# Sampling Considerations for Equilibrium Dissolved Oxygen [DO] Sensors

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Sampling Considerations for Equilibrium Dissolved Oxygen [DO] Sensors Abstract Boiler water in the low parts per billion dissolved oxygen concentration range can be significantly contaminated by oxygen intrusion from leaks and trapped air. Careful flow design is required to minimize errors resulting from external sources of oxygen. In many cases the flow dependence of the dissolved oxygen reading reveals the presence of contamination. The expected response for equilibrium sensors differs from that for diffusion rate sensors. Introduction At standard temperature [25 C] and pressure [1 atmosphere], clean water in equilibrium with air contains 8.24 milligrams of oxygen per liter of water. This is called the air saturation value at 25 C and is usually reported as parts per million, ppm. The control of corrosion in modern high pressure boilers commonly demands that the dissolved oxygen concentration in the water be less than 4 parts per billion, ppb, or 1/2000 th of the air saturation value. The 8.24 micrograms of oxygen contained in one milliliter [ml] of air saturated water is enough to raise the dissolved oxygen concentration in 2000 ml of water by 4 ppb. Thus, if 2000ml of water containing 4 ppb dissolved oxygen were exposed to just one ml of air saturated water, the dissolved oxygen concentration in the 2000 ml would be doubled. The concentration of oxygen in air is 20.9% by volume. The milligrams of oxygen per ml is calculated as follows: From the ideal gas law, PV = nrt where P is pressure, V is volume, n is the number of moles [gram molecular weights], R is the gas constant and T is the absolute temperature, the volume of one mole of gas at atmospheric pressure and 25 C [298 K] is V = 1 mole x [ Liter-atm/mole K] x 298K/1 atm = L The mg of oxygen per L is given by 1 L x [1 mole gas/24.45 Lgas] x [0.209 mole O 2/ mole gas] x [32,000mg O 2/mole O 2] = 274 mgo 2 per liter of air. This is about 33 times as concentrated as oxygen in air saturated water. The amount of oxygen in 1 ml of air is that which is required to double the concentration of [33 x 2000] 66,000 ml of water originally containing 4 ppb of dissolved oxygen. Leaks and Dead Spots Leaks can be either true leaks minute holes in the plumbing carrying the water usually observed at fittings or diffusion of oxygen from the surrounding air through the walls of non-metal tubing. Non-metal tubing is not recommended but is frequently used because of its flexibility. Any place where water fails to fill the plumbing or where the water flows down can become a dead spot and a source for oxygen contamination. An example of each is shown in Figure 1. Water flows up a sample line, over the oxygen-sensing portion of an equilibrium sensor, through the body of the sensor, out a side-arm outlet, down and then up a length of tubing leading to a funnel and, finally, over the top of the funnel. The funnel provides access to the water for the ampule of a wet chemical analysis of dissolved oxygen. The valve meters the sample flow and stops it for sensor calibration of servicing. Two things are wrong with this picture. First, as shown in the cutaway, the side arm outlet from the equilibrium probe is situated only part way up a large internal cavity. The water fills to the level of the side arm but, at least initially, an air pocket significantly larger that the 1 milliliter discussed in the introduction exists in the upper portion of the cavity.

2 Sampling Considerations for Equilibrium Dissolved Oxygen Sensors 2 While the air in this pocket must eventually dissolve into the water flowing through the sensor, until it does, it serves as a source for oxygen contamination that renders the results gathered by the wet chemical method unreliable. Consequently, independent measurement of dissolved oxygen in water flowing out of an equilibrium probe is not recommended. The second error is allowing the water to the sampling funnel to flow down. Before flow starts, the tube will be filled with air. It is necessary to sweep the air from the tube before reliable results can be expected. Upflowing water will tend to lift air out above it. Downflowing water, however, can leave air bubbles trapped in the downflowing portion of the tubing. As long as the buoyant upward velocity of the bubbles exceeds the sample flow, the bubbles will remain in the downflowing portion of the tubing and will serve as a source of oxygen contamination until they are dissolved. The valve on the common line stops flow for calibration and service. Again, two problems present themselves. When the flow to the sensor is stopped, the stagnant water in the sensor and the line to it will initially be at the sample dissolved oxygen concentration. However, unless the sensor has had water flowing through it long enough to dissolve the internal air pocket, oxygen from the pocket will immediately begin to diffuse into the water in the cavity and down the sample line. The question then will be whether this diffusion process is slow enough that the oxygen reaching the line flowing to the funnel will be negligible. The other problem is a general one concerning allowing any stagnant water to communicate with the sample flow. In the normal operation, flow is to the sensor and goes past the piping to the funnel. The water in this stagnant leg is sealed from contamination by valve below the funnel. Less obvious are the problems associated with the sampling system shown in Figure 2. Water normally flows through the equilibrium sensor and out its sidearm outlet as in Figure 1. When a wet chemical calibration check is to be performed, the valve in the sidearm outlet is closed, the valve below the funnel is opened diverting flow from the sensor to the funnel. The water in this stagnant leg is sealed from contamination by valve below the funnel. However, should this valve or any other fitting in this leg leak, oxygen can enter this stagnant leg and eventually reach the T below the sensor and begin contaminating the sample. This leak might be small enough to be negligible when its oxygen is continuously being diluted by the high flow recommended for the wet chemical analysis.

