# Vortex Flow Meter Wafer or Flange Connection. - Steam - Liquid - Gas

Save this PDF as:

Size: px
Start display at page:

Download "Vortex Flow Meter Wafer or Flange Connection. - Steam - Liquid - Gas"

## Transcription

1 Vortex Flow Meter Wafer or Flange Connection - Steam - Liquid - Gas

2 Working Principle & Circuit Diagram Working Principle When a column body placed in flowing fluids in pipe, a series of vortices will be generated alternately on each side of the object as shown as below, these eddies known as Karman Vortices, the frequency of the vortex shedding is related to the velocity of the fluid and the width of the body. Expressed by formula as below: f=st.v/d Thereinto f----frequency of Karman Vortex shedding St----Strouhal number v----velocity d----width of column object Because the frequency of the vortex shedding is proportional to the velocity, it can be used to calculate the instantaneous flowrate. Strouhal number is a very important coefficient in the Vortex Flowmeter. In the range of straight line of St 0.17 in curve, frequency of vortex shedding is proportional to the velocity, so as long as the frequency (f) be detected, the velocity (v) will be obtained, and volumetric flowrate will be got according to v. For KBLU Vortex Flowmeter, its frequency of the vortex shedding was detected by the stress force which exerted on the sensor (probe) using the piezoelectric unit which is built in sensor. 1

3 SPECIFICATIONS Accuracy: Liquid: Gas: Steam: ±1.0% of reading ±1.5% of reading ±1.5% of reading Materials: Housing: SS316 Shedder Bar: SS316 Converter housing and case cover: Aluminum Alloy Output Signal: (Where applicable) Sensor: Pulse signal (Low Level: 1V; High Level: 6V) Transmitter: 4 to 20 ma DC current signal Signal Transmission Distance: 500 m Electrical Connections: Protection Level: Vibration: ISO M Female IP g

4 OPERATION CONDITIONS Ambient: Temperature: -10C to +55 C Relative Humidity: 5% to 90% Power Supply: Sensor: Transmitter: Field Display Type: +12V DC (Optional: +24V DC) +24V DC Integral 3.2V Lithium Battery or +24V DC or 230V AC Fluid Temperature and Pressure: Temperature: Standard: -20 C to +70 C Optional: -20 C to +250 C; -20 C to +350 C Pressure: Media pressure should be limited according to flange rating. Measurable Flow Rate Range: For general liquids and gas: (See table 1) Table 1. Measurable Flow Rage Range for liquid and gas Nominal Liquid Gas Diameter (mm) (in.) Flow (m 3 /h) Frequency(Hz) Flow (m 3 /h) Frequency(Hz) to to to to to to to to to 35 9 to to to to 68 8 to to to to to to to to to to to to to to to to to to to to to to to to to to to to to to to 136 Note: The flow range for gas in table 1 is defined at operating state.

5 For Saturated Steam: (See table 2) Absolute pressure (MPa) Temperature ( C) Density (Kg/m 3 ) Table 2. Measurable Flow Rage for Saturated Steam DN25 DN40 DN50 DN65 DN80 DN100 DN125 DN150 DN200 DN250 DN300 Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Q min Q max Flow Unit Kg/h

6 Model Selection (See Table 4) Model Suffix Code Table 4. Model Selection Guide BFVM - / / / / / / Description 1 Flange connection Connection type 2 Wafer Connection (between flanges) Gas 1 Medium Liquid 2 Steam 3 Nominal Diameter (Code) Display Option Function Type EU Rating 25 25mm 40 40mm 50 50mm 65 65mm 80 80mm mm mm mm mm mm mm Z Integrated type (Optional: N; A; B; C; D) F Remote type (Optional: N; C) N A B C C1 D N Basic Type: +12V to +24V DC Power Supply; Pulse Output 4 to 20 ma current output Battery Power Supply with filed Display Field Display and 4 to 20 ma current output Field Display and RS485 24V DC; Temperature & Pressure Compensation Standard type

7

8 Mounting Orientation All of our vortex flow meters are designed to measure flow in only one direction. See marking on body Required Length of Straight pipe Runs Flow profile altering items such as elbows, valves and reducers can affect accuracy. See diagram 1 for typical flow meter system installation. Diagram 1. Typical Flow Meter System Installation The recommended guidelines are given to enhance accuracy and maximize performance. Distances given here are minimum requirements. Upstream: allow a minimum straight pipe length of at least 10 times the internal diameter of the pipe. For example, with 50mm pipe, there should be 500mm of straight pipe immediately upstream. Ideal upstream straight pipe length is 1000mm. Downstream: allow a minimum straight pipe length of at least 5 times the internal diameter of the pipe. For example, with 50mm pipe, there should be 250mm of straight pipe immediately upstream. Ideal upstream straight pipe length is 500mm.

