# A Primer on Vacuum Pressure Measurement

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 A Primer on Vacuum Pressure Measurement Steve Hansen This article appeared in Vacuum Coating & Technology, May 2009 It's often said that we live at the bottom of a sea of air. Looking more closely at the atmosphere around us it is composed of a variety of molecules such as nitrogen, oxygen, water vapor, carbon dioxide, argon and a host of other natural and manmade substances. The density of this mixture of molecules is determined by such factors as the total depth of the atmosphere, ambient temperature, altitude and meteorological conditions. At the microscopic level, the molecules are in motion and the combined force exerted by this vibrating mass of molecules on a surface is something we call pressure. Vacuum technology is concerned with reducing the density of molecules to some predetermined level in a contained environment and, more often than not, replacing the mixture that we call air with some other mixture of gases. The resulting environment: gas composition and density (pressure), is dictated by what we want to do in that environment, be it operating a surface analysis instrument, depositing films, or making a neon sign. That being said, the ability to accurately and reproducibly measure pressure is key to operating a successful vacuum application. Pressure Measurement - The Beginnings Ask a variety of people what "normal" sea-level atmospheric pressure is and you will get answers such as 14.7 pounds per square inch, 760 Torr, inches of mercury, 1 bar, 100,000 Pascal, etc. What do these units mean and where do they come from? In the middle of the 17th century, Evangelista Torricelli ( ) was the first person to "measure" atmospheric pressure. He did this will his invention, the mercury barometer. This device consists of a glass tube, about a meter in length, that is sealed at one end. To set up the barometer, the tube is first filled with mercury and then inverted with the open end placed in a cup of mercury. What is observed is that the mercury column drops to some level in the tube. Since the space at the top of the tube was originally full of mercury (that is, free of air), the space now contains a reasonably good vacuum. The actual height of the mercury column in the tube then represents the weight of mercury that can be lifted by the difference in pressure between the pressure exerted on the mercury in the cup by the atmosphere and the pressure at the top of the tube (close enough to zero for the Renaissance vacuum experimenter). At sea level, Torricelli observed that the typical height of the mercury column was about 760 millimeters. This would rise or fall depending upon the weather (hence the application of the barometer in meteorology). Other investigators found that the height of the column decreased when a barometer was carried up a hillside. Analyzing this a bit more, we can observe three things (see Figure 1): The barometer is measuring the pressure difference between the ambient atmosphere and a zero pressure reference. (Of course, the reference is not quite at zero pressure but it is at a pressure that is negligible as compared to what is being measured.) Thus, the pressure measurement is termed absolute because it is referenced to zero. If the barometer were carried toward space, the column height would drop to zero as the earth s atmospheric pressure dropped to zero.

2 It is the force of the atmosphere on the mercury as applied over the surface area of the cup i.e. force per unit area that is holding the mercury column up. This is the proper definition of pressure. Devices such as barometers where the pressure is measured by the action of the gas on a moveable element are termed direct gauges. The moveable element can be a liquid column, a diaphragm, a bellows, a coiled tube or anything else that will move in response to the pressure of the gas. The reading is the height of the column (length) for the particular working material (mercury). The proper way of expressing the reading is mmhg for millimeters of column height for the given working fluid. If water, which has a specific gravity 13 times lower than mercury were used, the column height for the same pressure would be that much more, i.e. about 30 feet. That brings us to the issue of pressure measurement units. We'll examine these in the next section. Figure 1. Absolute and atmosphere (gage) referenced pressures. Pressure Units For the reasons noted above, the traditional way to express pressure in a vacuum system is in terms of the height of a column of mercury that may be supported by the pressure in the system. This developed from the use of the barometer and later mercury column manometers (such as the McLeod gauge) as the primary way of measuring sub-atmospheric pressures. To summarize, one standard atmosphere of pressure will support a mercury column 760 millimeters (29.92 inches) high. One millimeter of mercury (mmhg) is the same as 1 Torr, a unit named in honor of Torricelli. A thousandth of 1 mm Hg is referred to as 1 micron Hg or, in more current terminology, 1 millitorr (mtorr). But, none of these is actually a measure of pressure. As stated before, pressure is properly defined as the force per unit area that the gas exerts on a surface. Depending upon what set of units we are working with or are accustomed. For example we can express

3 760 Torr as equivalent to 14.7 pounds per square inch (psi). If we are using the metric system the same pressure is about 1.03 bar (units of dynes/sq. cm) in cgs (centimeter-gram-second) units. In the professional scientific literature and in Japan the SI (Système International) units that are based on the meter and kilogram are normally used. Here, pressure is stated properly in terms of newtons/sq. meter or Pascal (Pa). In this system, 1 standard atmosphere is expressed as 101,300 Pa. So, in the US we tend to stay with torr as the unit has a long history and people are familiar with it. Just remember that, properly speaking, the torr isn't really a unit of pressure. Pressure References Since we live our lives at this thing we call atmospheric pressure, we have a perception that vacuum is anything less than 1 atmosphere and pressure is anything above. Thus, to a person who lives in Boston, anyone who lives in Denver is living in a vacuum. (Normal atmospheric pressure at Denver's mile-high altitude is about 660 Torr.) This atmospheric reference that we have produces some difficulties when we talk about measuring vacuum and we have to be careful about what the reference is against which we are measuring pressure. As we've seen, the barometer measures absolute pressure. The perfect zero pressure reference would be a vessel with absolutely no gas molecules in it. Once even one molecule is introduced, we have some pressure, albeit very small. Then, for every doubling of molecular density, we have a doubling of pressure. (See Table 1.) Pressure (Torr) Molecules/Liter x x x x x x x million Table 1. Pressure expressed as molecular density (at 0 0 C). Absolute pressure is generally used when it is important to know exactly how much "stuff" is in a given volume. Molecular densities, hence absolute pressure, are very important in vacuum apparatus and process tools as critical parameters such as mean free path and vapor pressures are related directly to absolute pressure. Another way of expressing pressure is with reference to our normal, but varying, atmospheric pressure. This is often called gage pressure. Gage pressure simply tells you how far away you are from the ambient reference. Common dial gauges are an example of a gage, or atmosphere referenced gauge. In process systems, gage pressure is useful in applications such as loadlock monitoring and control where it is important or necessary to open the chamber right at the prevailing atmospheric pressure. Differential pressure is another term that is often encountered. This is simply the pressure difference that exists between any two points in a system. For example, the differential pressure across an in-line filter might be monitored to insure proper air flow or to provide a warning in the even of clogging.

