# Product Catalogue MINICOMB. Switches for Pressure and Vacuum

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Product Catalogue MINICOMB Switches for Pressure and Vacuum PL ATEX

2 2 DEFINITION OF PRESSURE A force applied uniformly over a certain area is called pressure: p = F / A (pressure = force / area) Pressure (P) besides temperature is one of the most frequently measured physical units. The unit Pascal (Pa) is the SI unit of pressure within the metric unit system. In Europe bar is the most commonly used (SI) unit. It roughly equals with the magnitude of the atmospheric pressure. 1 bar = 0,1 MPa = 0,1 N/m 2 = 10 5 Pa Particularly in the anglo-american influenced region psi (pounds per square inch) is the most common unit. The general term pressure is not always very clear: In technical usage several types of pressure are differentiated, mainly differences between two pressure points, which in general linguistic usage all are called pressure. To avoid confusion, the various types of pressure are distinguished according to their point of reference: Absolute Pressure (Pabs) Absolute pressure always refers to the absolute vacuum, i.e. the zero-point is the absolute vacuum. A pressure gauge with measuring range 0-10 bar absolute shows the current ambient pressure (Pamb) when in nonoperating state/not installed. Ambient Pressure (Pamb) The atmospheric pressure is the ambient pressure. Atmospheric Pressure Difference (Pe) The atmospheric pressure difference, also called positive pressure (Pe+) respectively negative pressure (Pe-) is the most commonly measured type of pressure in the technical field. It refers to atmospheric pressure (Pamb) and is the difference between the atmospheric pressure (Pamb) and absolute pressure (Pabs). Pe = Pabs - Pamb Pe becomes positive when the absolute pressure is higher than the athmospheric pressure; Pe becomes negative when the absolute pressure is lower than the atmospheric pressure. A pressure gauge with measuring range 0-10 bar relative shows 0 bar when in nonoperating state/ not installed. Differential Pressure (DP) Differential pressure is the pressure difference ( P) between to measured pressures (P1, P2). P = P1 - P2 Differential pressure instruments are universal, as they can be used to as a relative pressure instrument or for hydrostatic level measurement.

3 3 WHAT IS A PRESSURE SWITCH? Pressure switches are signal elements, that can be used for measuring pressure in pressure lines for gases, vapours or liquids. Once the switching contact is triggered, setted pressure values are transformed into electrical or pneumatic signals which are necessary for the In general pressure switches have one or more fixed or adjustable switching contacts. Each switching contact has a setpoint ( SP). This setpoint corresponds to a pressure value setted on the pressure switch. When rising above or falling below this value the switching contact within the pressure switch is triggered. Due to inaccuracy the re-set point ( BSP) does not exactly match the setpoint. The difference between setpoint and re-setpoint is called hysteresis or switching differential ( SD) control and regulation of processes, e.g. safety and alarm devices. WHAT IS A MINICOMB PRESSURE SWITCH? The MINICOMB pressure switch series is a very compact-sized pressure switch for measuring compressed air, low-viscous media and non-aggressive gases. It is based on a force-balance measuring system with bellows sensors actuating one switching contact. The MINICOMB pressure switch series is approved for applications where PLc (Performance Level according to ISO 13849) is at least required (PLr) as well as for applications within hazardous area (zone 2 / 22 according to ATEX Directive 94/9/EC). The force-balance measuring system with bellows sensors is friction-free and features high repeatability and very good longterm-stability. The MINICOMB is available with pressure range 0,2-16 bar, -0,9-0 bar or -0, bar. Sub-base mounting or female threads are availbe for connecting to the process, while 4-pin plug according to ISO 4400 or plug M12x1 are available for electrical connection.

