# Hydrostatic & Submersible. Level instruments. for the Australian market

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Hydrostatic & Submersible Level instruments for the Australian market

2 Hydrostatic and Submersible level transmitters Hydrostatic and submersible level instruments use the head pressure of a fluid to infer level. It s one of the oldest and most common methods around, and with good reason. Using pressure to measure level is a simple, reliable and for the most part cost effective solutions for a wide range of applications across many industries. The equation to convert pressure (P) to level (L) is: L = P ρg Thus, the equation for pressure (P) generated is: P = Lρg Where: L = level of fluid in metres above the sensor P = pressure in kpa ρ = density of the fluid in g/cm 3 g = gravity, (Can be considered 9.81 m/s 2 for most locations) Example 1 - Level where pressure is known 1. Pressure (P) of 4.9 kpa 2. Fluid density (ρ) of 1 g/cm 3 3. Gravity (g) at sea level = 9.81 m/s 2 L = = 0. 5 metres Example 2 Pressure when level is known 1. Fluid level (L) is 0.5 metres 2. Fluid density of 1 g/cm 3 3. Gravity at sea level = 9.81 m/s 2 P = = 4. 9 kpa L = 0.5 m Measurement Solutions Pty Ltd - Australia 2

3 Installation Options 1. Top mount hydrostatic level transmitter with a flexible insertion length. 2. Rigid top mount hydrostatic level transmitter for pressurised vessels. 3. Rigid top mount hydrostatic level transmitter. 4. Hygienic gauge pressure/level for pharmaceutical or sanitary applications. 5. Gauge pressure or level transmitter. 6. Differential pressure transmitter for pressurised vessels. 7. Remote differential pressure transmitter. 8. Submersible level transmitter. Measurement Solutions Pty Ltd - Australia 3

4 Gauge pressure & level transmitters For non-pressurised vessels, a gauge pressure transmitter is a convenient and relatively low cost option for level measurement. They can be easily installed at the base of the tank for easy access and inspection.the volume of water produces a pressure proportional to the level of water above the sensor Advantages of this method are: 1. Simple installation 2. Easy removal and inspection 3. Device is often located in a convenient location for maintenance staff. 4. Not effected by foam and condensate. 5. Can withstand high pressures 6. Low cost, high value level solution APC-2000 Premium SMART Gauge transmitter PCE-28/ALW Gauge transmitter with local display % accuracy - Spans as low as 10 mm of water. - HART and Profibus protocals - DNV, IECEx, Hygienic and other approvals - 5-year warranty PCE-28.SMART SMART Gauge transmitter - Local display sealed to IP65 or IP and 316SS bodies - Spans as low as 255 mm of water. - DNV, IECEx, Hygienic and other approvals PCE-28/ALW Gauge transmitter - 0.1% accuracy - Spans as low as 10 mm of water. - HART and Modbus protocals - DNV, IECEx, Hygienic and other approvals and 316SS bodies - Spans as low as 255 mm of water. - DNV, IECEx, Hygienic and other approvals - High temp options Measurement Solutions Pty Ltd - Australia 4

5 Differential pressure & level transmitters A DP transmitter is ideal for pressurised or closed tank applications. A closed vessel may not equalise to atmosphere correctly and pressurised vessel will load up the measurement with the amount of pressure present. Advantages: 1. Compensate for vessel pressure and vapour space 2. Equalise vessel pressure where ventilation is poor 3. Greater measurement stability on shallow spans 4. Low cost, high value level solution APR-2000 Premium SMART DP transmitter % accuracy - Spans as low as 10 mm of water. - HART, Profibus and Modbus - DNV, IECEx, Hygienic and other approvals - 5-year warranty APR-2000 Y Single entry DP Level transmitter - Single vertical entry - Up to 0.16% accuracy - Spans as low as 160 mm of water. - IECEx approval PRE-28 DP transmitter % accuracy and 316SS bodies - Spans as low as 160 mm of water. - DNV, IECEx, Hygienic and other approvals Measurement Solutions Pty Ltd - Australia 5