3 Sampling Considerations for Equilibrium Dissolved Oxygen Sensors 3 However, a leak of similar magnitude might be large enough to contaminate the stagnant water and the water flowing to the sensor since this flow is usually much less than that recommended for the wet chemical method. An arrangement eliminating sample communication with stagnant water is shown in Figure 3. Flow may be directed to either the sensor or the funnel and the 3-way valve isolates the sample flow from the stagnant water in both cases. Two problems with the arrangement shown in Figure 3 remain. First, the sensor and the funnel cannot be supplied with sample simultaneously. It is desirable to be monitoring the dissolved oxygen concentration with the sensor exactly when the sample for the wet chemical analysis is gathered. Second, dissolved oxygen sensors are normally calibrated in air. It is desirable to be able to expose the sensor to air without opening plumbing connections. To achieve uncontaminated, simultaneous sampling and valvecontrolled air calibration, the plumbing arrangement in Figure 4 is recommended. Normal operation consists of allowing flow to be metered through the 3-way valve to the sensor. The stagnant water in the line to the funnel is protected from air contamination by a second shutoff metering valve. Simultaneous monitoring eliminates disagreements resulting from a fluctuating dissolved oxygen concentration. Simultaneous monitoring also allows a rapid return to the normal operation where the sensor only is monitoring without having to wait while air-contaminated water is purged from the line to the sample.

4 Sampling Considerations for Equilibrium Dissolved Oxygen Sensors 4 The sensor is exposed to air by switching the 3-way valve to drain. If this valve is solenoid-actuated, air calibration can be executed automatically by activating a relay in a sufficiently sophisticated analyzer. Furthermore, initiation of the calibration can be based on a programmed time or by an event. Simultaneous flow to the funnel and to the sensor is controlled by the two metering valves. Shown in Figure 5 is another arrangement that meets all of the requirements of uncontaminated simultaneous sampling and valve actuated calibration. The only drawback is that two connections into the main sample line flow are required. Consequences Detection of Oxygen Contamination In spite of all the precautions described above, dead spaces can exist in complicated sampling arrangements and leaks can always occur. Their presence can be revealed by the response of the sensor. The sensor s dependence on sample flow rate is a key diagnostic. Equilibrium dissolved oxygen sensors are independent of flow rate. Within the sensor, oxygen is consumed at an electrode and its concentration determined from the direct current resulting from this process. The amount of oxygen consumed is balanced by oxygen produced electrochemically at a counter electrode. No net oxygen is consumed; there is no oxygen depletion in the sample; the sample just outside of the sensor need not be replenished so flow dependence is zero and oxygen may be measured in still water if desired, see Figure 6. This is to be contrasted with oxygen sensors which consume oxygen electrochemically but which do not internally balance this consumption by a replenishing reaction. For these sensors, oxygen must continually diffuse from the sample, through a membrane, through some portion of an internal electrolyte solution to an electrode where it is consumed. Net oxygen is consumed from the sample. A concentration gradient develops from a point some distance from the sensor, through regions of diminishing dissolved oxygen concentration to the consuming electrode where the dissolved oxygen concentration is zero. The current from which the dissolved oxygen concentration is inferred depends on the diffusion path. If the flow rate past the sensor is diminished, replenishment of the consumed oxygen is lessened and the resulting drop in current is unrelated to the bulk dissolved oxygen concentration. Diffusion rate sensors demand a minimum flow rate below which the reported dissolved oxygen concentration will decrease with decreases in flow rate, see Figure 7. How does all of this allow inferring the presence of a leak? From this discussion of how these sensors work, we see that a decrease in flow rate should result in either [1] no change in signal for an equilibrium sensor or [2] a decrease in signal for a diffusion rate sensor. If, however, a decrease in flow leads to an increase in signal, this is almost surely related to oxygen leaking into the sample. For a given leak rate, a decrease in flow would correspond to an increase in dissolved oxygen concentration because lowering the flow rate dilutes the leaking oxygen less.

5 Sampling Considerations for Equilibrium Dissolved Oxygen Sensors 5 Distinguishing between a leak and trapped air involves comparing results over time. With a [constant] leak, the relationship between flow rate and the error in reported dissolved oxygen should be constant. With trapped air, the contamination and the positive flow rate dependence should decrease with time. Often this effect is reported as a sensor with slow response at low ppb concentrations. What s really happening is the slow dissolution of trapped oxygen into the flowing sample. Once the trapped oxygen source is depleted, the sensor responds correctly. Conclusion Sampling systems for dissolved oxygen sensors should be devoid of leaks to the atmosphere, regions where air or stagnant water can communicate with the sample and downflowing. Leaks and traps can be detected from the flow dependence and time-dependent flow dependence of the reported dissolved oxygen concentration. Each time a trap is exposed to air, such as during air calibration, the trapped oxygen source is reestablished and the trap symptoms described above reappear.