4 Direct and Indirect Gauges While the barometer will provide an absolute pressure indication, its resolution is low. Most vacuum processes operate at pressures below 1 Torr and you simply can't measure such a minuscule column height. As vacuum technology improved during the 19th century it was necessary to develop gauges that could accurately indicate lower and lower pressures. After all, how can you brag about the quality of your pump if you don't have a way of telling how good a vacuum it will produce? The answer to the gauging problem was a sort of amplifying barometer that was developed by Herbert McLeod ( ) in The McLeod gauge is a mercury manometer in which the gas whose pressure you are trying to measure is compressed by a known amount, usually in the range of ten to one hundredthousand fold. By compressing the gas the height of the mercury column is correspondingly increased, providing a visually observable reading. Good McLeod gauges could provide accurate gas pressure measurements to the 10-5 Torr range providing there were no condensable gases (such as water vapor) in the sample. (The compression action will turn these condensables to liquid and therefore will not be part of the pressure measurement.) McLeod and similar mercury manometers were difficult to fabricate, temperamental in operation, and fragile. These negatives led to the rise of another class of vacuum gauge that could be quite simple in construction and yet provide fairly good measurements. The trick in these gauges was not to measure pressure (force/area) but some property of the gas that predictably varies with pressure. These gauges are termed indirect gauges. Many 20th century researchers have spent a good part of their careers developing pressure gauges that measure anything but pressure. A large class of indirect gauge are those that depend upon the variation of thermal conductivity of the gas whose pressure is being measured. The specific quantity being measured is the molecule to molecule heat transfer from a heated element to the shell of the gauge tube. It turns out that thermal conduction is essentially constant in the higher pressure viscous flow regime (i.e. at pressures in excess of a few Torr) so these gauges have a high pressure limit. It also turns out that gases become very effective thermal insulators in the molecular flow regime (i.e. at pressures below about 10-3 Torr). However, in the region between these pressures, thermal conductivity will vary with pressure. So, if one has a heated filament within an envelope, at higher pressures there will be some level of heat transfer from the filament causing it to operate at some temperature, assuming a given power input to the filament. Then, as the pressure is lowered, the thermal conduction of the gas decreases and the filament will tend to increase in temperature. The temperature of the filament can be sensed with a thermocouple element that contacts the filament wire. This type of gauge is therefore termed a thermocouple gauge. Alternatively, the filament's temperature may be determined by the resistance of the filament wire: as the filament heats up, its resistance will also increase. This type of gauge is termed a Pirani gauge after its inventor, Marcello Pirani ( ). Pirani's gauges were based on electric light bulb technology and indeed, Pirani was the laboratory head at Siemens & Halske AG s bulb factory in Berlin. Simple and relatively inexpensive, the thermocouple gauge has some deficiencies. One is the mode of operation: as the pressure changes, the filament and thermocouple have to arrive at a new temperature.

5 Since these elements have some amount of mass, the reading will lag the actual pressure by as much as several seconds. As a result, this type of gauge will not sense fast changes in pressure. Although the principle is the same, the Pirani gauge has followed a different evolutionary path. Most notably, the filament is operated at a constant temperature. As the pressure changes, a bridge circuit varies the amount of power delivered to the filament in order to hold its resistance, hence its temperature, at a given value. Since there is no thermal time lag, the response of a Pirani gauge is quite fast, usually on the order of a tenth of a second or less. Other niceties in the Pirani gauge include temperature compensation to adjust for changing ambient temperature and a low filament temperature that helps to prevent contaminants that might be in the vacuum system from baking on to the filament. These provide for a more stable, repeatable gauge. Finally, the parameters associated with the Pirani filament (dimensions, temperature, electrical characteristics, etc.) permit usable pressure measurements up to about 100 Torr. Thermocouple and Pirani gauges are most frequently seen in pressure monitoring applications: indicating proper pump down performance, foreline pressures and so forth. As the pressure in the Pirani gauge approaches its upper limit, its sensitivity declines as a hot, stationary boundary layer forms around the filament. However, if the tube is properly constructed, as the pressure further increases strong convection currents of air will start to flow around the filament. The convection, as it gets stronger at higher pressures, will have a cooling effect on the filament. This convection cooling is exploited in a special form of the Pirani gauge, called the convection enhanced Pirani. With two modes of heat conduction, molecular at lower pressures and convection at higher pressures, a convection Pirani can provide continuous measurements over a 6-decade range, 1000 Torr down to 1 millitorr. (In some cases these gauges can measure to about 0.1 millitorr.) This type of gauge is frequently used to monitor the pump down of chambers from atmosphere. It must be noted that convection Pirani gauges must be mounted in a given orientation in order to work properly in the convection mode. Figure 2 shows a convection Pirani gauge and a typical pressure vs output curve. Note the non-linearity of the curve.

6 Pressure (Torr) 1000 Convection Mode To Electronics 100 Filament Pirani Mode (Molecular Conduction) 10-2 Molecular Conduction Convection Current Pirani Transducer Output Voltage (Nitrogen) Figure 2. Schematic of a convection Pirani gauge with pressure vs voltage relationship. Illustration courtesy of MKS Instruments, Inc. Figure 3. Convection Pirani sensor and controller. Photo courtesy of MKS Instruments, Inc. Measuring Lower Pressures - Ion Gauges While the thermal conductivity gauges provided a good solution to the measurement of pressures down to the upper end of the high vacuum realm, it won't work at lower pressures because the physical basis upon which the gauge works simply ceases to operate. The early researchers had to find another gas property that varies with pressure, one that would work at lower pressures. This lead to the development