6 Submersible Level Sensors Submersible level transmitters offer a convenient and low-cost solution for measuring fluid and slurry levels in dams, tanks, weirs, wells and just about any atmospheric body of water. The sensor is fully submerged in the fluid and signal sent via the cable or to a remote head for display and transmission. Calibration features can also be provided via remote indications heads or via HART, Modbus and Profibus protocols Advantages - Levels up to 500 metres of H2O - Low cost, high value - Easy installation - Low power options for battery and solar SGE-25 and SGE-25.SMART Submersible level transmitter - Compact and low cost - Ranges from 0.15 to 500 metres of H2O - Also available with PT100 to PT1000 temperature outputs - Optional HART and Modbus outputs SGE-25S and SGE-25S.Smart Submersible level transmitter for waste and slurries - Large diaphragm for thick mediums - Resists build up - Available in titanium for brines and corrosives - Optional HART and Modbus outputs SGE-16 Slim line Submersible level transmitter - 16 mm diameter for narrow bores - Ranges up to 500 metres of H2O - Also available with PT100 to PT1000 temperature outputs APC-2000ALW/L SMART Submersible level transmitter - Configurable - High stability and accuracy - IECEx approvals Measurement Solutions Pty Ltd - Australia 6

7 Rigid hydrostatic level transmitters Aplisen s hydrostatic level transmitters can also be provided in a rigid a configuration that fixes its position. This is ideal for applications that have moving or agitated product which could cause a conventional submersible sensor to move around. Advantages: 1. Fixed position 2. Resistance to agitation 3. Can be mounted at an angle to accommodate particular geometry or to assist clearing internal obstacles. APC-2000ALW/P Rigid SMART hydrostatic level transmitter - Ranges from 200 mm up to 3000 mm - High accuracy - HART compatible - Programable via local display - Temperatures up to 260 deg C - Fixed or adjustable insertion lengths PCE-28P Rigid hydrostatic level transmitter - Ranges from 200 mm up to 3000 mm - Temperatures up to 260 deg C - IECEx and DNV approval - Fixed or adjustable insertion lengths Measurement Solutions Pty Ltd - Australia 7

8 Diaphragm Seals Diaphragm seals are a great way of isolating pressure instruments from coatings, solids, temperature, pulsation, corrosion and more. S-CompCH S-Comp S-Nord S-Mazut Large diaphragm with teflon or tantalum wetted parts for corrosives Seal with seperatable lower section and high linearity high-temperature applications with low ambient temperature Compact seal for integral and remote mounitng. SP ST S-Comp 10M Sanitary Flush faced seal for thick or coating mediums Extended face to accommodate nozzles High pressure DIN, Tri-Clover, Universal S-TK S-Homogenizer S-RC S-RCF50 Seal cleaning Homogeniser High temperature compact with flush face High temperature, high linearity with flush face Measurement Solutions Pty Ltd - Australia 8

9 Indicators IP66, 4-20 ma loop monitor Single Channel indicators with alarms Dual Channel monitor with alarm 6 Channel universal monitor with alarms Mounting Brackets C-2 PC AL FI25 Valves/Manifolds Needle Valve 2 valve manifold 3 valve manifold 5 valve manifold Universal HART Communication Bluetooth and USB compatible HART modem - Universal connectivity - Compatible with other brands - USB for communication and charging - Bluetooth for wireless communication - Programable features - Mobile app available for android devices Measurement Solutions Pty Ltd - Australia 9

10 Example Applications Foaming products DP level transmitters are a good choice for products prone to producing or experiencing foam. As the foam is light, it generates very little in the way of head pressure and its impact can be considered negligible. Large flush flanged or sanitary diaphragm seals can be equipped to address other process requirements. Monitoring of small spans Aplisens range of hydrostatic transmitters are cable of measuring levels as low as a few centimetres which makes them ideal for shallow sumps and small containers. The rigid sensing elements on the APC-2000ALW/P and PCE-28P can also be equipped with quick change mountings for rapid change over from one vessel to another. Dam and river levels An easy way to monitor levels in rivers and dams is to use a submersible level transmitter. They are simply dropped to bottom of the body of water and ideally suspended from their lifting handle. Fixed cable lengths in excess of 2000 metres are available for long runs back to a suitable panel or junction box. Measurement Solutions Pty Ltd - Australia 10