6 Sampling Considerations for Equilibrium Dissolved Oxygen Sensors 6 More Information For more information on Equilibrium Dissolved Oxygen Sensors, visit or contact your Honeywell account manager. Honeywell Process Solutions Honeywell 1250 West Sam Houston Parkway South Houston, TX Honeywell House, Arlington Business Park Bracknell, Berkshire, England RG12 1EB UK Shanghai City Centre, 100 Junyi Road Shanghai, China SO ENG January Honeywell International Inc.

### Dissolved Oxygen Guide

Educat i onser i es Di ssol vedoxygengui de Dissolved Oxygen Guide Introduction Dissolved oxygen probes provide a convenient approach to essentially direct measurement of molecular oxygen. The membrane

### ASTM 3612/TOGA/Dissolved Gas GC Revised October 2013

The SRI ASTM 3612/TOGA/ Dissolved Gas GC configuration ( TOGA GC ) permits measurement of gasses dissolved in water, oil or other liquids, and is based on the requirements of ASTM method 3612, option C,

### Evaluation copy. Interdependence of Plants and Animals. computer OBJECTIVES MATERIALS

Interdependence of Plants and Animals Computer 14 Plants and animals share many of the same chemicals throughout their lives. In most ecosystems, O 2, CO 2, water, food and nutrients are exchanged between

### CTB3365x Introduction to Water Treatment

CTB3365x Introduction to Water Treatment D4b Aeration Doris van Halem Did you know that there are not just gasses in your glass of sparkling coke, but also in the tap water you drink? Welcome to the water

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### Oxygen Measurement in Natural Gas

Oxygen Measurement in Natural Gas Application Background Natural Gas Fuel is primarily methane. If it were pure methane, it would be very simple to transport and use. But it comes out of the well as a

### Dextens Thermo Conductivity Sensors. Type 52101, 54101, 55101

Dextens Thermo Conductivity Sensors Type 52101, 54101, 55101 Operating Manual May 2012 Table of Content 1 GENERAL INFORMATION...4 1.1 DISCLAIMER...4 1.2 SYMBOLS AND CONVENTIONS...4 1.3 SERVICES AND REPAIRS...4

### The Decomposition of Potassium Chlorate

The Decomposition of Potassium Chlorate Small quantities of molecular oxygen (O 2 ) can be obtained from the thermal decomposition of certain oxides, peroxides, and salts of oxoacids. Some examples of

### Behavior of Gases Chapter 12 Assignment & Problem Set

Behavior of Gases Name Warm-Ups (Show your work for credit) Date 1. Date 2. Date 3. Date 4. Date 5. Date 6. Date 7. Date 8. Behavior of Gases 2 Study Guide: Things You Must Know Vocabulary (know the definition

### Instruction Bulletin Addendum For Pressure-Balanced In Situ Oxygen Probes

Instruction Bulletin Addendum For Pressure-Balanced In Situ Oxygen Probes Instruction Bulletin Addendum IB PB1000 Part no. Serial no. Order no. Oxymitter 4000 Applicability: Used with pressure-balancing

### AMT-Ex Dewpoint Transmitter

AMT-Ex Dewpoint Transmitter Instruction Manual Alpha Moisture Systems Alpha House 96 City Road Bradford BD8 8ES England Tel: +44 1274 733100 Fax: +44 1274 733200 email: mail@amsytems.co.uk web: www.amsystems.co.uk

### LOW PRESSURE EFFUSION OF GASES revised by Igor Bolotin 03/05/12

LOW PRESSURE EFFUSION OF GASES revised by Igor Bolotin 03/05/ This experiment will introduce you to the kinetic properties of low-pressure gases. You will make observations on the rates with which selected

### Calibration Gas Instrument INSTRUCTION MANUAL. Release I. Advanced Calibration Designs, Inc.

Advanced Calibration Designs, Inc. Calibration Gas Instrument INSTRUCTION MANUAL Release I www.goacd.com Instruction Manual Gas Generator Release I TABLE OF CONTENTS I. General Description Page 2 II. Start-Up

### MULTIPLE CHOICE. Choose the one alternative that best completes the statement or answers the question.

Test General Chemistry CH116 UMass Boston Summer 2013 Name MULTIPLE CHOICE. Choose the one alternative that best completes the statement or answers the question. 1) The pressure exerted by a column of