7 of the ion gauge. Ion gauges developed from early vacuum tube technology where it was observed that the current drawn from the positive element (called the plate in a vacuum tube and a collector in a gauge tube) in a triode tube was related to the gas pressure in the tube, other conditions being constant. Figure 4 illustrates the principle. Here we have an electron tube with three elements: an incandescent filament of refractory metal (pure tungsten or coated iridium) which provides a source of electrons through the process of thermionic emission, a positively biased fine wire grid that accelerates the electrons to a fixed energy though the volume of the tube, and a negatively biased collector. Some of the electrons that are passing through the tube will interact with the gas molecules, producing positive ions through a process called electron impact ionization. If the electron flux is held constant (done by controlling the current between the filament and the grid), the number of ions that are generated will be directly related to the density of gas molecules within the tube, that is, the collector current is directly related to the pressure. Collector Ion + Grid Constant Emission (Electron) Current Volts e Volts Filament Ion Current Measurement (Pressure Indication) Figure 4. Schematic of a hot cathode ion gauge. Bias voltages are typical values. A number of ion gauge configurations are in use with most being based upon the geometry of Bayard and Alpert where the collector is a fine wire that is located along the central axis of the sensor. The collector is surrounded by a helical grid and the filament is located outside the grid. This configuration was introduced to eliminate spurious signals that interfered with measurements below about 10-7 Torr. The common configurations include glass envelope tubes and "nude" gauges where the electrodes are mounted to a vacuum flange. An example of a nude gauge is shown in Figure 5. Ion gauges of the Bayard-Alpert configuration can measure pressures over the range of to 10-3 Torr. The upper limit is determined by the short mean free path which limits the free movement of the electrons and ions. The lower limit is determined by a variety of noise sources. Another type of ion gauge was developed by F.W. Penning in The Penning gauge consists of just two electrodes that are immersed in a magnetic field. With about 2,000 volts placed across the electrodes, a glow discharge is formed in the gap. The magnetic field causes the electrons to move in a spiral trajectory, increasing the probability of electron-ion collisions, even at very low pressures. In this de-

8 vice, the pressure is related to the current flowing between the electrodes. Because there is no filament, this type of gauge is called a cold cathode ion gauge. These are often used where the chamber gases might attack the filament of a hot cathode tube. Properly designed cold cathode gauges are also more immune to some of the noise sources that occur with hot cathode gauges, permitting operation to even lower pressures. Figure 5. Nude hot cathode ion sensor and controller. The sensor elements are mounted on a 2-3/4 inch CF flange. Photo courtesy of MKS Instruments, Inc. Trends in Indirect Vacuum Gauges While the traditional pairing of sensor/readout is still very popular there has been a trend toward integrated transducers where a scalable output is available directly from the gauge. In addition to analog voltage outputs, a variety of digital interfaces have been incorporated in these transducers. Miniaturization has been part of this transformation. Most of the sensor technologies have remained the same, just made more compact. An exception to this is the introduction of the micromachined Pirani sensor. The integrated/miniature nature of these transducers has further led to the combining of sensors within a single package. For example, a Pirani sensor can be combined with an ion gauge to provide continuous wide range coverage. Furthermore, the menu of indirect sensors may also be coupled with strain gauge (piezoresistive) diaphragm gauges to extend reading ranges to atmospheric pressure and above. Typical combinations include: Pirani and hot cathode or cold cathode ion gauge for wide range measurement (UHV to low vacuum) Pirani and absolute strain gauge for medium vacuum to positive pressure measurement Pirani and atmosphere referenced strain gauge for loadlock control

9 Figure 6. Integrated piezo differential and Pirani sensors in a loadlock gauge. Illustration courtesy of MKS Instruments, Inc. Accuracy in Indirect Gauges While simple and relatively inexpensive, the indirect gauges discussed above have some performance limitations that must be taken into account when they are used. A major factor is gas sensitivity. In the case of thermal conductivity gauges, different gases have varying thermal conductivities. Gases that are lighter than air, such as helium, are better thermal conductors because the molecules travel faster. Gases that are heavier than air, such as argon and xenon, are relatively poor thermal conductors. Since thermocouple and Pirani gauges work on the principle of thermal conductivity, their readings will change with gas type. The errors are particularly large at higher pressures. Some gauges, usually Piranis, will have calibration curves for various pure gases and corrections for a few common gases (helium, argon) may be incorporated in the electronics of the controller. When calibrating a thermal conductivity gauge (or any other kind of indirect gauge) the gauge must be calibrated with the gas that the gauge will be used with. With regard to ion gauges, as we have seen, the pressure indication produced by an ion gauge is dependent upon the quantity of ions that are produced and the ion yield is proportional to pressure. However, for a given electron energy, it is more difficult to ionize the smaller atoms which have tightly bound electrons than it is to ionize heavier atoms. Thus a given density of helium atoms will produce fewer ions than the same density of argon atoms. Another general issue with most of the commonly used indirect gauges is the fact that the gauges are not individually calibrated. Thus the calibration curve that is imbedded in the readout is a typical calibration curve for that make and model of gauge. Manufacturing tolerance, handling issues, etc. will cause gauge to gauge variations as well as variations over time. For the most part, manufacturers will not specify absolute accuracy values for these gauges. Repeatabilities may be specified and the better thermal and ionization gauges can achieve repeatabilities in the 5% range.

10 In the earlier days of vacuum, indirect gauges were usually adequate because the variety of gases used was fairly minimal and great precision was generally not required. In the few critical applications a McLeod manometer could be used. The situation changed with the advent of wide ranging microelectronics and thin film vacuum processes. Process pressure measurement and control became a major issue and a host of exotic gas mixtures came into use. The old style direct reading mercury manometers simply could not adapt to this new environment. To address this situation, the capacitance manometer (also called a capacitance diaphragm gauge) was developed. The Capacitance Manometer The requirements that are placed on a process control gauge are as follows: made of non-corrosive materials, providing accurate pressure readings independent of gas type, a wide range of pressure ranges, highly accurate and stable, high resolution, fast response and providing an electrical (analog or digital) output in order to integrate the gauge into pressure control systems. In a capacitance manometer, the pressure that the gas exerts is applied to a thin tensioned diaphragm. The thickness and diameter of the diaphragm are specified to permit the gauge to respond over a Diaphragm Electrode Structure (Metal on Ceramic) Chemical Getter Pump Electronics Baffle Measurement (Px) Side, Connected to System Electrical Connector Reference (Pr) Side, High Vacuum Oscillator Reference Constant Temperature Oven (Heated-Temperature Controlled Units Only) Bridge Buffer, Scaling, Amplification Output Figure 7. Schematic depiction of a capacitance manometer for absolute pressure measurement and a representative process manometer. Illustration and photo courtesy of MKS Instruments, Inc. particular range of pressures. It turns out the motions of the diaphragm are exceedingly small and therefore the bending of the diaphragm is sensed electrically by means of a capacitance bridge. Figure 7 shows the concept. The sensor consists of a sealed capsule that is divided by the diaphragm. Behind the diaphragm is a ceramic disk upon which there is a metal pattern in a form similar to that of a circular bull's eye surrounded by a ring. As a result of this geometry, between each electrode and the