11 Monitoring of dams, bores and ground water With measurement ranges as deep as 500 mh2o, and industry leading long term stability, Aplisens family of submersible level transmitters can reliably monitor levels of ground and bore water for year of unattended service. All titanium devices can be provided for high salinity and flush faces for coating prone or particle rich fluids. The SGE family of submersibles is also available in an ultra-low power consumption format consuming only a handful of mw. Great for remote locations running on battery and solar supply. Storage vessels One of the most cost effective and reliable ways to monitor storage tank levels. Hot Oil Levels Integral mount sensors can be used on fluids in excess of 300 degrees. A high temperature radiator keeps the electronics cool while a special diaphragm seal and fill ensures stability over the operating range. Measurement Solutions Pty Ltd - Australia 11

12 Toll Free: Head Office Address 39/10 Gladstone Road Castle Hill NSW 2154 Postal address PO BOX 6826 Baulkham Hills BC NSW 2153 Phone: (02) Fax: (02) Web:

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### Pressure Measurement Single-range transmitters for general applications

Siemens AG 07 Overview Function U const. p U I EM Sensor Connection for auxiliary power supply Vent pipe with humidity filter Protective conductor connection/ Equipotential bonding Diaphragm The pressure

### 3.0 Pressure Transmitter Selection

3.0 Pressure Transmitter Selection Each Tronic Line pressure transmitter has different features to meet specific performance, environmental, and price requirements. It is not possible to describe every

### Pressure measurement is a family affair

Get to know the team of specialists! Process Instrumentation Pressure measurement is a family affair Discover our P family. usa.siemens.com/pressure Whether it be hygienic requirements or rugged and harsh

### CONTROL and INSTRUMENTATION

CONTROL and INSTRUMENTATION COURSE 500: 5 DAYS: Max 8 Candidates This course covers the key aspects of current instrumentation and process control technology and is designed to enable maintenance personnel

### L3 Pressure and Level transmitter

SENSORS FOR AND BIOPHARMA. Product Information L3 L3 Pressure and Level transmitter Range of applications Hydrostatic Level measurement in dynamic temperature environments Pressure measurement in pipes

### Series INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries. Description ALL STAINLESS DESIGN

INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries Series 2000 ALL STAINLESS DESIGN EASY CALIBRATION WITHOUT TEST PRESSURE BY 3 PUSH BUTTONS ACCURACY 0,1% 4-20 ma AND HART PROTOCOL ATEX II

### Liquid level measurement using hydrostatic pressure and buoyancy

iquid level measurement using hydrostatic pressure and buoyancy This worksheet and all related files are licensed under the Creative Commons Attribution icense, version 1.0. To view a copy of this license,

### 9700 Transmitter Submersible hydrostatic level transmitter

Specification sheet IP0078 October 2007 Level 9700 Transmitter Submersible hydrostatic level transmitter Description The 9700 series hydrostatic level transmitter is designed to perform in the most arduous

### MB260/MB261/MB264 Submersible Level Transmitter

MB260/MB261/MB264 Submersible Level Transmitter Features Separate construction. Full sealed stainless steel construction for submersible part; sensor and amplifier circuit are all in the sealed housing;

### Installation, operating and maintenance Instructions for Seemag bypass level indicator

Issue: S Date: 05-09-14 Type G35 General information The Seetru bypass magnetic level indicator, abbreviate SEEMAG, serves to show the filling level of fluids in tanks, basins, tubes etc. The Seemag operates

### Mobrey 9700 Transmitter Submersible Hydrostatic Level Transmitters

Product Data Sheet Mobrey 9700 Transmitter Mobrey 9700 Transmitter Submersible Hydrostatic Level Transmitters Hydrostatic electronic level transmitters Factory sealed and tested for submersed duty 4 20

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### Pressure Measurement Single-range transmitters for general applications

Siemens A 207 Overview Application The SITRANS P Compact pressure transmitter is designed for the special requirements of the food, pharmaceutical and biotechnology industries. The use of high-grade materials

### Expert Hydrostatic Level Transmitters

Expert Hydrostatic s General Features MJK Expert hydrostatic level transmitters are designed for level measurement by submerging the transmitter in open channels, drains and tanks. Expert hydrostatic level

### Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters

ABB MEASUREMENT & ANALYTICS WHITE PAPER Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters Remote seals protect transmitters when measuring difficult