### L 100. Bubble-Tube Level System. Installation, Operation and Maintenance Instructions

L 100 Bubble-Tube Level System Installation, Operation and Maintenance Instructions Figure 1 Contents Section Description Page 1.0 Introduction 2 2.0 Specifications 3 3.0 Installation 3 4.0 Warranty 6

### Potter Nitrogen Generators

Potter Nitrogen Generators What is nitrogen? Nitrogen the inert gas Earth s atmosphere the air we breathe is composed of nitrogen, oxygen, and water vapor. While this atmosphere is great for life, it is

### Topics For Discussion Acadiana Gas Measurement School

1) Principle of Operation The Electrolytic Principle How the cell works 2) Electronic Verification Basic field check procedure Cell simulation 3) Flow Calibration Topics For Discussion Acadiana Gas Measurement

### Instruction Manual HI Portable Dissolved Oxygen Meter. DO meter HI 8043 STB C OFF O 2 ZERO SLOPE C. ins rum n s

Instruction Manual HI 8043 Portable Dissolved Oxygen Meter DO meter HI 8043 ZERO SLOPE C ins rum n s This Instrument is in Compliance with the CE Directives Dear Customer, Thank you for choosing a Hanna

### OxiQuant S. Operating Instructions

ENVITEC -WISMAR GMBH Environmental & Medical Sensor Technology OxiQuant S Operating Instructions P/N: 46-00-0021 EnviteC-Wismar GmbH Alter Holzhafen 18, D-23966 Wismar, Germany Tel.: +49-(0)-3841-360-1

### 4.) There are no forces of attraction or repulsion between gas particles. This means that

KINETIC MOLECULAR (K-M) THEORY OF MATTER NOTES - based on the idea that particles of matter are always in motion - assumptions of the K-M Theory 1.) Gases consist of large numbers of tiny particles that

### Powers Controls RC 195 Single Input Receiver-Controller

Document No. 155-119P25 RC 195-4 Powers Controls RC 195 Single Input Receiver-Controller Description The RC 195 Single lnput Receiver-Controller is a pneumatic instrument which receives one pneumatic input,

### Automatic Permeability Testing: The Challenges and Solutions Author: Alyce Hartvigsen, PBI-Dansensor A/S, Ringsted, Denmark

Automatic Permeability Testing: The Challenges and Solutions Author: Alyce Hartvigsen, PBI-Dansensor A/S, Ringsted, Denmark In recent years, numerous product developments and major changes in the distribution

### What does the % represent on the beakers?

DISSOLVED OXYGEN VIDEO FAQs What does the % represent on the beakers? What are the glass tubes to beakers for? How is the temperature being kept the same (at 5 o then 35 o )? What is salinity in parts

### Gases Day 12. Phases of Matter

Phases of Matter Gases Day 12 Kinetic Molecular Theory ( Ideal Gases ) 1) The molecules of a gas are in continual, and random, motion of varying speeds. 2) The average kinetic energy of the gas molecules

### Chapter 13 Gases, Vapors, Liquids, and Solids

Chapter 13 Gases, Vapors, Liquids, and Solids Property is meaning any measurable characteristic of a substance, such as pressure, volume, or temperature, or a characteristic that can be calculated or deduced,

### CORROSION AT THE AIR-WATER INTERFACE REMOVE THE GAS

CORROSION AT THE AIR-WATER INTERFACE There are three (3) physical attributes of oxygen that explain why almost all corrosion in wet pipe fire sprinkler systems occurs in close proximity to trapped air

### PURE SUBSTANCE. Nitrogen and gaseous air are pure substances.

CLASS Third Units PURE SUBSTANCE Pure substance: A substance that has a fixed chemical composition throughout. Air is a mixture of several gases, but it is considered to be a pure substance. Nitrogen and

### DF1 and DF2 Diffusers

1550650/4 IM-P155-07 ST Issue 4 and Diffusers Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation 4. Commissioning 5. Operation 6. Maintenance

### INTRODUCTION Porosity, permeability, and pore size distribution are three closely related concepts important to filter design and filter performance.

Measurement of Filter Porosity using a Custom-Made Pycnometer George Chase The University of Akron INTRODUCTION Porosity, permeability, and pore size distribution are three closely related concepts important

### Per Section I of the ASME

notifies the control room or other Maintenance Safer & Easier remote locations that a probe column The system s intelligence can BOILER INSPECTION GUIDELINES FOR distinguish dirty probes from BOILER INSPECTION

### Determination of the Gas-Law Constant (R) using CO2

Determination of the Gas-Law Constant (R) using CO2 EXPERIMENT 11 Prepared by Edward L. Brown and Miranda Raines, Lee University The student will become familiar with ideal gases and how their properties

### Experiment C-6 Gas Solubility

1 Experiment C-6 Gas Solubility Objectives To measure dissolved oxygen in water by using an oxygen sensor. To learn about physical factors that influence oxygen solubility in water. To examine the dissolved

The Gas Law Reference Sheet 1. The Kelvin Temperature Scale Degrees Kelvin = C + 273 Convert to K 1) 27 C 2) 0. C 3) 48 C 4) 16 C 5) 106 C 2. Pressure Conversions (Add to your Conversion Sheet 1 mm Hg

### maintaining storage tank dissolved oxygen levels utilizing gas transfer membranes

Water Technologies & Solutions technical paper maintaining storage tank dissolved oxygen levels utilizing gas transfer membranes Authors: W.E. Haas, SUEZ, J. Helmrich, Florida Power and Light and J.E.