4 4 MINICOMB Pressure Switch friction-free force-balance measuring system high repeatability very good longterm stability pressure ranges -0, (+1) bar or 0,2-16 bar process connection 1/4 ISO 228 female or subbase mounting electrical connection plug ISO 4400 or plug M12x1 approved for applications with PLr = PLc approved for ATEX zones 2 / 22 MINICOMB with sub-base mounting Description The MINICOMB Series is a very compact-sized pressure switch for measuring compressed air, low-viscous media and non-aggressive gases. Operating Principle The principle of operation is the balance of forces. The working force within each bellow is counter-balanced by adjustable steel springs. This system actuates a microswitch when the working pressure overcomes the opposing force. The measuring system operates without any pistons or diaphragms, thus completly friction-free, resulting in minimal wear and no maintenance. MINICOMB with 1/4 female ISO 228 thread Approvals PL ATEX Performance Level (ISO 13849) PLc ATEX-Directive 94/9/EC Zone 2 / Zone 22

5 5 Technical Data Standard Options Function Life Cycle Pressure Range Vacuum Range Pressure-/Vacuum Range) Overpressure Safety Vacuum Safety Enclosure Material mechanical pressure switch; force-balance measuring system with bellows sensor at least 10 mio switch cycles 0,2-16 bar -0,9-0 bar -0, bar 25 bar -1 bar Aluminium Wetted Parts Material Aluminium, Brass, NBR (Gasket) FKM (Gasket) Permissalbe Media Temperature Permissable Ambient Temperature C C Temperature Deviation approx. 1% per 20 C Factory Adjustment Temperature 20 C on request Switching Contact Adjustment Accuracy Switching Accuracy Repeatability 1 micro switch (SPDT) < 1,0% FS < 1,0% FS < 1,0% FS Switching Differential (Hysteresis) < 4,0% FS on request Process Connection (alternatively) Electrical Connection (alternatively) Weight Protection sub-base mounting; sub-base mounting CNOMO; 1/4 female ISO 228 thread; Stecker EN A (ISO 4400); Stecker M12x1 approx. 160 g IP65 (IP67 with plug M12x1) Schock Resistance (XYZ-Direction) 15 g (IEC ) Vibration Resistance (XYZ-Direction) 10 g ( Hz) (IEC )

6 6 DIMENSIONS (VERSION WITH PLUG ISO 4400) MINICOMB sub-base mounting and plug ISO 4400 MINICOMB sub-base mounting CNOMO and plug ISO 4400 MINICOMB 1/4 female ISO 228 thread mounting and plug ISO 4400

7 7 DIMENSIONS (VERSION WITH PLUG M12X1) MINICOMB sub-base mounting and plug M12x1 MINICOMB sub-base mounting CNOMO and plug M12x1 MINICOMB 1/4 female ISO 228 thread mounting and plug M12x1

8 8 ELECTRICAL DATA Category AC-12 / DC-12 Switching Capacity max. steady current I max [A] on inductive load Electrical Connection shown in zero pressure condition version with plug ISO4400 version with plug M12x1 U [V] 30V 48V 60V 125V 250V I [A] AC I [A] DC 2 0,55 0,4 0, I [A] AC I [A] DC plug ISO 4400 plug M12 reference switch cycle: 30/min reference temperature: +30 C AC = cos phi ca. 0,7 DC = L/R ca 10ms Switching Capacity max. steady current I max [A] on ohmic load U [V] 30V 48V 60V 125V 250V micro switch plug ISO4400 plug M12 +UB 1 1 normally closed 2 2 normally open 3 4 GND version with plug ISO4400 version with plug M12x1 I [A] AC I [A] DC 3 1,2 0,8 0,4 --- I [A] AC I [A] DC reference switch cycle: 30/min reference temperature: +30 C

9 9 ORDER-CODES most common options Pressure Range Electrical Connection Process Connection Remarks Order-Code Sub-base mounting 1 C0011F-199-PAO ATEX, 1 C0111F-199-PAO ISO 4400 Sub-base mounting CNOMO 1 C0011F-199-CAO 0,2-16 bar 1/4 female ISO 228 thread Sub-base mounting 2 C0051F-199-4AO ATEX, 2 C0151F-199-4AO 1 C0011F-199-PDO ATEX, 1 C0111F-199-PDO M12 Sub-base mounting CNOMO 1 C0011F-199-CDO 1/4 female ISO 228 thread 2 C0051F-199-4DO ATEX, 2 C0151F-199-4DO -0,9...0 bar ISO 4400 M12 Sub-base mounting 1/4 female ISO 228 thread Sub-base mounting 1/4 female ISO 228 thread 1 C0011F-300-PAO ATEX, 1 C0111F-300-PAO 2 C0051F-300-4AO ATEX, 2 C0151F-300-4AO 1 C0011F-300-PDO ATEX, 1 C0111F-300-PDO 2 C0051F-300-4DO ATEX, 2 C0151F-300-4DO -0, bar ISO 4400 M12 Sub-base mounting 1/4 female ISO 228 thread Sub-base mounting 1/4 female ISO 228 thread 1 C0011F-301-PAO ATEX, 1 C0111F-301-PAO 2 C0051F-301-4AO ATEX, 2 C0151F-301-4AO 1 C0011F-301-PDO ATEX, 1 C0111F-301-PDO 2 C0051F-301-4DO ATEX, 2 C0151F-301-4DO 1) Counter plug and installation screws within scope of delivery 2) Counter plug within scope of delivery