### ATM.1ST. Pressure transmitters CUSTOMER BENEFITS

Pressure transmitters ATM.1ST CUSTOMER BENEFITS High precision pressure sensor over the whole temperature range Electronic compensation Modular sensor structure (any process and electrical connections

### Programmable Submersible Level Transmitters

Programmable Submersible Level Transmitters PTM/N/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available as

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 26.06.2014 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 03-28-2012 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### Precision Liquid Settlement Array Manual

Precision Liquid Settlement Array Manual All efforts have been made to ensure the accuracy and completeness of the information contained in this document. RST Instruments Ltd reserves the right to change

### Exercise 5-2. Bubblers EXERCISE OBJECTIVE DISCUSSION OUTLINE. Bubblers DISCUSSION. Learn to measure the level in a vessel using a bubbler.

Exercise 5-2 Bubblers EXERCISE OBJECTIVE Learn to measure the level in a vessel using a bubbler. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Bubblers How to measure

### Hitachi Differential Pressure/Pressure Transmitter

Hitachi Differential Pressure/Pressure Transmitter Hitachi Differential Pressure/Pressure Transmitter Series is based on our accumulated technology and is available in many types. This instrument is expected

### Vortex Flow Meter Wafer or Flange Connection. - Steam - Liquid - Gas

Vortex Flow Meter Wafer or Flange Connection - Steam - Liquid - Gas Working Principle & Circuit Diagram Working Principle When a column body placed in flowing fluids in pipe, a series of vortices will

### S265 Remote seals with capillary tube. 2600T Series Pressure Transmitters. Engineered solutions for all applications

Contents Data Sheet S265 2600T Series Pressure Transmitters Engineered solutions for all applications Wide range of remote seal types Allows optimum design for each application without compromising performance

### Liquid Level Measurement

Technical Article Liquid Level Measurement A pressure transmitter can be used to determine the liquid level in a tank, well, river or other body of liquid. The pressure at the bottom of a liquid filled

### THERMAL MASS FLOW MEASUREMENT FOR GASES

THERMAL MASS MEASUREMENT FOR GASES Make the Wise Choice. Choose Sage Flow Meters. A Commitment to Higher Performance SAGE METERING is a manufacturer of high performance Thermal Mass Flow Meters which measure

### Parametrable / Programmable Pressure Transmitter

Parametrable / Programmable Pressure Transmitter PTM/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available

### Technical Manual. Liquid Level Transmitter CT801-LB/S

CT801-LB/S Technical Manual SAS au Capital de 2 158 244-444 871 933 R.C.S. Bourges - APE : 2651B Headquarter : 9, rue Isaac Newton - 18000 Bourges - France Technical Manual CT801-LB/S 1 st Edition Released

### Botlek Studiegroep Badotherm. Solving measurements issues with Diaphragm seals 3 Case Studies

Botlek Studiegroep 9-4-2015 Badotherm Solving measurements issues with Diaphragm seals 3 Case Studies Prepared: 10/04/2015 Badotherm Company Summary Badotherm introduction Family owned company Founded

### Analogue and Digital Mass Flow Meters and Controllers for Gases MASS-STREAM

Analogue and Digital Mass Flow Meters and Controllers for Gases MASS-STREAM M+W Instruments Your partner Key Facts M+W Instruments was founded in 1988 and has always specialised in thermal mass flow meters

### Over 20,000 Strain Gage Target flowmeters installed since 1952.

Over 20,000 Strain Gage Target flowmeters installed since 1952. Liquid, gases, superheated and saturated steam from -320 to 500 O F and up to 15,000 psi. No frictional moving parts. Designed to withstand

### SSL. Smart Valve Positioner.

Smart Valve Positioner Smartest valve control device meeting a dynamic performance and a precise setting with a piezoelectric technology and an optimized auto-calibration program Features Auto-Calibration

### DTG - LCD Digital Temperature Gauge USER MANUAL

DTG - LCD USER MANUAL Page 1 of 7 USERS GUIDE page 1. Description 3 2. Function 3 3. Safety Instruction 3 3.1. Safety Conventions 3 3.2. Proper Use 3 4. Installation 4 4.1. Unpacking 4 4.2. Storage 4 4.3.