### FUNDAMENTALS OF PRESSURE RELIEF VALVES IN NATURAL GAS INSTALLATION - OPERATION - MAINTENANCE. Gary S. Beckett

FUNDAMENTALS OF PRESSURE RELIEF VALVES IN NATURAL GAS INSTALLATION - OPERATION - MAINTENANCE Gary S. Beckett Flow Safe Supply 10727 Tower Oaks Boulevard Houston, Texas 77070 What Are They and Why Are They

### Instruction Manual. Model DSP-Ex Portable Dewpoint Meter. Alpha Moisture Systems Alpha House 96 City Road Bradford BD8 8ES England

Model DSP-Ex Portable Dewpoint Meter Instruction Manual Alpha Moisture Systems Alpha House 96 City Road Bradford BD8 8ES England Tel: +44 1274 733100 Fax: +44 1274 733200 Email: info@amsytems.co.uk Web:

### BASIC PHYSICS APPLIED TO ANAESTHESIOLOGY

BASIC PHYSICS APPLIED TO ANAESTHESIOLOGY Dr.R.Selvakumar.M.D.D.A.DNB Professor of Anaesthesiology, K.A.P.Viswanatham Govt medical college, Trichy. The current practice of Anaesthesiology demands knowledge

### Heavy Duty Dissolved Oxygen Meter

User's Manual Heavy Duty Dissolved Oxygen Meter Model 407510 Test Equipment Depot - 800.517.8431-99 Washington Street Melrose, MA 02176 FAX 781.665.0780 - TestEquipmentDepot.com Introduction Congratulations

### Determination of R: The Gas-Law Constant

Determination of R: The Gas-Law Constant PURPOSE: EXPERIMENT 9 To gain a feeling for how well real gases obey the ideal-gas law and to determine the ideal-gas-law constant R. APPARATUS AND CHEMICALS: KClO

### The Ideal Gas Constant

Chem 2115 Experiment # 8 The Ideal Gas Constant OBJECTIVE: This experiment is designed to provide experience in gas handling methods and experimental insight into the relationships between pressure, volume,

### CP Chapter 13/14 Notes The Property of Gases Kinetic Molecular Theory

CP Chapter 13/14 Notes The Property of Gases Kinetic Molecular Theory Kinetic Molecular Theory of Gases The word kinetic refers to. Kinetic energy is the an object has because of its motion. Kinetic Molecular

### LOW PRESSURE EFFUSION OF GASES adapted by Luke Hanley and Mike Trenary

ADH 1/7/014 LOW PRESSURE EFFUSION OF GASES adapted by Luke Hanley and Mike Trenary This experiment will introduce you to the kinetic properties of low-pressure gases. You will make observations on the

### INTRODUCTION. The Quantum Technology system has the following advantages:, as it does not need plastic gas-bags which are volume-consuming,

1 INTRODUCTION Quantum Technology is a leading scientific equipment supplier. For the last thirty years we served our customers by tailoring our products to each customer s unique requirements. Quantum

### Autocalibration Systems for In Situ Oxygen Analyzers

Product Data Sheet PDS 106-340AC.A01 February, 2006 SPS 4001B, IMPS 4000 and MPS 3000 Autocalibration Systems for In Situ Oxygen Analyzers Cost-effective autocalibration systems for installations ranging

### Dissolved Oxygen/Temp Waterproof Tester

User's Guide AM-DO-01 Battery Compartment Cap Main Display Dissolved Oxygen/Temp Waterproof Tester Lanyard Connection old/max/min Button Function Mode Button Power / Calibration Button Probe Coller Probe

### Degassing ERC KK Saitama, Japan

Degassing Degassing Solutions IDEX Health & Science degassers improve fluidic instrument precision and reliability by removing dissolved gases from fluids before they outgas and form problem-causing bubbles

### EASIDEW TRANSMITTER with Current Source Output

EASIDEW TRANSMITTER with Current Source Output INSTALLATION, OPERATION AND MAINTENANCE MANUAL Issue March 2002 2 TABLE OF CONTENTS SECTION PAGE 1. INTRODUCTION 3 1.1 General 3 1.2 Ceramic Sensing Element

### Exercise 5-2. Bubblers EXERCISE OBJECTIVE DISCUSSION OUTLINE. Bubblers DISCUSSION. Learn to measure the level in a vessel using a bubbler.