11 diaphragm there is a capacitance that is defined by the areas of each electrode, the dielectric constant of the intervening gas (a constant since the capsule is sealed), and the separation of the each electrode from the diaphragm. Each of these variable capacitors is a leg in a bridge circuit. When the diaphragm is flat, the bridge is balanced and there is an output of 0 volts. However, as the diaphragm becomes curved as pressure is applied, the bridge becomes unbalanced and the degree of imbalance is reflected in the output voltage. The configuration shown in the figure represents an absolute gauge. The backside of the diaphragm is evacuated and maintained at a very low reference pressure. In order to maintain a high vacuum reference over the life of the instrument, a getter material is fired during the evacuation process. The getter is a chemical that scavenges any gases that may evolve. When the measurement port is open to any environment where the pressure is greater than that of the reference pressure, the diaphragm will curve and there will be a positive output voltage. The signal conditioning electronics then translates this to an output that is linear with pressure. Each capacitance manometer is specified according to its full scale (FS) range and it will be able to measure pressure for some number of decades below that full scale value. In the case of a 10 Torr FS manometer, the output at 10 Torr will be 10 volts. At 1 Torr it will be 1.0 volts, at 0.1 Torr it will be 0.1 volts and so forth. Needless to say, as the pressure goes to, say, a hundredth of a percent of the full scale rating, the output voltage will be exceedingly small and will hinder measurements. To measure lower pressures, one would use a lower full scale instrument. By varying the dimensions of the diaphragms, currently offered manometers have full scale ranges from 1000 Torr down to 20 millitorr. High pressure capacitance manometer have also been introduced that will measure pressures to 3000 psi. The materials that are used in the construction of capacitance manometers are crucial to accuracy and stability. Inconel R (Inconel R is a registered trademark of Inco Alloys International, 3200 Riverside Dr., Huntington, WV) diaphragms have been in continuous use since the introduction of commercial capacitance manometers in This material offers great chemical resistance and thermal-mechanical stability and is used in all of the high resolution and high accuracy manometers. Ceramic diaphragm instruments, first introduced about 20 years ago, have been in and out of production but have found some ongoing success in general purpose applications. For some applications the instrument may incorporate a temperature regulated oven to improve stability or to prevent the condensation of process gases on the diaphragm. Concerning accuracy, all capacitance manometers are individually calibrated and are traceable to national standards laboratories. They are available with guaranteed accuracy specifications of 1% of reading to 0.12% of reading. Premium capacitance manometers are available with 0.05% of reading accuracies.

12 The Capacitance Manometer Enters the 21 st Century Digital Capacitance Manometers The past decade has seen a steady increase in vacuum gauges with digital interfaces. This has been driven by the need to reduce the bulk of interconnect wiring, to improve noise immunity and to enhance diagnostics capabilities. The first generation of digital capacitance manometers consisted of an analog to digital module that was added to the standard analog circuitry. These instruments were calibrated as standard analog gauges by adjusting the span and linearity pots to match as much as possible the readings of the calibration standard. In the latest generation of digital instruments the digital section is integral to the manometer s functionality, not just an interface. Figure 8 shows the architecture along with a representative product. Note that both analog and digital connections may be made to the gauge. Many options are available for digital interfaces including DeviceNet, Profibus, RS-485 and ethernet protocols. Oscillator Reference Bridge Buffer D/A, Scaling, Amplification Digital Section Calibration Table Digital Interface Digital Communication Analog Output Figure 8. Digital capacitance manometer architecture and a representative instrument with analog and ethernet connectors. Photo courtesy of MKS Instruments, Inc. Note that the digital block contains a calibration table. As noted previously analog instruments are calibrated by adjusting 3 pots (zero, span, linearity) to match, as closely as possible, the readings of the calibration standard. With true digital instruments, the internal calibration table is populated with as many as 20 pressure points between zero and full scale. Because of the number of calibration points the linearity (and accuracy) of the instrument is improved. This improvement is generally in the range of 20%. This is depicted in Figure 9.

13 Analog Standard Digital Span Linearity Zero Figure 9. Analog vs digital capacitance manometer calibration. Contamination Control Capacitance manometers are directly exposed to the process conditions and are therefore subject to byproduct and/or particle build up on the diaphragm. As material accumulates on the diaphragm, the zero will shift and eventually it will be impossible to zero the gauge. The only remedy at this point is to have the sensor replaced. Using a heated manometer is a primary method to mitigate the effects of condensable gases. However, in some cases (especially some etch processes) these contaminants are still difficult to control. Figure 10 shows a variety of features that may be incorporated into process manometers to insulate the diaphragm from contaminants. The features shown include long/narrow passageways and particle sumps. The inlet screens and traps are usually replaceable elements whereas the internal baffles are integral to the sensor design. For the most part, these features will lengthen (often substantially) the life of a manometer but they may not render the instrument immune from contamination.

14 Figure 10. Features to shield the diaphragm from particulates and condensables. Summary Selecting a vacuum gauge for a particular application requires an understanding of how the gauge operates, what factors will cause erroneous readings, and what degree of accuracy and repeatability are required. The indirect gauges are generally economical and offer satisfactory performance for many monitoring applications. High precision capacitance manometers are used for critical control applications where known levels of accuracy and stability are required. Figure 11 shows a vacuum system with a variety of gauges in appropriate locations. This brief survey has only touched the surface of the more commonly encountered vacuum gauges. Please consult the references for articles that go into further detail or touch on other aspects of vacuum gauging.

15 Atmosphere Referenced Gauge for At-Atmosphere Indication Convection Pirani for Pumpdown Monitoring Capacitance Manometer for Process Pressure Control Loadlock Process Chamber Convection Pirani for Monitoring of Roughing Phase To Mechanical Pump Ion Gauge for Base Pressure Monitoring High Vacuum Pump Thermocouple Gauge for Foreline Monitoring To Mechanical Pump Figure 11. Vacuum system with a variety of gauges and the function of each. A single combo transducer could replace the two loadlock gauges. Acknowledgements Many aspects of this article are drawn from my years with MKS Instruments. I would also like to extend my thanks to Philip Sullivan of MKS for his input on the more recent innovations in capacitance manometers. Further Reading 1. Chris Stone, Process Improvement through Optimized Loadlock Operations, Vac Technology and Coating (December 2008), pp Bob Hyman, Effect of Aggressive Gases on Capacitance Manometers, Solid State Technology, October Baratron Etch Manometer Reduces Thermal Drift, Semiconductor Online, October Robert Traverso, U.S. Patent #7,252,011, Surface Area Deposition Trap, August 7, 2007.