10 10 CONVERSION TABLE FOR PRESSURE UNITS

11 11 INQUIRY CHECKLIST PRESSURE SWITCHES (BY FAX TO )

12 PINTER Mess- und Regeltechnik GmbH Kraichgaublick 17 Technologiepark Neckartal-Odenwald Obrigheim, Germany Phone Fax Internet All given information and/or technical data in this document have been prepared very carefully and reflect the state of the art when printed. Information and/or technical data may change without prior notice. All given information and/or technical data in this document are not binding and for information purposes only. Binding information and/or technical data can be obtained from our quotations and/or order confirmations. Please understand that we cannot be held responsible for the correctness of any given information and/or technical data in this document. When installing or maintaining PINTER products always refer to the corresponding operating manual and technical data sheet. All mentioned product names, product designations, product descriptions and logos are trademarks and property of their respective owners. CHEMSEAL, DIMIO, INDUSENS, INDUSWITCH, INTELLICOMB, MA- NOCOMB, MINICOMB are trademarks and/or registered trademarks of the PINTER Mess- und Regeltechnik GmbH and/or their affiliated companies in Germany, the European Union, Switzerland and/or other countries. The use of PINTER trademarks is prohibited if not clearly agreed otherwise. CATALOGUE.MINICOMB.12R3.EN

### Product Catalogue. P-SERIES Pressure Gauges

Product Catalogue P-SERIES Pressure Gauges 2 3 CONTENT INTRODUCTION PRESSURE GAUGES...4 P-SERIES (Gauges for Pressure and Vacuum)...6-25 P-SERIES Model P1 (Stainless Steel Bourdon-Tube Pressure Gauge)...6

### Product Catalogue INDUSENS Pressure Transmitters

SIL Product Catalogue INDUSENS Pressure Transmitters 2 3 CONTENT Introduction Pressure Transmitters... 4 Product Matrix / Products...5-41 INDUSENS Model 100 - Universal Pressure Transmitter for Pressure/High

### 1 Overview. Pressure Measurement Transmitters for basic requirements. 1/16 Siemens FI SITRANS P220 for gauge pressure

Siemens AG 204 Overview The pressure transmitter SITRANS P220 measures the gauge pressure of liquids, gases and vapors. Stainless steel measuring cell, fully welded Measuring ranges 2.5 to 600 bar (36.3

### Electropneumatic Positioner and Pneumatic Positioner Type 3760 JIS

Electropneumatic Positioner and Pneumatic Positioner Type 3760 Application Single-acting positioners for direct attachment to pneumatic control valves. Supplied with an electric input signal of 4 to 20

### Electropneumatic Positioner and Pneumatic Positioner Type 3760 JIS

Electropneumatic Positioner and Pneumatic Positioner Type 3760 Application Single-acting positioners for direct attachment to pneumatic control valves. An electric standardized signal of 4 to 20 ma or

### SITRANS P measuring instruments for pressure

SITRANS P measuring instruments for pressure Z series for gage pressure Siemens AG 008 Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and

### Instruction Manual Contact Pressure Vacuum Gauge

MS10 Instruction Manual Contact Pressure Vacuum Gauge Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance

### Electropneumatic Positioner Type 4763 Pneumatic Positioner Type 4765

Electropneumatic Positioner Type 4763 Pneumatic Positioner Type 4765 Application Single-acting positioner for attachment to pneumatic control valves. Supplied with either an electric input signal from

### Pneumatic pressure switch 20DD

Sensor system: stainless steel 0,2 to 6 bar Microswitch with gold plated contacts (suitable for intrinsically safe operation) High accuracy (max. scattering < 1,5%) Excellent sealing properties (better