9 Transducer Installed in Vertical Pipe When used in gas, transducer could be installed in vertical pipe with no considering of flow direction, but for gas contains few liquid, the gas flow direction should be from bottom to top. When used in liquid, flow direction should be from bottom to top, avoid extra weight of the liquid to exert on the probe. Lateral Installation in Horizontal Pipe Transducer can be installed laterally in horizontal pipe for all fluids. Particularly for superheated steam, saturated steam and cryogenic liquids, if allowed, transducer should be installed laterally to avoid amplifier from being affected greatly by temperature. Inverse Installation in Horizontal Pipe It is not suggested to install transducer inversely for normal gas or superheated steam, but it is suitable to saturated steam, high temperature liquid or the condition that dirty pipe needs cleaning frequently. 15

10 5. Requirement for Ambient Condition 1) Avoid transducer being installed in place where temperature changes greatly or heat radiation occurs. If it is necessary, take some measures to isolate thermal or ventilation. 2) Avoid transducer being installed in place where corrosive atmospheres surrounded. If it is necessary, take on some enforced ventilation measures. 3) It had better transducer be installed indoors. If sensor must be installed outdoors, bend the cable into U shape at the electrical port to avoid raining water into the box of amplifier along with the cable. 4) Sufficient space should be around transducer for convenient installation and regular maintenance. 5) Connection of transducer should be far away from electrical noise interference, such as big power transformer, electromotor and radio frequency interference etc. 6) There should be no frequency converter near transducer. Otherwise, normal performance will be affected by high frequency interference. 7) There should be no strong vibration source near transducer installing point. Damping measures should be taken once needed. Transducer Installed in Horizontal Pipe It is the most common way to install transducer in horizontal pipe. When used in gas, transducer should be installed at the top of pipeline if the gas contains few liquid. When used in liquid, transducer should be installed at the bottom of pipeline if the liquid contains little gas. 14

11 How to Select Pressure and Temperature Measuring Point If you require measuring pressure or temperature near the flowmeter, pressure measuring point should be 3~5D at downstream of transducer and temperature measuring point should be 6~8D at downstream of transducer. Preparation before Operation 1. Check the transducer mounting and wiring if it is correct; 2. Power on indicator and check if there is flow indication; 3. Open the valve slowly and stop once get a small pressure, then check if any leakage happens around the transducer or flow indication available in the indicator; 4. If condition is normal, open the valve fully and stabilize for a few minutes to check if it works properly. 20

12 See diagram 2 for straight pipe length requirement when there is a flow profile altering item Diagram 2. Number of Pipe Diameter (D=Diameter)

### For Multi-Parameter Meters see mvx

Design Features VORTEX IN-LINE FLOW METERS For Multi-Parameter Meters see m Principles of Operation n No moving parts to wear or fail. n Electronics can be remotely mounted up to 30.5 m (0 ft). n No fluid

### For Multi-Parameter Meters see mvx

VORTEX IN-LINE FLOW METERS Design Features For Multi-Parameter Meters see mvx Principles of Operation VX No moving parts to wear or fail. Electronics can be remotely mounted up to 30.5 m (0 ft). No fluid

### Vortex flowmeters. Product family introduction Principle of operation Product review Applications Key product features

Vortex flowmeters introduction Product review s Key product features This document should not be duplicated, used, distributed, or disclosed for any purpose unless authorized by Siemens. Page 1 Vortex

### PanaFlow MV82. Insertion Style Multivariable Flowmeter. GE Measurement & Control. Key Benefits. Applications

GE Measurement & Control PanaFlow MV82 Insertion Style Multivariable Flowmeter Key Benefits Multivariable vortex flowmeter for measuring volumetric flow, temperature, pressure, density, and mass flow using

### M111 Spiraflo Transducer

3309551/ 7 IM-P330-38 MI Issue 7 M111 Spiraflo Transducer Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Technical data 4. Mechanical installation: M111

### Coriolis Mass Flow Meter

Coriolis Mass Flow Meter TMFW Series Mini Type Coriolis Mass Flow Meter Xi an Tosilon Automation Co., Ltd No.299, Daqing Rd, Lianhu District, Xi'an Shaanxi, China Tel: +86-29-8823 8550 info@tosilon.com;

### TRZ 03-K Volumeter. Proven Technology. Superior Performance. TRZ 03-K Volumeter is a Turbine Gas Flowmeter for Secondary Gas Metering.