Exercise 5-2 Bubblers EXERCISE OBJECTIVE Learn to measure the level in a vessel using a bubbler. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Bubblers How to measure

### Aerobic Respiration. Evaluation copy

Aerobic Respiration Computer 17 Aerobic cellular respiration is the process of converting the chemical energy of organic molecules into a form immediately usable by organisms. Glucose may be oxidized completely

### The Discussion of this exercise covers the following points: Range with an elevated or suppressed zero Suppressed-zero range Elevated-zero range

Exercise 4-3 Zero Suppression and Zero Elevation EXERCISE OBJECTIVE In this exercise, you will learn the effect that mounting a pressure transmitter above or below the reference level has on the hydrostatic

### The HumiSys. RH Generator. Operation. Applications. Designed, built, and supported by InstruQuest Inc.

The HumiSys RH Generator Designed, built, and supported by InstruQuest Inc. Versatile Relative Humidity Generation and Multi-Sensor System The new HumiSys with single or dual RH probes capabilities is

### A Journal of Practical and Useful Vacuum Technology. By Phil Danielson

A Journal of Practical and Useful Vacuum Technology From By Phil Danielson Thermal Conductivity Gauges Thermal conductivity pressure gauges are extremely common in vacuum technology, but an understanding

### Compact Triple Cabinet Outlet Station Model Installation and Operating Instructions

Compact Triple Cabinet Outlet Station Model 6258-1 Installation and Operating Instructions The Porter Compact Triple Outlet Station (6258-1) provides a quick, safe, and reliable method of connection to

### Accelerated Corrosion Test Chambers

Accelerated Corrosion Test Chambers Salt Spray The Series Chambers create an humid atmosphere with NaCl (salt), which is completely controlled. The samples are exposed to that atmosphere, and in a short

### Operating instruction

Operating instruction MV, XV, HG, HP, RKO, D2G, TV, BV, WB & SLV 1 Introduction 2 2 Stafsjö s knife gate valves 2 3 Technical information 2 3.1 Pressure test 2 3.2 Labelling 2 4 Storage 3 5 Transportation

### How To Use Getters and Getter Pumps

A Journal of Practical and Useful Vacuum Technology From By Phil Danielson How To Use Getters and Getter Pumps Gettering is a vacuum pumping technology that has been with us, in many forms, for almost

### OPERATING PROCEDURES

OPERATING PROCEDURES 1.0 Purpose This element identifies Petsec s Operating Procedures for its Safety and Environmental Management System (SEMS) Program; it applies to all Petsec operations. Petsec is

### UNIT 10 - GASES. Notes & Worksheets - Honors

Ideal Gas Equation 1 WKSHT 1.) What is the pressure exerted by 2.0 moles of an ideal gas when it occupies a volume of 12.0 L at 373 K? 2.) A flashbulb of volume 2.6 cm 3 contains O 2 gas at a pressure

### GM Series Dual-Block Multi-Function Gas Control Valves

Installation Sheets Manual 121 Gas Combustion Combination Controls and Systems Section G Technical Bulletin GM Issue Date 0297 GM Series Dual-Block Multi-Function Gas Control Valves Figure 1: GM Series

### Instructions for Installation, Operation, Care and Maintenance

Model B1 Accelerator with Integral Accelo-Check Bulletin 323 Rev. E Bulletin 323 Rev. E Instructions for Installation, Operation, Care and Maintenance Features 1. Quickens the operation of dry pipe valves.

### Autocalibration Systems For In Situ Oxygen Analyzers

Product Data Sheet PDS 106-340AC SPS 4000, IMPS 4000, and MPS 3000 January, 1999 Autocalibration Systems For In Situ Oxygen Analyzers Cost-effective autocalibration systems for installations ranging from

### Name: Class: Date: SHORT ANSWER Answer the following questions in the space provided.

CHAPTER 11 REVIEW Gases SECTION 1 SHORT ANSWER Answer the following questions in the space provided. 1. Pressure =. For a constant force, when the surface area is tripled the pressure is (a) doubled. (b)

### Figure Vapor-liquid equilibrium for a binary mixture. The dashed lines show the equilibrium compositions.

Another way to view this problem is to say that the final volume contains V m 3 of alcohol at 5.93 kpa and 20 C V m 3 of air at 94.07 kpa and 20 C V m 3 of air plus alcohol at 100 kpa and 20 C Thus, the

### VIC offers a variety of Calibrated Gas Leaks

CALIBRATED GAS LEAKS VIC offers a high quality line of calibrated gas leaks traceable to National Institute of Standards Technology (NIST). Calibrated gas leaks are used to provide a controlled delivery

### Application Note AN-107

SPEC Sensor TM Characterization & Calibration Considerations Scope This document is provided to describe the considerations needed to characterize, calibrate, verify and validate the measurement performance

### Design Requirements for Eyewash and Safety Shower Units

Design Requirements for Eyewash and Safety Shower Units Application: Eyewash Stations must be provided in the following areas. Laboratories Plumbed emergency eyewash equipment shall be installed in every