### Instruction Manual Differential Pressure Transmitter

DE61 Instruction Manual Differential Pressure Transmitter Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance

### 2 Overview. SITRANS P measuring instruments for pressure. Transmitters for gage and absolute pressure. Z series for gage pressure

Z series for gage pressure Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and electronics plate Process connection Electrical connection

### Pressure transmitters EMP 2

Data sheet Pressure transmitters EMP 2 The high accuracy pressure transmitter EMP 2 is designed for monitoring and control in marine and industial applications and offers a reliable pressure measurement,

### OPTIBAR P 1010 C Technical Datasheet

Technical Datasheet Pressure transmitter with recessed diaphragm for general applications Wide variety thanks to modular design Measuring ranges up to 250 bar / 3750 psi High overload and temperature stability

### 4/2 way Pneumatic Solenoid Valve

4/2 way Pneumatic Solenoid Valve Compact design Push-over solenoid coil Exhaust air can be regulated Tube, threaded and sub-base connections Type combined with Seat valve version Type 2508 Type 2510/11

### 1 Overview. Pressure Measurement Single-range transmitters for general applications. 1/16 Siemens FI US Edition

Siemens AG 206 Overview Design Device structure without explosion protection The pressure transmitter consists of a piezoresistive measuring cell with a diaphragm installed in a stainless steel enclosure.

### Compact pressure switches for gas and air

Compact pressure es for gas and air / 5.0 Printed in Germany Edition 0.8 Nr. 9 544 6 Technical description The pressure is an adjustable compact pressure according to EN 854 for combustion plants. The

### Solenoid valves VZWF, force pilot operated

Key features and overview Function Normally closed solenoid valve with diaphragm and forced lifting. When the solenoid is energised, the differential pressure from the secondary side of the diaphragm is

### Pressure Measurement. Transmitters for basic requirements SITRANS P Z for gauge pressure. 2/4 Siemens FI Overview

f gauge pressure Siemens AG 010 Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and electronics plate Process connection Electrical connection

### Pressure Measurement Single-range transmitters for general applications

Siemens AG 206 Overview Design Device structure without explosion protection The pressure transmitter consists of a piezoresistive measuring cell with a diaphragm installed in a stainless steel enclosure.

### Model range*1) Switching pressure difference

Pneumatic pressure switch electromechanical actuated -...00 bar, G/2 Microswitch with gold plated contacts Intrinsically safe operation Electrical connection: connector acc. to IN EN 7530-803 (form A)

### Pressure Switch for gas and air GGAO-A4

Pressure Switch for gas and air GGAO-A4 CSA Certified CSA C22.2 No. LR 53222 Certification file # 201527 Commonwealth of Massachusetts Approved Product Approval code G3-0106-191 Gas pressure switch Codes

### Pressure switch for gas and air

Pressure switch for gas and air GAO-A4 GMH-A4 GML-A4 UL Listed UL 353 File # MH 16628 CSA Certified CSA C22.2 No. LR 53222 Certification # 201527 FM Approved Class 3510, 3530 File # J.I. 1Y919.AF GXX_A4

### Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC

Mechatronic pressure measurement Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC WIKA data sheet PV 35.41 Process Compact Series Applications Differential pressure monitoring

### Installation and Operation Manual

Manual Static pressure transducer with controller Differential static pressure transducer with analog output and optional PI control mode Large diaphragm element with differential transformer Transducer

### Bourdon tube pressure gauge for diaphragm seals

Bourdon tube pressure gauge for diaphragm seals and switch function, Type series BR42.. Features Bourdon tube pressure gauge for diaphragm seal and switch function Nominal range -1 3 bar to -1 15 bar,

### Series 44 Self-operated Pressure Regulators. Type 44-1 B Pressure Reducing Valve. Type 44-6 B Excess Pressure Valve. ANSI version

Series 44 Self-operated Pressure Regulators Type 44-1 B Pressure Reducing Valve Type 44-6 B Excess Pressure Valve ANSI version Application Set points from 3 to 290 psi (0.2 to 20 bar) with valves ½ NPT

### DMK 458. Pressure Transmitter for Marine and Offshore. Ceramic Sensor. accuracy according to IEC 60770: standard: 0.25 % FSO option: 0.