### Pressure Measurement

Pressure Measurement Manometers Sensors, Transducers Ashish J. Modi Lecturer, Dept. of Mech.Engg., Shri S.V.M. inst. Of Technology, Bharuch Pressure Pressure is a force per unit area exerted by a fluid

### A Journal of Practical and Useful Vacuum Technology. By Phil Danielson

A Journal of Practical and Useful Vacuum Technology From By Phil Danielson Thermal Conductivity Gauges Thermal conductivity pressure gauges are extremely common in vacuum technology, but an understanding

### This educational seminar discusses creating, measuring, and troubleshooting Rough Vacuum.

This educational seminar discusses creating, measuring, and troubleshooting Rough Vacuum. Specifically, today s talk will cover: Brief review of Vacuum Fundamentals Applications Using Rough Vacuum Rough

### Chapter 5: Subatmospheric Total Pressure Gauges

Chapter 5: Subatmospheric Total Pressure Gauges The pressure gauges that will be described in this section are used to monitor the environment in a vacuum vessel so that processes conducted under subatmospheric

### Pressure Measurement. Introduction. Engr325 Instrumentation. Dr Curtis Nelson 3/1/17

3/1/17 Pressure Measurement Engr325 Instrumentation Dr Curtis Nelson Introduction A cluster of 72 helium-filled balloons over Temecula, California in April of 2003. The helium balloons displace approximately

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### Pressure & Vacuum Measurement. Series 370. Solutions. Stabil-Ion Vacuum gauge and Controller

w w w. m k s i n s t. c o m Pressure & Vacuum Measurement Solutions Series 370 Stabil-Ion Vacuum gauge and Controller Features & Benefits All-metal, rack-mount controller for Stabil-Ion and Convectron

### Gases and Pressure SECTION 11.1

SECTION 11.1 Gases and In the chapter States of Matter, you read about the kineticmolecular theory of matter. You were also introduced to how this theory explains some of the properties of ideal gases.

### Vacuum Science Techniques and Applications Dan Dessau Adv. Lab 2007

Vacuum Science Techniques and Applications Dan Dessau Adv. Lab 2007 Vacuum increases the mean-free-path of gas molecules. Vacuum prevents chemical reaction. Vacuum removes contaminants from surfaces. Vacuum

### Unit 9 Packet: Gas Laws Introduction to Gas Laws Notes:

Name: Unit 9 Packet: Gas Laws Introduction to Gas Laws Notes: Block: In chemistry, the relationships between gas physical properties are described as gas laws. Some of these properties are pressure, volume,

### LOW PRESSURE EFFUSION OF GASES revised by Igor Bolotin 03/05/12

LOW PRESSURE EFFUSION OF GASES revised by Igor Bolotin 03/05/ This experiment will introduce you to the kinetic properties of low-pressure gases. You will make observations on the rates with which selected

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### 3.0 Pressure Transmitter Selection

3.0 Pressure Transmitter Selection Each Tronic Line pressure transmitter has different features to meet specific performance, environmental, and price requirements. It is not possible to describe every

### SPD Pressure Sensor Families

DATASHEET SPD Pressure Sensor Families 1/7 Introduction to the principals of Smart Pressure Devices 1 Basic principles Pressure sensors are based on the principle of bending a membrane caused by the pressure

### Precision level sensing with low-pressure module MS

The task on hand Level sensing as it is understood in modern process automation is much more than simply "tank half full" or "tank a quarter full". Using suitable sensors, levels, inlets and outlets can

### LOW PRESSURE EFFUSION OF GASES adapted by Luke Hanley and Mike Trenary

ADH 1/7/014 LOW PRESSURE EFFUSION OF GASES adapted by Luke Hanley and Mike Trenary This experiment will introduce you to the kinetic properties of low-pressure gases. You will make observations on the

### Another convenient term is gauge pressure, which is a pressure measured above barometric pressure.

VACUUM Theory and Applications Vacuum may be defined as the complete emptiness of a given volume. It is impossible to obtain a perfect vacuum, but it is possible to obtain a level of vacuum, defined as

### Experiment. THE RELATIONSHIP BETWEEN VOLUME AND TEMPERATURE, i.e.,charles Law. By Dale A. Hammond, PhD, Brigham Young University Hawaii

Experiment THE RELATIONSHIP BETWEEN VOLUME AND TEMPERATURE, i.e.,charles Law By Dale A. Hammond, PhD, Brigham Young University Hawaii The objectives of this experiment are to... LEARNING OBJECTIVES introduce

### 8. Now plot on the following grid the values of T (K) and V from the table above, and connect the points.

Charles s Law According to Charles s law, the volume of a fixed mass of gas varies directly with its Kelvin temperature if its pressure is constant. The following table contains Celsius temperature and

### PSI Chemistry: Gases Multiple Choice Review

PSI Chemistry: Gases Multiple Choice Review Name Kinetic Molecular Theory 1. According to the kinetic-molecular theory, particles of matterare in constant motion (A) have different shapes (B) have different

### Dissolved Oxygen Guide

Educat i onser i es Di ssol vedoxygengui de Dissolved Oxygen Guide Introduction Dissolved oxygen probes provide a convenient approach to essentially direct measurement of molecular oxygen. The membrane

### OMNISTAR / THERMOSTAR. The efficient solution for gas analysis. Intelligent software. Wide range of applications.

OMNISTAR / THERMOSTAR The efficient solution for gas analysis. Intelligent software. Wide range of applications. OMNISTAR /THERMOSTAR The efficient solution for gas analysis. Intelligent software. Wide

### Chapter 13 Gases and Pressure. Pressure and Force. Pressure is the force per unit area on a surface. Force Area. Pressure =

Chapter 13 Gas Laws Chapter 13 Gases and Pressure Pressure and Force Pressure is the force per unit area on a surface. Pressure = Force Area Chapter 13 Gases and Pressure Gases in the Atmosphere The atmosphere

### Chemistry HP Unit 6 Gases. Learning Targets (Your exam at the end of Unit 6 will assess the following:) 6. Gases

Chemistry HP Unit 6 Gases Learning Targets (Your exam at the end of Unit 6 will assess the following:) 6. Gases 6-1. Define pressure using a mathematical equation. 6-2. Perform calculations involving pressure,

### AMS 2710 PCB pressure sensor module with V output

FEATURES Universal pressure sensor module with 0.. 10 V voltage output Fully calibrated and temperature compensated sensor module Variants for (bidirectional) differential, gage, absolute and barometric

### Cryo-Evaporator Operation

Cryo-Evaporator Operation Cara Ricci November 25, 2003 Thin Film Deposition THE UNIVERSITY OF TEXAS AT DALLAS ERIK JOHNSON SCHOOL OF ENGINEERING DOCUMENT NUMBER: FA2003-TF-008 EDITION: 1.1 PAGE: 1 of 28