### Applied Physics Topics 2

Applied Physics Topics 2 Dr Andrey Varvinskiy Consultant Anaesthetist Torbay Hospital, UK EDAIC Paper B Lead and Examiner TOPICS 2 Gas Laws Other Laws: Dalton, Avogadro Critical temperature Critical pressure

### KOBOLD KSM FLOWMETER. User Instructions

KOBOLD KSM FLOWMETER User Instructions USA 1801 Parkway View Drive Pittsburgh, PA 15205 PH 412-788-2830 Canada 9A Aviation Point Claire, QC H9R 4Z2 PH 514-428-8090 www.koboldusa.com KSM_manual_08-10 Table

### Installation, Operation and Maintenance Manual for Back Pressure Regulator

Installation, Operation and Maintenance Manual for Back Pressure Regulator Model 8860 2009 Groth Corporation IOM-8860 Rev. B 12541 Ref. ID: 95565 Page 2 of 13 Table of Contents I. INTRODUCTION 3 II. DESIGN

### UNDERSTANDING A DIVE COMPUTER. by S. Angelini, Ph.D. Mares S.p.A.

UNDERSTANDING A DIVE COMPUTER by S. Angelini, Ph.D. Mares S.p.A. Dive Computer UNDERSTANDING A DIVE COMPUTER The decompression algorithm in a dive computer is an attempt to replicate the effects of a dive

### Experiment 8 GAS LAWS

Experiment 8 GAS LAWS FV 6/25/2017 MATERIALS: Amontons Law apparatus, Boyle s Law apparatus, Avogadro s Corollary apparatus, four beakers (2 L), warm-water bath, ice, barometer, digital thermometer, air

### A Reliable and Tracer Gas Independent Leak Detector for Food Packages

19 th World Conference on Non-Destructive Testing 2016 A Reliable and Tracer Gas Independent Leak Detector for Food Packages Silvio DECKER 1 1 INFICON GmbH, Köln, Germany Contact e-mail: Silvio.Decker@inficon.com-

### Dissolved Oxygen in Water. Evaluation copy. Table 1. Temperature Range ( C) Trout Smallmouth bass Caddisfly larvae

Dissolved Oxygen in Water Computer 12A Although water is composed of oxygen and hydrogen atoms, biological life in water depends upon another form of oxygen molecular oxygen. Oxygen is used by organisms

### Aeration of Dugouts or Ponds with Compressed Air

Revised April 2000 Agdex 716 (B36) Why aerate? Aeration of Dugouts or Ponds with Compressed Air D ugout aeration improves water quality by maintaining dissolved oxygen levels. Of all the methods used for

Chapter 13 Gases Copyright Cengage Learning. All rights reserved 1 Section 13.1 Pressure Why study gases? An understanding of real world phenomena. An understanding of how science works. Copyright Cengage

### Design, Building and Teaching with a Hydrostatic and Buoyancy Apparatus

Design, Building and Teaching with a Hydrostatic and Buoyancy Apparatus Mir M. Atiqullah and Norman Russell Southern Polytechnic State University Marietta, GA. ABSTRACT A typical Fluid Mechanics laboratory

### Appendix D: SOP of INNOVA 1412 Photoacoustic Multi-Gas Monitor. Description and Principle of Operation

Page 1 of 19 : SOP of INNOVA 1412 Photoacoustic Multi-Gas Monitor Description and Principle of Operation The photoacoustic multi-gas monitor (INNOVA 1412, Innova AirTech Instruments, Denmark) is a highly

### Chapter 13 Gases and Pressure. Pressure and Force. Pressure is the force per unit area on a surface. Force Area. Pressure =

Chapter 13 Gas Laws Chapter 13 Gases and Pressure Pressure and Force Pressure is the force per unit area on a surface. Pressure = Force Area Chapter 13 Gases and Pressure Gases in the Atmosphere The atmosphere

### Elements that exist as gases at 25 o C and 1 atmosphere H 2, N 2, O 2, F 2, Cl 2, He, Ne, Ar, Kr, Xe, Rn

AP Chemistry Chapter 5 Sections 5. 5.9 Note Organizer Pressure, The Gas Laws of Boyle, Charles, and Avogadro, The Ideal Gas Law, Gas Stoichiometry, Dalton s Law of Partial Pressure, The Kinetic olecular

### Compact Triple Cabinet Outlet Station Model B Installation and Operating Instructions

PORTER Parker Hannifin Corporation Porter Instrument Division 245 Township Line Rd. P.O. Box 907 Hatfield, PA 19440-0907 USA (215) 723-4000 / fax (215) 723-5106 Compact Triple Cabinet Outlet Station Model

### Gas Laws Chapter 14. Complete the following pressure conversion. Be sure to show how units cancel.

Gas Laws Chapter 14 Complete the following pressure conversion. Be sure to show how units cancel. 1 atm = 760 mm Hg = 760 torr = 101.3 kpa = 14.7 psi = 1.013 bar 1. The air pressure for a certain tire