Pressure Transmitter for Marine and Offshore Ceramic Sensor accuracy according to IEC 60770: standard: 0.5 % FSO option: 0. % FSO Nominal pressure from 0... 40 mbar up to 0... 0 bar Output signals -wire:

### Gas Pressure Switch for DMV safety shutoff valves

Gas Pressure Switch for DMV safety shutoff valves GAO-A GMH-A GML-A UL Listed UL File # MH 668 CSA Certified CSA C. No. LR File # 07 FM Approved Class 0, 0 File # J.I. Y9A9.AF Commonwealth of Massachusetts

### Mounting and Operating Instructions EB EN. Type Supply Pressure Regulator. with increased air capacity

Type 4708-45 Supply Pressure Regulator with increased air capacity Translation of original instructions Mounting and Operating Instructions EB 8546-1 EN Edition March 2016 Note on these mounting and operating

### Pressure Switches Application Guideline

Pressure Switches Application Guideline Product Overview The Fema Pressure Switch product portfolio provides devices suitable for many applications. The portfolio contains Special functions and equipment

### Level switches Series LD Vibrating fork level switch for liquids and solids

Level switches Series LD Vibrating fork level switch for liquids and solids Robust and compact Suitable for liquids (model LD61) and solids (model LD60) No moving parts, low maintenance Corrosion resistant

### Level switches Series LD Vibrating fork level switch for liquids and solids

Level switches Series LD Vibrating fork level switch for liquids and solids Robust and compact Suitable for liquids (model LD61) and solids (model LD60) No moving parts, low maintenance Corrosion resistant

### SITRANS. Pressure transmitter SITRANS P, Z series for gauge and absolute pressure. Introduction. Safety instructions 2.

Introduction 1 Safety instructions 2 SITRANS Pressure transmitter SITRANS P, Z series for gauge and absolute pressure 7MF1564 Description 3 Assembly and connection 4 Technical data 5 Dimensional drawings

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Type 8713 can be combined with... Direct flow measurement by MEMS- Technology for nominal flow rates from 1 ml N /min to 8 l N /min (N 2 ) High accuracy and repeatability

### Proportional pressure regulators VPPE

Proportional pressure regulators VPPE Proportional pressure regulators VPPE Product range overview Function Version Pneumatic connection Proportional pressure regulator Nominal size for pressurisation/

### Pressure switch Type BCP

Data sheet Pressure switch Type BCP The BCP type is a series of dedicated switches for safety and monitoring of steam and hot water boilers. The BCP incorporates a single-pole changeover microswitch where

### Differential pressure gauge For the process industry Model , all-metal media chamber

Pressure Differential pressure gauge For the process industry Model 732.51, all-metal media chamber WIKA data sheet PM 07.05 for further approvals see page 4 Applications For gaseous and liquid aggressive

### Pressure Switch for gas and air

Switch for gas and air GAO-A4 GMH-A4 GML-A4 UL Listed UL 353 File # MH 16628 CSA Certified CSA C22.2 No. 13 and No. 0 Certification file # 201527 FM Approved Class 3510, 3530 File # J.I. 1Y919.AF Commonwealth

### Measurement of Low GAS & Liquid Flows

Measurement of Low GAS & Liquid Flows The Measurement Measurement with a float in a tapered borosilicate glass tube. Series 2100 with a 100 mm flow tube Series 2150 with a 150 mm flow tube Series 2300

### Precision pressure sensor Basic version Model CPT6020

Calibration Precision pressure sensor Basic version Model CPT6020 WIKA data sheet CT 25.13 Applications Calibration technology High-accuracy pressure monitoring Pressure sensing in critical applications

### Pressure controls, differential pressure controls type RT

Data sheet Pressure controls, differential pressure controls type RT Introduction Features An RT pressure control contains a pressure operated single-pole changeover contact, the position of which depends

### MANUAL KPS Pressure Control Valve

TetraTec Instruments GmbH Gewerbestrasse 8 71144 Steinenbronn Deutschland E-Mail: info@tetratec.de Tel.: 07157/5387-0 Fax: 07157/5387-10 MANUAL Pressure Control Valve *** VERSION 1.0 *** Update: 17.11.2006