TRZ 0-K Volumeter TRZ 0-K Volumeter is a Turbine Gas Flowmeter for Secondary Gas Metering. TRZ 0-K Turbine Gas Flowmeter continues the tradition of Honeywell s field-proven turbine meter technology, delivering

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### Flow Measurement SITRANS F X

SITRANS FX00 Overview The main applications of SITRANS FX00 can be found in the following sectors: Chemical Petrochemical Oil & Gas Power plants -Air - Heating - Cooling -Chilling Food & beverage - Pharmaceutical

### No E. General Information. Averaging Pitot Tube Flow Sensor HITROL CO., LTD.

No. 2000-03E Averaging Pitot Tube Flow Sensor HITROL CO., LTD. M e a s u r e m e n t Pr i n c i p l e Measurement principle of Hitbar sensor is explained below. When a Hitbar is installed in the pipeline

### 3-way control valves VK 3

Data sheet 3-way control valves VK 3 Description / Application VK 3 valves are a range of 3 port high specification flanged valves for chilled water, LPHW, MPHW, HPHW (low, medium or high pressure hot

### Steam Flow Meter STEAMcube Model MVC30A (Integral Type) Model MVC31A (Integral Type for large flow)

OVERVIEW Steam Flow Meter STEAMcube Model MVC30A (Integral Type) Model MVC31A (Integral Type for large flow) The steam flow meter STEAMcube is designed exclusively to measure saturated steam. Because steam

### Over 20,000 Strain Gage Target flowmeters installed since 1952.

Over 20,000 Strain Gage Target flowmeters installed since 1952. Liquid, gases, superheated and saturated steam from -320 to 500 O F and up to 15,000 psi. No frictional moving parts. Designed to withstand

### 2-Port Seat Valves with Flange, PN 16

4 45 2-Port Seat Valves with Flange, PN 6 VVF45... dular cast iron EN-GJS-400-5 valve body DN 50...50 k vs 9...00 m /h Can be equipped with SKB...- or SKC...- electrohydraulic actuators Use For use in

### Certified according to DIN EN ISO 9001 Technical Datasheet TRICOR Series Mass Flow Meters

www.kem-kueppers.com info@kem-kueppers.com Certified according to DIN EN ISO 9001 Technical Datasheet TRICOR Series Mass Flow Meters Description The Tricor Mass Flow Meters measure simultaneously mass

### The Discussion of this exercise covers the following points:

Exercise 3-2 Orifice Plates EXERCISE OBJECTIVE In this exercise, you will study how differential pressure flowmeters operate. You will describe the relationship between the flow rate and the pressure drop

### Vortex Flowmeter EF73

() 26 May 2011 ISO 9001/ ISO 14001 Kakogawa, Japan is approved by LRQA LTD. to ISO 9001/14001 Vortex Flowmeter EF73 Copyright 2009 by TLV CO., LTD. All rights reserved Rev 2-2009 Contents Contents... 2

### Pressure Measurement Single-range transmitters for general applications

Siemens A 207 Overview Application The SITRANS P Compact pressure transmitter is designed for the special requirements of the food, pharmaceutical and biotechnology industries. The use of high-grade materials

### SITRANS F C MASSFLO. MASS flowmeters Sensor type MASS MC1 DN 50 - DN 100. Manual, supplement *083R9505* Order no.: FDK:521H1145

s Manual, supplement SITRANS F C MASSFLO MASS flowmeters [ ] Order no.: FDK:521H1145 SFIDK.PS.28.P3.2 - A5E253681 *83R955* Contents Page Introduction... 2 Siemens Flow Instruments range of coriolis mass

### Liquid level instruments BM 24 /BM 51

KROHNE 09/2001 7.02362.21.00 GR/PRINTO Liquid level instruments / Status: 10/99 Variable area flowmeters Vortex flowmeters Flow controllers Electromagnetic flowmeters Ultrasonic flowmeters Mass flowmeters