### Unit 2 Pressure Measurement Techniques 1. State importance of pressure measurement in process industries.

Unit 2 Pressure Measurement Techniques 1. State importance of pressure measurement in process industries. The measurement of pressure is one of the most important measurements, as it is used in almost

### Chapter 3 PRESSURE AND FLUID STATICS

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Chapter 3 PRESSURE AND FLUID STATICS Lecture slides by Hasan Hacışevki Copyright The McGraw-Hill

### Truck-mounted Mass Flow Metering for LPG Delivery

Truck-mounted Mass Flow Metering for LPG Delivery Itron, Inc. 1310 Emerald Rd. Greenwood, SC 29646 Introduction The mathematician Gustave De Coriolis first described the Coriolis Effect, which is an inertial

### 800 Series Mass Flow Meters and Controllers. High Performance Mass Flow Meters and Controllers

800 Series Mass Flow Meters and Controllers High Performance Mass Flow Meters and Controllers WHY MASS FLOW? In most processes it is mass, not volume, which is the critical variable. Volumetric flow measurements

### Atmospheric Pressure. Conceptual Physics 11 th Edition. Atmospheric Pressure. Atmospheric Pressure. The Atmosphere

Atmospheric Pressure Conceptual Physics 11 th Edition Chapter 14: GASES Atmospheric pressure Caused by weight of air Varies from one locality to another Not uniform Measurements are used to predict weather

### CP Chapter 13/14 Notes The Property of Gases Kinetic Molecular Theory

CP Chapter 13/14 Notes The Property of Gases Kinetic Molecular Theory Kinetic Molecular Theory of Gases The word kinetic refers to. Kinetic energy is the an object has because of its motion. Kinetic Molecular

### The Ins and Outs of I/P Transducers

The Ins and Outs of I/P Transducers By Mark B. Levine, ControlAir Inc. General description I/P transducers are versatile instruments that use an electrical control signal to proportionally regulate gas

### How To Use Getters and Getter Pumps

A Journal of Practical and Useful Vacuum Technology From By Phil Danielson How To Use Getters and Getter Pumps Gettering is a vacuum pumping technology that has been with us, in many forms, for almost

### Pressure of the atmosphere varies with elevation and weather conditions. Barometer- device used to measure atmospheric pressure.

Chapter 12 Section 1 Pressure A gas exerts pressure on its surroundings. Blow up a balloon. The gas we are most familiar with is the atmosphere, a mixture of mostly elemental nitrogen and oxygen. Pressure

### Fluid Mechanics. Liquids and gases have the ability to flow They are called fluids There are a variety of LAWS that fluids obey

Fluid Mechanics Fluid Mechanics Liquids and gases have the ability to flow They are called fluids There are a variety of LAWS that fluids obey Density Regardless of form (solid, liquid, gas) we can define

### Chapter 11: Gases: Homework: Read Chapter 11. Keep up with MasteringChemistry and workshops

C h e m i s t r y 1 2 C h a p t e r 11 G a s e s P a g e 1 Chapter 11: Gases: Homework: Read Chapter 11. Keep up with MasteringChemistry and workshops Gas Properties: Gases have high kinetic energy low

### Precision Pressure Regulators. Precision Fluidics

Precision Pressure Regulators Precision Fluidics This is increased efficiency. Precision low flow carrier gas control for gas chromatography. Parker precision regulators, valves and controllers are designed

### Chapter 13 Fluids. Copyright 2009 Pearson Education, Inc.

Chapter 13 Fluids Phases of Matter Density and Specific Gravity Pressure in Fluids Atmospheric Pressure and Gauge Pressure Pascal s Principle Units of Chapter 13 Measurement of Pressure; Gauges and the

### Matter is made up of particles which are in continual random motion Misconception: Only when a substance is in its liquid or gas state do its

Kinetic Theory of Matter Matter is made up of particles which are in continual random motion Misconception: Only when a substance is in its liquid or gas state do its particles move because in these two

### WATER MADE EASY MARINE ENERGY MUNICIPAL INDUSTRIAL

Capital Controls Series 71V3000 The Series 71V3000 electrically-heated vaporizer automatically vaporizes and superheats liquid chlorine, sulfur dioxide or ammonia at a rate controlled by the gas feed system.

### Predicted Dispense Volume vs. Gravimetric Measurement for the MICROLAB 600. November 2010

Predicted Dispense Volume vs. Gravimetric Measurement for the MICROLAB 600 November 2010 Table of Contents ``Abstract...3 ``Introduction...4 ``Methods & Results...6 ``Data Analysis...9 ``Conclusion...12

### CHAPTER 3 RESULTS AND DISCUSSION

CHAPTER 3 RESULTS AND DISCUSSION The research included the development and verification of the earlier apparatus, and the collection and validation of moisture diffusion data under both isothermal and

### Chapter 5: Gases 5.1 Pressure Why study gases? An understanding of real world phenomena. An understanding of how science works.

Chapter 5: Gases 5.1 Pressure Why study gases? An understanding of real world phenomena. An understanding of how science works. A Gas Uniformly fills any container. Easily compressed. Mixes completely

### Application Note AN-107

SPEC Sensor TM Characterization & Calibration Considerations Scope This document is provided to describe the considerations needed to characterize, calibrate, verify and validate the measurement performance

### EASIDEW TRANSMITTER with Current Source Output

EASIDEW TRANSMITTER with Current Source Output INSTALLATION, OPERATION AND MAINTENANCE MANUAL Issue March 2002 2 TABLE OF CONTENTS SECTION PAGE 1. INTRODUCTION 3 1.1 General 3 1.2 Ceramic Sensing Element

### LEAP CO 2 Laboratory CO 2 mixtures test facility

LEAP CO 2 Laboratory CO 2 mixtures test facility THE PROJECT AIM CO 2 flows made available by several capture techniques are contaminated with impurities and this affects the design and operations of the

### Revision 2013 Vacuum Technology 1-3 day Good Vacuum Practice 1 Day Course Outline

Revision 2013 Vacuum Technology 1-3 day Good Vacuum Practice 1 Day Course Outline This training course outline is intended to cover the following: Introduction to vacuum Measurement Lubricated rotary pumps

### VS Series Leak Detectors for Vacuum Furnaces LEAK DETECTION FOR VACUUM FURNACES

VS Series Leak Detectors for Vacuum Furnaces LEAK DETECTION FOR VACUUM FURNACES VS Series Leak Detectors for Vacuum Furnaces indicate the presence of a leak. If the vacuum level is slow to reach the original

### Chapter 13: The Behavior of Gases

Chapter 13: The Behavior of Gases I. First Concepts a. The 3 states of matter most important to us: solids, liquids, and gases. b. Real Gases and Ideal Gases i. Real gases exist, ideal gases do not ii.