### Series 44 Self-operated Pressure Regulators. Type 44-0 B Pressure Reducing Valve. ANSI version

Series 44 Self-operated Pressure Regulators Type 44-0 B Pressure Reducing Valve ANSI version Application Set points from 3 to 290 psi (0.2 to 20 bar) with valves ½ NPT to 1 NPT as well as NPS ½ and 1 Pressure

### Series 42 Self-operated Regulators Differential Pressure Regulators with Type 2424/Type 2428 Actuator (closing) Type Type 42-28

Series 42 Self-operated Regulators Differential Pressure Regulators with Type 2424/Type 2428 Actuator (closing) and balanced Type 2422 Valve Type 42-24 Type 42-28 Application Differential pressure regulators

### SPD310/SPD360 Air Differential

SPD310/SPD360 Air Differential Pressure SPD310 / SPD360 SPD310 / SPD360 differential pressure transmitters are intended for use in air handling systems for the monitoring of air ducts, filters and fans.

### PC10 Piezoresistive Silicon Pressure Sensor

PC10 Piezoresistive Silicon Pressure Sensor Features With constant current and constant voltage excitation options Imported highly reliable pressure die Wide temperature compensation Normalized output

### Pressure Transmitters 36 F-27839

Pressure Transmitters 36 F-27839 SPD310/SPD360 Air Differential schneider-electric.com 37 SPD310/SPD360 SPD310/SPD360 differential pressure transmitters are intended for use in air handling systems for

### Instruction Manual Contact Pressure Gauge

MS11 Instruction Manual Contact Pressure Gauge Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance 7.

### Mechanical Pressure Switches for overpressure, vacuum pressure and differential pressure

Mechanical Pressure Switches for overpressure, vacuum pressure and differential pressure Switching range: -250 +100 mbar, 1 16 mbar to 16 63 bar Temperature: max. 70 C Material: Copper, brass, stainless

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Bypass MFC with capillary technology for nominal flow rates from 5 ml N /min to 15 l N /min Applicable for aggressive gases Compact design and digital communication

### NivuBar Plus II

NivuBar Plus II Ceramic diaphragm Capacitive measuring cell 2-wire technology (output 4-20 ma) Integrated overload protection Extensive, almost front flush diaphragm Ex approval for zone 0 as standard

### Gas Pressure Switch for DMV safety shutoff valves

Gas Switch for DMV safety shutoff valves GAO-A GMH-A GML-A UL Listed UL 353 File # MH 1668 CSA Certified CSA C. No. LR 53 File # 0157 FM Approved Class 3510, 3530 File # J.I. 1Y9A9.AF Commonwealth of Massachusetts

### Glass tube flowmeters Series 2000

Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases Short mounting length and compact construction, especially indicated for control panels Easy installation. No

### Pressure transmitter PASCAL Ci4

Pressure transmitter PASCAL Ci4 for general application,type Series CI4100 Features Pressure transmitter for the measuring of relative and absolute pressures of gases, vapors and liquids High-resolution

### Series 44 Self-operated Pressure Regulators. Type 44-1 B Pressure Reducing Valve. Type 44-6 B Excess Pressure Valve

Series Self-operated Pressure Regulators Type -6 B Excess Pressure Valve Application Set points from 0.2 to 20 bar with valves G ½, G ¾, and G as well as DN, DN 2, DN 0, and DN 0 Nominal pressure PN 2

### Operating instructions Safety Rope Emergency Stop Switches ZB0052 / ZB0053 ZB0072 / ZB0073

Operating instructions Safety Rope Emergency Stop Switches UK ZB0052 / ZB0053 ZB0072 / ZB0073 7390878 / 02 03 / 2011 Contents 1 Safety instructions...3 2 Installation / set-up...4 2.1 Applications...4

### Pilot HON 625. Entwurf. Product information. serving the gas industry worldwide

Pilot HON 62 Entwurf Product information serving the gas industry worldwide Pilot HON 62 Application, characteristics Application Pilot for the gas pressure regulator HON 02 Pilot for outlet pressure control

### Pressure switch and Thermostat CAS

Data sheet Pressure switch and Thermostat CAS The CAS Series consists of a series of pressure controlled switches and temperature controlled switches. In this series, special attention has been given to

### Technical explanations for mechanical pressure switches

Technical explanations for mechanical pressure switches 7 6 5 4 7 5 6 4 Diaphragm pressure switch NO Piston pressure switch NC Change over 3 2a 1 3 2b 10 9 What is a mechanical pressure switch? Mechanical