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Bypass MFC with capillary technology for nominal flow rates from 5 ml N /min to 15 l N /min Applicable for aggressive gases Compact design and digital communication

### JUMO dtrans p30 Pressure Transmitter

609 Fulda, Germany Postal address: 605 Fulda, Germany Phone: +49 661 600-0 Fax: +49 661 600-607 Phone: +44 179 6 55 Data Sheet 40466 Page 1/6 JUMO dtrans p0 Pressure Transmitter General application Pressure

### Orifice Plate Flow Meter

Orifice Plate Flow Meter FOFT Measuring Principle The principle of measuring the differential pressure is based on the fact that a differential pressure is created across that section of a pipe where its

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### MEASUREMENT BEST PRACTICES FORIMPROVEDREFINERY SAFETY, AVAILABILITY & EFFICIENCY

MEASUREMENT BEST PRACTICES FORIMPROVEDREFINERY SAFETY, AVAILABILITY & EFFICIENCY Mark Menezes, PlantWeb Marketing Manager Rosemount Inc. Eric Wickberg, Pressure Applications Marketing Manager Rosemount

### ProcessMaster FEP610 Electromagnetic flowmeter

Change from one to two columns Data Sheet DS/FEP61-EN Rev. A ProcessMaster FEP61 Electromagnetic flowmeter The new flowmeter: The first choice for all industrial standard applications. Measurement made

### Technical Manual. Liquid Level Transmitter CT801-LB/S

CT801-LB/S Technical Manual SAS au Capital de 2 158 244-444 871 933 R.C.S. Bourges - APE : 2651B Headquarter : 9, rue Isaac Newton - 18000 Bourges - France Technical Manual CT801-LB/S 1 st Edition Released

### Liquid Level Measurement

Technical Article Liquid Level Measurement A pressure transmitter can be used to determine the liquid level in a tank, well, river or other body of liquid. The pressure at the bottom of a liquid filled

### Unique mounting features for Thermal Mass Flow Meters. By Martin Raab in cooperation with CaTeC b.v.

Unique mounting features for Thermal Mass Flow Meters By Martin Raab in cooperation with CaTeC b.v. Development and production of sensors and transmitters for: humidity dew point CO2 O2 air velocity mass

### Ratio Regulator Zero Governor. FRG/6 Series

Ratio Regulator Zero Governor FRG/6 Series CSA Certified ANSI Z21.18 / CSA 6.3 CSA Requirement # 5.01 Zero Governor and Gas/Air Regulator File # 157406 US and Canadian Models FRG 705/6 FRG 707/6 FRG 710/6

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 03-28-2012 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### POP Safety Valve. POP Safety Valve INTRODUCTION DEFINITIONS

POP Safety Valve POP Safety Valve INTRODUCTION The effects of exceeding safe pressure levels in an unprotected pressure vessel or system, can have catastrophic effects on both plant and personnel. Safety

### Pressure and/or Temperature Pilot Operated Steam Regulators Series 2000

Hoffman Specialty Regulators Regulators Pressure and/or Temperature Operated Regulators Series 2000 The Hoffman Specialty Series 2000 consists of main valves, pilot valves, wells and hardware kits. They

### Programmable Submersible Level Transmitters

Programmable Submersible Level Transmitters PTM/N/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available as

### Two-port seat valves

4 345 Two-port seat valves with flange, PN16 VVF45... Two-port seat valves with flange, PN16 Spheroidal cast iron GGG-40 DN50...150 mm k vs 19...300 m 3 /h Stroke 20 or 40 mm Can be equipped with actuators

### MB260/MB261/MB264 Submersible Level Transmitter

MB260/MB261/MB264 Submersible Level Transmitter Features Separate construction. Full sealed stainless steel construction for submersible part; sensor and amplifier circuit are all in the sealed housing;

### THERMISTOR TYPE VORTEX FLOWMETER for GAS MEASUREMENT. DELTAFLOW Type F

Gas THERMISTOR TYPE VORTEX FOWMETER for GAS MEASUREMENT DETAFOW Type F Standard and Gas Purge Types MODE: VF series GENERA SPECIFICATION GS.No. GENERA This flowmeter is a kind of vortex flowmeters, which

### GA24 Technical Datasheet

GA24 Technical Datasheet Variable-area flowmeter Sturdy construction for several applications Local indication without auxiliary power Replaceable mounting parts KROHNE CONTENTS GA24 1 Product features