Chapter 13 Gases Copyright Cengage Learning. All rights reserved 1 Section 13.1 Pressure Why study gases? An understanding of real world phenomena. An understanding of how science works. Copyright Cengage

### Memorandum Background: Results and Discussion:

Memorandum To: Dr. Abercromby From: Jason Rapp and Matthew Stumbo Date: January 17, 2012 Subject: Vacuum Chamber Basics Technical Memo Background: We explored several aspects of vacuum chambers, in preparation

### The HumiSys. RH Generator. Operation. Applications. Designed, built, and supported by InstruQuest Inc.

The HumiSys RH Generator Designed, built, and supported by InstruQuest Inc. Versatile Relative Humidity Generation and Multi-Sensor System The new HumiSys with single or dual RH probes capabilities is

### CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES

SENSORS FOR MASS FLOW AND PRESSURE CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES INTRODUCTION Bronkhorst High-Tech BV is the worldwide pioneer in the field of low flow liquid metering

PLASMA PHYSICS ADVANCED LAB Rev. 2010 Laboratory Cautions 1. The energy from the RF amplifier can can cause burns or even be lethal. Be aware of all electrical connections to the output of this device.

### Ideal gas law. Introduction

Ideal gas law Introduction We think of a gas as a collection of tiny particles in random, thermal motion. When they collide with the sides of a container, they exert a force on the container walls. The

### The Discussion of this exercise covers the following points: Range with an elevated or suppressed zero Suppressed-zero range Elevated-zero range

Exercise 4-3 Zero Suppression and Zero Elevation EXERCISE OBJECTIVE In this exercise, you will learn the effect that mounting a pressure transmitter above or below the reference level has on the hydrostatic

### P900. P900, Datenblatt Seite 1 DESCRIPTION

, Datenblatt Seite 1 Field proven rugged construction High overpressure capability High reliability for demanding environments Application specific customization Excellent media compatibility Shock and

### Outcomes: Example the historical development of the measurement of pressure. Include: Galileo, Toricelli, von Gureick, Pascal, Huygens, Avogadro,

History of Pressure Outcomes: Example the historical development of the measurement of pressure. Include: Galileo, Toricelli, von Gureick, Pascal, Huygens, Avogadro, Dalton. Describe the various units

### Point level switches for safety systems

Point level switches for safety systems By: Bill Sholette Level Products Business Manager Northeast US - Endress+Hauser Point level switches are often used in applications designed to prevent accidents.

### Gas Laws. Introduction

Gas Laws Introduction In 1662 Robert Boyle found that, at constant temperature, the pressure of a gas and its volume are inversely proportional such that P x V = constant. This relationship is known as

### Botlek Studiegroep Badotherm. Solving measurements issues with Diaphragm seals 3 Case Studies

Botlek Studiegroep 9-4-2015 Badotherm Solving measurements issues with Diaphragm seals 3 Case Studies Prepared: 10/04/2015 Badotherm Company Summary Badotherm introduction Family owned company Founded

### Hitachi Differential Pressure/Pressure Transmitter

Hitachi Differential Pressure/Pressure Transmitter Hitachi Differential Pressure/Pressure Transmitter Series is based on our accumulated technology and is available in many types. This instrument is expected

### The HumiPyc - Model 1 - Gas Pycnometer; Density, Moisture, Permeation Analyzer; RH sensor Calibrator

The HumiPyc - Model 1 - Gas Pycnometer; Density, Moisture, Permeation Analyzer; RH sensor Calibrator Designed, built, and supported by InstruQuest Inc. Temperature controlled, multi-technique volumetric

### L 100. Bubble-Tube Level System. Installation, Operation and Maintenance Instructions

L 100 Bubble-Tube Level System Installation, Operation and Maintenance Instructions Figure 1 Contents Section Description Page 1.0 Introduction 2 2.0 Specifications 3 3.0 Installation 3 4.0 Warranty 6

### Figure 1 Schematic of opposing air bearing concept

Theoretical Analysis of Opposing Air Bearing Concept This concept utilizes air bearings to constrain five degrees of freedom of the optic as shown in the figure below. Three pairs of inherently compensated

### Over 20,000 Strain Gage Target flowmeters installed since 1952.

Over 20,000 Strain Gage Target flowmeters installed since 1952. Liquid, gases, superheated and saturated steam from -320 to 500 O F and up to 15,000 psi. No frictional moving parts. Designed to withstand

### High Precision Gauge Pressure Indicators

Series 1500 8½" Dial High Precision Gauge Pressure Indicators Series 1500 Gauges have capsule-type pressure elements up to and including the 150-psig range. Ranges of 200 psig and above have Bourdon tubes.

### Long term stability tests of INO RPC prototypes

Long term stability tests of INO RPC prototypes While the problem of sudden aging in the glass RPC prototypes is being investigated, a few RPCs of dimension 40 cm 30 cm were fabricated, using glass procured

### VACUUM TESTING PRECAST CONCRETE MANHOLES

1 OF 5 testing is a quick, safe and practical way to validate manhole system integrity. Manhole sections can be tested at the precast concrete plant prior to delivery or on site prior to backfilling. Here

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Bypass MFC with capillary technology for nominal flow rates from 5 ml N /min to 15 l N /min Applicable for aggressive gases Compact design and digital communication

### Exercise 4-2. Centrifugal Pumps EXERCISE OBJECTIVE DISCUSSION OUTLINE DISCUSSION. Pumps

Exercise 4-2 Centrifugal Pumps EXERCISE OBJECTIVE Familiarize yourself with the basics of liquid pumps, specifically with the basics of centrifugal pumps. DISCUSSION OUTLINE The Discussion of this exercise

### . In an elevator accelerating upward (A) both the elevator accelerating upward (B) the first is equations are valid

IIT JEE Achiever 2014 Ist Year Physics-2: Worksheet-1 Date: 2014-06-26 Hydrostatics 1. A liquid can easily change its shape but a solid cannot because (A) the density of a liquid is smaller than that of

### Students measure the change in pressure by varying the volume of trapped air in a syringe while:

How Does a Trapped Gas Behave? Teacher Information Objective Students investigate the effect of changes in the volume of a confined gas on pressure at constant temperature. Using the pressure sensor, students

### JANIS OPERATING INSTRUCTIONS FOR SUPERCONDUCTING MAGNET CRYOSTATS

OPERATING INSTRUCTIONS FOR SUPERCONDUCTING MAGNET CRYOSTATS INTRODUCTION The Janis Research Company's Superconducting Magnet/Cryostat System is one of the most versatile tools available to the scientist

### Section 5.1 Pressure. Why study gases? An understanding of real world phenomena. An understanding of how science works.