### Gas Pressure Regulator HON 300

Product information serving the gas industry worldwide Applications, characteristics, technical data Applications direct acting gas pressure regulator, for systems in accordance with DVGW working instruction

### BSP Pressure Sensor Balluff GmbH, BA Marco Zorcic.

1 2 Standard Pressure Sensor with display Overview Pressure Range Relative nominal Overload Burst Permitted negative -1..2 bar 2 bar 4 bar 10 bar -1..10 bar 10 bar 20 bar 35 bar 0..2 bar 2 bar 4 bar 10

### Note: the lock-out circuit must be in accordance with Section 8.7 of DIN/VDE Twin thermostats ATH.-SW-2020 ATH.-SW-2070 ATH.

M. K. JUCHHEIM GmbH & Co Delivery address:mackenrodtstraße 14, 36039 Fulda, Germany Postal address: 3603 Fulda, Germany Phone: +49 661 6003-0 Fax: +49 661 6003-60 E-mail: mail@jumo.net Internet: www.jumo.net

### MB610 Industrial Pressure Transmitter

MB60 Industrial Pressure Transmitter Features Extreme Impact and vibration resistance, anti-overload, anti-corrosion; Digital compensation, wider operating ambient temperature range; High accuracy, low

### Gas density monitor With integrated transmitter Model GDM-100-TI

SF 6 gas solutions Gas density monitor With integrated transmitter Model GDM-100-TI grid Products WIKA data sheet SP 60.05 for further approvals see page 5 Applications Gas density monitoring of closed

### Pressure switch and Thermostat KPS

Data sheet Pressure switch and Thermostat KPS The KPS Series consists of a series of pressure and temperature controlled switches. In this series, special attention has been given to meeting demands for

### Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases

Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases Short mounting length and compact construction, especially indicated for control panels Easy installation. No

### Old documentation - Only for your information! Product is not available anymore!

Double solenoid valve Regulator and safety combination Infinitely variable floating operation DM-EF 07-7. Printed in Germany Rösler Druck Edition 0.0 Nr. 9 Flanges, pressure switches, line sockets, pulse

### Mechanical Pressure Measurement

Mechanical Pressure Measurement Differential Pressure Gauges Model GDP, Universal Version, with Diaphragm Element High Overpressure Safety PN 40, 100, 250 or 400 Applications Differential pressure measurement

### Pressure Switch for gas and air

Switch for gas and air GAO-A4 GMH-A4 GML-A4 UL Listed UL 353 File # MH 16628 CSA Certified CSA C22.2 No. LR 53222 Certification file # 201527 FM Approved Class 3510, 3530 File # J.I. 1Y919.AF Commonwealth

### Mechanical Pressure Switches for overpressure, vacuum pressure and differential pressure

Mechanical Pressure Switches for overpressure, vacuum pressure and differential pressure Switching range: -250 +100 mbar, 1 16 mbar to 16 63 bar Temperature: max. 70 C Material: Copper, brass, stainless

### HYDRA-line. HYDRA-Gauges Ultra High Purity Pressure Gauge Model HG. Pressure Systems. Applications. Special features. Description

Pressure Systems HYDRA-Gauges Ultra High Purity Pressure Gauge Model HG WIKA Data Sheet SP 99.20 Applications Semiconductor industry UHP-liquid chemicals distribution Special features Patented double diaphragm

### Mechanical pressure switches

18 Technical features / Advantages Diecast aluminium housing IP 54 or IP 65 version also available Wall mounting or directly on the pressure line Switching element (microswitch) Lead sealable setpoint

### Data sheet. Pressure switches and thermostats Type KPS 10/2003 DKACT.PD.P10.E B0782

Data sheet Pressure switches and thermostats Type KPS 10/2003 DKACT.PD.P10.E2.02 520B0782 Survey Type KPS pressure switches 1. Standard pressure switches Range Further -1 0 10 20 30 40 50 60 bar P e Type

### ATM - Analog Pressure Transmitter

Pressure Transmitter ATM - Analog Pressure Transmitter CUSTOMER BENEFITS Fast customization thanks to configurable product design Demountable electrical connector option to allow adjustment of zero and