### THERMAL MASS FLOW MEASUREMENT FOR GASES

THERMAL MASS MEASUREMENT FOR GASES Make the Wise Choice. Choose Sage Flow Meters. A Commitment to Higher Performance SAGE METERING is a manufacturer of high performance Thermal Mass Flow Meters which measure

### SINGLE VALVE WITH LOW-FLOW BYPASS

CONTROL VALVES Pressure Reducing Valve Sizing Guide Sizing pilot operated reducing valves is not a complicated process. It starts with determining requirements and following these guidelines in valve size

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### INTERNATIONAL OIML R 32 RECOMMENDATION. Rotary piston gas meters and turbine gas meters ORGANISATION INTERNATIONALE INTERNATIONAL ORGANIZATION

INTERNATIONAL OIML R 32 RECOMMENDATION Edition 1989 (E) Rotary piston gas meters and turbine gas meters Compteurs de volume de gaz à pistons rotatifs et compteurs de volume de gaz à turbine OIML R 32 Edition

### HUMOR 20. High-precision Humidity Calibrator HUMOR 20. Operation

High-precision Humidity Calibrator The role of humidity calibrations that are accurate, reproducible, and documentable is becoming more and more important. ISO quality guidelines and regulations according

### Pressure Measurement Single-range transmitters for general applications

Siemens AG 07 Overview Function U const. p U I EM Sensor Connection for auxiliary power supply Vent pipe with humidity filter Protective conductor connection/ Equipotential bonding Diaphragm The pressure

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 26.06.2014 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### E8AA. Pressure Sensor of Stainless Steel Construction Is Ideal for a Wide Range of Applications. Pressure Sensor (Stainless Steel Diaphragm)

Pressure Sensor (Stainless Steel Diaphragm) CSM DS_E_3_1 Pressure Sensor of Stainless Steel Construction Is Ideal for a Wide Range of Applications Incorporates double diaphragms consisting of SUS316L stainless

### GasMultiBloc Combined regulator and safety valve Infinitely variable air/ gas ratio control mode MBC-300-VEF MBC-700-VEF MBC-1200-VEF

GasMultiBloc Combined regulator and safety valve Infinitely variable air/ gas ratio control mode MBC-00-VEF 7.03 Max. operating pressure 30 mbar Compact design High flow values Low weight Low power consumption

### Regulating valves Type REG 6-65 Type REG-SS REFRIGERATION AND AIR CONDITIONING. Technical leaflet

Regulating valves Type REG 6-65 Type REG-SS 15-40 REFRIGERATION AND AIR CONDITIONING Technical leaflet Content Page Introduction........................................................................................3

### Truck-mounted Mass Flow Metering for LPG Delivery

Truck-mounted Mass Flow Metering for LPG Delivery Itron, Inc. 1310 Emerald Rd. Greenwood, SC 29646 Introduction The mathematician Gustave De Coriolis first described the Coriolis Effect, which is an inertial

### Expert Hydrostatic Level Transmitters

Expert Hydrostatic s General Features MJK Expert hydrostatic level transmitters are designed for level measurement by submerging the transmitter in open channels, drains and tanks. Expert hydrostatic level

### Wedge Type Flow Meter Installation and Operation Manual

Wedge Type Flow Meter Installation and Operation Manual 358-EN Please read and save these instructions. Wedge Type Flow Meter PLEASE READ THIS FIRST! The purpose of this document is to provide guidance

### Transmitter CS 21 Operation Manual

Transmitter CS 21 Operation Manual Content Page For your Safety 3 General Description 3 Detection Principle 4 Operational Notes 4 Design 4 Mounting Position of CS21 5 Mounting 6 Installation of Electrical

### Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters

ABB MEASUREMENT & ANALYTICS WHITE PAPER Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters Remote seals protect transmitters when measuring difficult

### Proline Prowirl 72F, 72W, 73F, 73W

Technical Information Proline Prowirl 72F, 72W, 73F, 73W Vortex flow measuring system Reliable flow measurement of gas, steam and liquids Application For the universal measurement of the volume flow of

### 776 Cryogenic Safety Valve

776 Cryogenic Safety Valve INTRODUCTION 776 Cryogenic Safety Valve The effects of exceeding safe pressure levels in an unprotected pressure vessel or system, can have catastrophic effects on both plant