### Figure 4 Figure 5 Figure 6

PICCV versus conventional two-way control s A with an equal percentage characteristic has historically been used to counteract the nonlinear heat emission of the coils. The result is a linear heat emission

### 2600T Series Pressure Transmitters Plugged Impulse Line Detection Diagnostic. Pressure Measurement Engineered solutions for all applications

Application Description AG/266PILD-EN Rev. C 2600T Series Pressure Transmitters Plugged Impulse Line Detection Diagnostic Pressure Measurement Engineered solutions for all applications Increase plant productivity

### Diagnosing EVAP systems By: Bernie Thompson

Diagnosing EVAP systems By: Bernie Thompson Perhaps one of the toughest jobs in the automotive field is diagnosing small leaks, especially A/C and Fuel Vapor (EVAP) leaks. Finding these leaks can be time

### LGSS Liquefied Gas & VSS Vapor (Gas) LGSS VSS LGSS. Features and Benefits

N LGSS Liquefied Gas & VSS Vapor (Gas) Sampling Systems The SENSOR Cylinder Sampling Systems allow collection of samples from gas and light liquid hydrocarbon applications in a manner that is safe to both

### Operating Guidelines for PermSelect Modules Liquid Contacting

Operating Guidelines for PermSelect Modules Liquid Contacting PermSelect PDMSXA-2500 (0.25m 2 ) PermSelect PDMSXA-7500 (0.75m 2 ) PermSelect PDMSXA-1.0 (1m 2 ) PermSelect PDMSXA-2.1 (2.1m 2 ) PermSelect

### Kinetic Molecular Theory

Kinetic Molecular Theory Name Period Unit 7 HW 1 Worksheet (Goals 1 & 2) 1. Describe how gases, liquids, and solids compare using the following table. Volume (definite or indefinite) Molecular Motion (high,

### Final Gas Law Review

Name: ate: 1 t which temperature is the vapor pressure of ethanol equal to 80 kpa?. 48. 73. 80. 101 4 Gas Molecular Mass (g/mol) 2 4 17 20 The table shown lists four gases and their molecular mass. Which

### UV Curing in an Inerted Atmosphere Equipment Update

UV Curing in an Inerted Atmosphere Equipment Update Jim Borsuk Senior Application Engineer Nordson UV Systems Group 09Apr14 Abstract Quality is never an accident. It is the result of intelligent effort

### VACOMASS. The modular air supply system in wastewater treatment plants

VACOMASS The modular air supply system in wastewater treatment plants Efficient and economical plant operation made possible by precise control of air supply VACOMASS Biology securely under control When

### Solenoid Valves used in Safety Instrumented Systems

I&M V9629R1 Solenoid Valves used in Safety Instrumented Systems Operating Manual in accordance with IEC 61508 ASCO Valves Page 1 of 7 Table of Contents 1 Introduction...3 1.1 Terms and Abbreviations...3

### INSTALLATION & START-UP INSTRUCTIONS. Clack WS1 & WS1.25 METER WATER SOFTENER SYSTEMS

INSTALLATION & START-UP INSTRUCTIONS Clack WS1 & WS1.25 METER WATER SOFTENER SYSTEMS 1999-2009 QualityWaterForLess.com - 1 - info@qualitywaterforless.com Preface: Thank you for your purchase of a new Water

### ISE-730 Oxygen Electrode and DO2-100 Currentto-Voltage

Technical Note ISE-730 and DO2-100 Overview In 1954, Dr. Leland Clark invented the first membrane-covered electrode designed to measure the concentration of oxygen in blood, or solution. This electrode

### The Next Generation for Level Control. By Ryan Spooner, Singer Valve

The Next Generation for Level Control By Ryan Spooner, Singer Valve The Next Generation for Level Control Water is an essential resource to everyday life and therefore having a way to store and distribute

### Maintenance and Troubleshooting of Pneumatic Conveying Systems for Sand in a Foundry

Maintenance and Troubleshooting of Pneumatic Conveying Systems for Sand in a Foundry Article Takeaways: Chris Doerschlag President ALB Klein Technology Group www.albkleinco.com 1. Troubleshooting Guide

### Boyle s Law: Pressure-Volume Relationship in Gases

Boyle s Law: Pressure-Volume Relationship in Gases The primary objective of this experiment is to determine the relationship between the pressure and volume of a confined gas. The gas we will use is air,

### OxyCheq Expedition-X Oxygen Analyzer. Operator s Manual

OxyCheq Expedition-X Oxygen Analyzer Operator s Manual OxyCheq 3528 Russell Road Marianna, Florida 32446 USA Tel: 850.482.0385 Email: oxy@oxycheq.com Web: http://oxycheq.com CONTENTS 1.0 Introduction..