### ARI-DP32 / DP33 / DP34 / DP34T / DP34Tri / DP35 Pneumatic actuator ARI-DP

ARI-DP32 / DP33 / DP34 / DP34T / DP34Tri / DP35 ARI-DP32 / DP33 Pneumatic actuator Page 2 DP-Actuator Extended stem on air failure ARI-DP34 Pneumatic actuator Page 2 ARI-DP34T Pneumatic actuator Page 6

### Hand lever valves VHER

Hand lever valves VHER Hand lever valves VHER Key features Powerful Flexible Practical -M- Flow 170 3800 l/min 4/3-way valve mid-position closed mid-position exhausted mid-position pressurised Connections:

### Mounting and Operating Instructions EB 2558 EN. Self-operated Pressure Regulators. Type Pressure Build-up Regulator

Self-operated Pressure Regulators Type 2357-31 Pressure Build-up Regulator with safety function and integrated excess pressure valve Type 2357-31 with non-return unit at port C Ports A and B with soldering

### Pressure switch and Thermostat Type KPS

MAKING MODERN LIVING POSSIBLE Data sheet Pressure switch and Thermostat Type KPS The KPS Series consists of a series of pressure and temperature controlled switches. In this series, special attention has

### Type 4708 Supply Pressure Regulator

Type 708 Pressure Regulator Application pressure regulator used to provide pneumatic measuring and control equipment with a constant air supply Set point range 0.5 to bar (8 to 90 psi) The pressure regulator

### Pressure switches and thermostats, type KPS

Description KPS units are pressure-controlled switches. The position of the contacts depends on the pressure in the inlet connection and the set scale value. In this series, special attention has been

### High-speed pressure sensor Model CPT6140

Calibration technology High-speed pressure sensor Model CPT6140 WIKA data sheet CT 25.11 Applications Testing technology Calibration technology Laboratories and maintenance shops Leak and burst applications

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### Safety Relief Valve HON 832

Safety Relief Valve HON 832 product information serving the gas industry worldwide Application, characteristics, technical data Application 2 As leak gas SRV with internal vent connection (function class

### Contents. Operating instructions. Pressure switch, heavy-duty version Model PSM-520

Operating instructions Contents EN Pressure switch, heavy-duty version Model PSM-520 1. General information 2. Design and function 3. Safety 4. Transport, packaging and storage 5. Commissioning, operation

### Safety. Operating instructions Solenoid valve for gas VG 6 VG 15/10 DANGER. Contents WARNING CAUTION. Changes to edition 09.14

15 Elster GmbH Edition 7.15 Translation from the German 519 D F NL I E DK S N P GR TR CZ PL RUS H www.docuthek.com Operating instructions Solenoid valve for gas VG VG 15/1 Contents Solenoid valve for gas

### GasMultiBloc Combined regulator and safety valve Infinitely variable air/ gas ratio control mode MBC-300-VEF MBC-700-VEF MBC-1200-VEF

GasMultiBloc Combined regulator and safety valve Infinitely variable air/ gas ratio control mode MBC-00-VEF 7.03 Max. operating pressure 30 mbar Compact design High flow values Low weight Low power consumption

### 2/2-Way Solenoid Control Valve

2/2-Way Solenoid Control Valve Excellent range (1:200) Very good response Compact valve design Orifice sizes 0.05... 2.0 mm Port connection 1/8 or sub-base Type 2871 can be combined with Type 8605 Digital

### ED 701 Hygenic Pressure Transmitter

ED 701 Hygenic Pressure Transmitter Hygienic process connections Complete range of electrical connections 4... 20 ma and Voltage outputs Accuracy: 0.1%, 0.2% and 0.4% FS Quick response time < 25 ms (Option

### Pressure controls for hydro carbons, zone II applications, Type KPE

MAKING MODERN LIVING POSSIBLE Technical brochure Pressure controls for hydro carbons, zone II applications, Type KPE KPE pressure controls for use in refrigeration and air conditioning systems with hydrocarbons

### DM01. Battery Powered Precision Digital Gauge. Stainless Steel Sensor. class 0.05

Battery Powered Stainless Steel Sensor class 0.05 Nominal pressure from 0 00 mbar up to 0... 00 bar Special characteristics modular sensor concept data logger graphic display stainless steel housing Ø