### Process Solutions. Electronic Turbine Meter TERZ 94 PROVEN TECHNOLOGY. SUPERIOR PERFORMANCE.

Process Solutions Electronic Turbine Meter TERZ 9 PROVEN TEHNOOGY. SUPERIOR PERFORMNE. Electronic Turbine Meter TERZ 9 Method of Operation The TERZ 9 electronic turbine meter is a flow meter which directly

### Level transmitter LT 100

KROHNE 09/2001 7.02363.21.00 GR/PRINTO Level transmitter LT 100 Status: 10/99 Variable area flowmeters Vortex flowmeters Flow controllers Electromagnetic flowmeters Ultrasonic flowmeters Mass flowmeters

### Flow Measurement SITRANS F C

Overview The flow measuring principle is based on the Coriolis Effect. The FCS400 sensor s measuring tubes are energized by an electromechanical driver circuit which oscillates them at their resonance

### Rosemount 8600 Series Utility Vortex Flowmeter

Rosemount 8600 Series Utility Vortex Flowmeter Product Data Sheet November 2013 00813-0100-4860, Rev BA The Rosemount 8600 delivers superior reliability for general purpose applications. Rosemount Reliability

### STANDARD SPECIFICATION FOR FLOW INSTRUMENTS DOCUMENT NO : 44-LK /J.02/0006-A4

Page 1 of 9 ` STANDARD SPECIFICATION FOR FLOW INSTRUMENTS DOCUMENT NO : 44-LK-5102-00/J.02/0006-A4 Rev No. Issue Date Pages Rev Description Prepared A 12.08.08 9 Issued for Comments / Approval By Checked

### Turbine Flowmeters for Gas Applications

Turbine Flowmeters for Gas Applications Description Flow Technology s FT Series turbine flowmeters utilize a proven flow measurement technology to provide exceptionally reliable digital outputs. Because

### Electronic gas volume corrector model DGVC-04

Electronic gas volume corrector model DGVC-04 1. Introduction The following user s manual gives information about the installation, configuration, usage and storage of the electronic gas volume corrector

### Proportional pressure regulators VPPE

Proportional pressure regulators VPPE Proportional pressure regulators VPPE Product range overview Function Version Pneumatic connection Proportional pressure regulator Nominal size for pressurisation/

### Exercise 5-2. Bubblers EXERCISE OBJECTIVE DISCUSSION OUTLINE. Bubblers DISCUSSION. Learn to measure the level in a vessel using a bubbler.

Exercise 5-2 Bubblers EXERCISE OBJECTIVE Learn to measure the level in a vessel using a bubbler. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Bubblers How to measure

### Exercise 4-2. Centrifugal Pumps EXERCISE OBJECTIVE DISCUSSION OUTLINE DISCUSSION. Pumps

Exercise 4-2 Centrifugal Pumps EXERCISE OBJECTIVE Familiarize yourself with the basics of liquid pumps, specifically with the basics of centrifugal pumps. DISCUSSION OUTLINE The Discussion of this exercise

### Hydrostatic & Submersible. Level instruments. for the Australian market

Hydrostatic & Submersible Level instruments for the Australian market Hydrostatic and Submersible level transmitters Hydrostatic and submersible level instruments use the head pressure of a fluid to infer

### GRUNDFOS DATA BOOKLET. Hydro Solo-E. Complete pressure boosting systems 50/60 Hz

GRUNDFOS DATA BOOKLET Complete pressure boosting systems 5/6 z Contents Product data Performance range 3 Operating conditions Inlet pressure Type key Product range 5 Construction 5 Installation 5 Mechanical

### Type 299 Pressure Reducing Regulators

Type 299 Pressure Reducing Regulators Fisher, Fisher-Rosemount, and Managing The Process Better are marks owned by Fisher Controls International, Inc. or Fisher-Rosemount Systems, Inc. All other marks

### DTG - LCD Digital Temperature Gauge USER MANUAL

DTG - LCD USER MANUAL Page 1 of 7 USERS GUIDE page 1. Description 3 2. Function 3 3. Safety Instruction 3 3.1. Safety Conventions 3 3.2. Proper Use 3 4. Installation 4 4.1. Unpacking 4 4.2. Storage 4 4.3.