Chapter 5 Gases Section 5.1 Pressure Why study gases? An understanding of real world phenomena. An understanding of how science works. Copyright Cengage Learning. All rights reserved 2 Section 5.1 Pressure

### Lab 3 Introduction to Quantitative Analysis: Pumps and Measurements of Flow

Georgia Institute of Technology School of Earth and Atmospheric Sciences EAS 4641, Spring 2008 Lab 3 Introduction to Quantitative Analysis: Pumps and Measurements of Flow Purpose of Lab 3: 1) To gain a

### PHYS 101 Previous Exam Problems

PHYS 101 Previous Exam Problems CHAPTER 14 Fluids Fluids at rest pressure vs. depth Pascal s principle Archimedes s principle Buoynat forces Fluids in motion: Continuity & Bernoulli equations 1. How deep

### The Discussion of this exercise covers the following points:

Exercise 5-3 Wet Reference Leg EXERCISE OBJECTIVE Learn to measure the level in a vessel using a wet reference leg. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Measuring

### Additel 761 Automated Pressure Calibrators

Automated Calibrators UPDATED Fully automated pressure calibrator with built-in pressure generator / controller to as high as 600 psi (40 bar) or as low as 0.01 Pa (0.00004 inh 2 O) accuracy Dual pressure

### To convert to millimeters of mercury, we derive a unit factor related to the equivalent relationship 29.9 in. Hg = 760 mm Hg.

Example Exercise 11.1 Gas Pressure Conversion Meteorologists state that a falling barometer indicates an approaching storm. Given a barometric pressure of 27.5 in. Hg, express the pressure in each of the

### Self Heating Sensing Applications

Self Heating Sensing Applications The main sections of this white paper are laid out as follows: Section 1 Overview Section 2 Section 3 Liquid Level Sensing Fluid Flow Sensing. Section 4 Gas Analysis using

### EXPERIMENT 8 Ideal Gas Law: Molecular Weight of a Vapor

EXPERIMENT 8 Ideal Gas Law: Molecular Weight of a Vapor Purpose: In this experiment you will use the ideal gas law to calculate the molecular weight of a volatile liquid compound by measuring the mass,

### Phys101 Lectures Fluids I. Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle. Ref: 10-1,2,3,4,5,6,7.

Phys101 Lectures 24-25 luids I Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle Ref: 10-1,2,3,4,5,6,7. Page 1 10-1 Phases of Matter The three common phases of matter are solid,

### HumiSys HF High Flow RH Generator

HumiSys HF High Flow RH Generator Designed, built, and supported by InstruQuest Inc. Versatile Relative Humidity Generation and Multi-Sensor System The HumiSys HF is a high flow version of the previously

### Basic Concepts of Chemistry Notes for Students [Chapter 10, page 1] D J Weinkauff - Nerinx Hall High School. Chapter 10 Gases

Basic Concepts of Chemistry Notes for Students [Chapter 10, page 1] Chapter 10 Gases We have talked a little about gases in Chapter 3 and we dealt briefly with them in our stoichiometric calculations in

### Micro Channel Recuperator for a Reverse Brayton Cycle Cryocooler

Micro Channel Recuperator for a Reverse Brayton Cycle Cryocooler C. Becnel, J. Lagrone, and K. Kelly Mezzo Technologies Baton Rouge, LA USA 70806 ABSTRACT The Missile Defense Agency has supported a research

### - Introduction: a) Metallized Paper & Film :The market Outlook The following diagram shows the estimated usage of the main metallized materials

Vacuum Metallization of Paper and Outgassing Materials By Fabiano Rimediotti Galileo Vacuum Systems Prato Italy Phone : +390574564380 - f.rimediotti@galileovacuum.com Abstract Materials like paper, some

### InstruTech, Inc. Vacuum Gauge Controller B-RAX User Manual. InstruTech, Inc S. Fordham St. Longmont, CO USA

InstruTech, Inc. Vacuum Gauge Controller B-RAX 3100 User Manual InstruTech, Inc. 1475 S. Fordham St. Longmont, CO 80503 USA Phone: +1-303-651-0551 Fax: +1-303-678-1754 E-mail info@instrutechinc.com www.instrutechinc.com

### Gun Vacuum Performance Measurements February 2003

JLAB-TN-03-013 Gun Vacuum Performance Measurements February 2003 Marcy Stutzman Abstract. The vacuum in the polarized photoguns at Jefferson Lab is crucial to long lifetime operation of the polarized source.

### THERMAL MASS FLOW MEASUREMENT FOR GASES

THERMAL MASS MEASUREMENT FOR GASES Make the Wise Choice. Choose Sage Flow Meters. A Commitment to Higher Performance SAGE METERING is a manufacturer of high performance Thermal Mass Flow Meters which measure

### Leak Checking Large Vacuum Chambers

Leak Checking Large Vacuum Chambers Technical Overview Vacuum Technologies Introduction Understanding the pump-down characteristics of a large vacuum vessel is critical for determining whether the vacuum

### Chapter 13 Gases, Vapors, Liquids, and Solids

Chapter 13 Gases, Vapors, Liquids, and Solids Property is meaning any measurable characteristic of a substance, such as pressure, volume, or temperature, or a characteristic that can be calculated or deduced,

### Practical approach and problems in in-situ RGA calibration

Practical approach and problems in in-situ RGA calibration Oleg Malyshev and Keith Middleman Vacuum Science Group, ASTeC Accelerator Science and Technology Centre STFC Daresbury Laboratory UK Workshop

### Major Design Features

Major Design Features 3Flex Features Design Benefits Discussion 316 SS fittings and pneumatically actuated, hard-seal valves Provides virtually leak-free gas management with the lowest outgassing rate

### Ch. 11 Mass transfer principles

Transport of chemical species in solid, liquid, or gas mixture Transport driven by composition gradient, similar to temperature gradients driving heat transport We will look at two mass transport mechanisms,

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### Pressure Measurement Single-range transmitters for general applications

Siemens A 207 Overview Application The SITRANS P Compact pressure transmitter is designed for the special requirements of the food, pharmaceutical and biotechnology industries. The use of high-grade materials