### GRUNDFOS alldos product information. Vaccuperm 111 / 113. Vacuum gas dosing systems for chlorine gas, ammonia, sulphur dioxide and carbon dioxide

GRUNDFOS alldos product information Vacuum gas dosing systems for chlorine gas, ammonia, sulphur dioxide and carbon dioxide Construction and function Modular system Vacuum regulator for direct mounting

### Residential Meters. Available innov8 Outputs 900 MHz Radio Virtual Network 3W Standard AMR 2W AMR 2W Standard Touchpad Scalable Pulse 4-20mA SCADA

The Spectrum Single-Jet Meter is the widest range, single measuring element meter available to U.S. utilities. They have been designed to replace limited range displacement and multi-jet meters. The Spectrum

### INSTRUCTION MANUAL. process. Ultrasonic Flowmeter Range

INSTRUCTION MANUAL process TM Ultrasonic Flowmeter Range process TM CONTENTS Page No 1 General 4 2 Order Codes 4 3 Installation 5 & 6 4 Specifications 7 & 8 1 General 3 Installation The Process Atrato

### VORTEX FLOWMETER. Operation Manual Version:2012-6V

VORTEX FLOWMETER Operation Manual Version:2012-6V01 2012.06 1 I. Summary Vortex Flowmeter is one kind of main flowmeters in the international for detection and metering the liquid, gas and steam. It is

### Autocalibration Systems For In Situ Oxygen Analyzers

Product Data Sheet PDS 106-340AC SPS 4000, IMPS 4000, and MPS 3000 January, 1999 Autocalibration Systems For In Situ Oxygen Analyzers Cost-effective autocalibration systems for installations ranging from

### Components for air preparation and pressure adjustment. OUT port position ( ) connected Rear side. of IN port. Air tank. directly.

Components preparation and pressure adjustment ABP Overview ABP is a component that enables boosting by s only up to twice primary pressure (.0MPa max.) in combination with using air tank but not using

### AMS 2710 PCB pressure sensor module with V output

FEATURES Universal pressure sensor module with 0.. 10 V voltage output Fully calibrated and temperature compensated sensor module Variants for (bidirectional) differential, gage, absolute and barometric

### L3 Pressure and Level transmitter

SENSORS FOR AND BIOPHARMA. Product Information L3 L3 Pressure and Level transmitter Range of applications Hydrostatic Level measurement in dynamic temperature environments Pressure measurement in pipes

### Bourdon tube pressure gauge for diaphragm seals

Bourdon tube pressure gauge for diaphragm seals and switch function, Type series BR42.. Features Bourdon tube pressure gauge for diaphragm seal and switch function Nominal range -1 3 bar to -1 15 bar,

### Seated valves VRB 3, VRG 3

Data sheet Seated valves VRB 3, VRG 3 Description VRB, VRG valves provide a quality, cost effective solution for most water and chilled applications. These valves may be used with glycol concentrations

### 7000 Series OPTIMASS Mass Flowmeters. Titanium Hastelloy Stainless Steel

KROHNE 04/2003 7.02445.71.00 GR 7000 Series OPTIMASS Mass Flowmeters with single straight measuring tube Titanium Hastelloy Stainless Steel One tube no limits No limits with tube material No limits with

### Purgemaster Series 10A6100 SNAP-IN. Instruction Bulletin Replaceable Parts List. D184B003U35 Rev. 01 /

SNAP-IN Purgemaster Series 0A600 Instruction Bulletin Replaceable Parts List D84B003U35 Rev. 0 / 03.200 You have purchased a high quality, modern instrument from ABB Automation Products. We thank you for

### Flow Sensors & Hydrometers Specification Sheet

Baseline Flow s and Hydrometers Baseline s smart flow sensors and hydrometers are two-wire ready flow measurement and control devices that are pre-programmed with the correct K and Offset values. In addition,

### 756 Safety Relief Valves

756 S a fe t y R e l i e f Va l ve s INTRODUCTION 756 Safety Relief Valves The effects of exceeding safe pressure levels in an unprotected pressure vessel or system, can have catastrophic effects on both

### Averaging Pitot Tubes

Data Sheet DS/TORBAR-EN Issue 1 Averaging Pitot Tubes Torbar Unique profile shape offers high flow turndown No drift in co-efficient ensures long term stability One-piece outer tube for pipes up to 5000mm

### MASS-STREAM TM Instruction Manual. D-6200 Analog Mass Flow Meters / Controllers

MASS-STREAM TM Instruction Manual D-6200 Analog Mass Flow Meters / Controllers Doc. No.: 9.17.112A Date: 17-03-2017 ATTENTION Before installing and operating the instrument it is strongly recommended that

### Materials : Galvanized steel flanges

Size : Ends : Min Temperature : Max Temperature : DN 32 to 200 Flanges PN10/16-10 C + 80 C Max Pressure : 16 Bars Specifications : Absorb vibrations and noises Linear and angular compansion Single NBR

### Rosemount 2130 Extreme Temperature Vibrating Fork Liquid Level Switch

Technical Note ommunicate Rosemount 2130 Rosemount 2130 Extreme Temperature Vibrating Fork Liquid Level Switch The Rosemount 2130 Vibrating Fork Level Switch, based on vibrating short fork technology,

### Turbine Flowmeters for Gas Applications (Metric)

Turbine Flowmeters for Gas pplications (Metric) Description Flow Technology s FT Series turbine flowmeters utilize a proven flow measurement technology to provide exceptionally reliable digital outputs.

### New! ProFlow. Sliding Vane Meters DN (½ - 2 ) _ 142. Product Bulletin

New! ProFlow Sliding Vane Meters DN 15-5 (½ - 2 ) _ 142 Product Bulletin WWW.VAF.NL 1 Introduction VAF Instruments ProFlow positive displacement sliding vane type liquid flowmeters are used in continuous

### SSL. Smart Valve Positioner.

Smart Valve Positioner Smartest valve control device meeting a dynamic performance and a precise setting with a piezoelectric technology and an optimized auto-calibration program Features Auto-Calibration

### Thermal Energy Flowmeter Model KEC. Application Range. Special Features

Thermal Energy Flowmeter for gases measuring monitoring analysing KEC O O Direct mass flow rate measurement of gases OMeasuring O accuracy: ± 0.3% of full scale ±1.5% of reading on request: ± 0.3% of full

### EcoPRO-Series 4-20mA SUBMERSIBLE PRESSURE TRANSMITTER INSTRUCTIONS

EcoPRO-Series 4-20mA SUBMERSIBLE PRESSURE TRANSMITTER INSTRUCTIONS EcoPRO-Series 4-20mA SUBMERSIBLE PRESSURE TRANSMITTER INSTRUCTIONS 9001:2008 ISO CERTIFIED COMPANY Table of Contents Introduction - EcoPRO-Series

### EXPERIMENTAL RESULTS OF GUIDED WAVE TRAVEL TIME TOMOGRAPHY

18 th World Conference on Non destructive Testing, 16-20 April 2012, Durban, South Africa EXPERIMENTAL RESULTS OF GUIDED WAVE TRAVEL TIME TOMOGRAPHY Arno VOLKER 1 and Hendrik VOS 1 TNO, Stieltjesweg 1,

### Technical Data Sheet. OPTIMASS 7000 Custody Transfer Mass Flowmeter

Technical Data Sheet KROHNE 12/2004 7.02516.21.00 CH/GR OPTIMASS 7000 Custody Transfer Mass Flowmeter with single straight measuring tube OIML R117 Pattern Approval Accuracy classes 0.3 / 0.5 / 1.0 / 1.5

### Regulating valves REG-SA and REG-SB

MAKING MODERN LIVING POSSIBLE Technical brochure Regulating valves REG-SA and REG-SB REG-SA and REG-SB are angleway and straightway hand regulating valves, which act as normal stop valves in closed position.

### STAD-D. Balancing valves Balancing valve for domestic water systems, DN Balancing valves

STAD-D Balancing valves Balancing valve for domestic water systems, DN 15-50 Balancing valves IMI TA / Balancing valves / STAD-D STAD-D The STAD-D balancing valve delivers accurate hydronic performance

### KMT. Thermal Mass Flowmeter. for compressed air and gases. Measuring range: Nm³/h Nm³/h

Thermal Mass Flowmeter for compressed air and gases measuring monitoring analysing KMT Measuring range: 0.32... 63 Nm³/h - 263.4... 263350 Nm³/h Accuracy: ± 1.5% of reading + 0.5% of full scale ( DN 50)

### HUMOR 20. High Accuracy Humidity Calibrator. Water T, p1 HUMOR 20 V1.0

High Accuracy Humidity Calibrator The role of humidity calibrations that are accurate, reproducible, and documentable is becoming more and more important. ISO quality guidelines and regulations according