### APG. Intrinsically Safe, Heavy Duty Pressure Transducer Series: PT-400

Intrinsically Safe, Heavy Duty Pressure Transducer Series: PT-400 C US The PT-400 offers high accuracy and reliability over a wide range of pressures. The small size, integrated electronics, wide operating

### Pneumatic dead-weight tester Model CPB3500

Calibration technology Pneumatic dead-weight tester Model CPB3500 WIKA data sheet CT 31.22 Applications Primary standard for calibrating the scale in a pneumatic range from -1... 120 bar / -14.5... 1,600

### The Discussion of this exercise covers the following points:

Exercise 5-3 Wet Reference Leg EXERCISE OBJECTIVE Learn to measure the level in a vessel using a wet reference leg. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Measuring

### Standard Grout Header. Protected Pressure Gauge. High Pressure Gauge. In-Line Grout Header. Inline Protected Pressure Gauge

G A U G E S G R O U T H E A D E R S Protected Pressure Gauge In-line mounting accurately measures grout discharge pressure. Gauge is equipped with quick disconnect fittings that easily snaps into grout

Pressure Variation with Depth Pressure in a static fluid does not change in the horizontal direction as the horizontal forces balance each other out. However, pressure in a static fluid does change with

### CO, NO, NO 2, NO x, SO 2, CH 4, HCl, CO 2 & H ² O. Extractive low cost, low maintenance devices for continuous process and emissions monitoring.

Product Data Sheet CODEL Continuous Emission Monitoring GCEM 40 Series Extractive Gas Analyser CO, NO, NO 2, NO x, SO 2, CH 4, HCl, CO 2 & H ² O Extractive low cost, low maintenance devices for continuous

### BaseCal 3, calculating the performance of Diaphragm Seals

BaseCal 3, calculating the performance of Diaphragm Seals BASECAL: THE DIAPHRAGM SEAL PERFORMANCE CALCULATION TOOL BaseCal is the web-based performance calculation tool for diaphragm seal applications,

### THE NEW SP SERIES MAXIMUM CAPACITY MINIMUM FOOTPRINT

THE NEW SP SERIES MAXIMUM CAPACITY MINIMUM FOOTPRINT SPORTS INDUSTRY & SAFETY 2 THE NEW SP SERIES BAUER KOMPRESSOREN BAUER KOMPRESSOREN THE NEW SP SERIES 3 JOINT INNOVATION THE CONCEPT BEHIND THE NEW SP

### MJK Expert 700 / 800 / 900 / 7070 / 7060 MJK Expert 1400 / 3400

MJK Expert 700 / 800 / 900 / 7070 / 7060 MJK Expert 1400 / 3400 Submersible Hydrostatic s Rugged and versatile 2.75 GB 2.75 Expert 090311 We reserve the right to continuously improve our products and make

### CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES

SENSORS FOR MASS FLOW AND PRESSURE CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES INTRODUCTION Bronkhorst High-Tech BV is the worldwide pioneer in the field of low flow liquid metering

### STANDARD SPECIFICATION FOR LEVEL INSTRUMENTS DOCUMENT NO : 44-LK /J.02/0007-A4

Page 1 of 8 STANDARD SPECIFICATION FOR LEVEL INSTRUMENTS DOCUMENT NO : 44-LK-5100-00/J.02/0007-A4 Rev No. Issue Date Pages Rev Description Prepared A 12.08.08 8 Issued for Comments / Approval By Checked

### AMT-Ex Dewpoint Transmitter

AMT-Ex Dewpoint Transmitter Instruction Manual Alpha Moisture Systems Alpha House 96 City Road Bradford BD8 8ES England Tel: +44 1274 733100 Fax: +44 1274 733200 email: mail@amsytems.co.uk web: www.amsystems.co.uk

### Pressure measurement. Pressure gauges

Pressure gauge Pressure measurement There are four types of pressure measurements: Measurement of absolute pressure : Measurements are made with respect to absolute pressure, for example as with atmospheric

### Differential Pressure Level Solutions. Bulletin 01C25A04-01EN-Y

Differential Pressure Level Solutions Bulletin 01C25A04-01EN-Y Differential Pressure Level Solutions Liquid tank level is one of the basic measurements used in process plants but, its importance cannot

### UNIK 5600/5700 Marine Certified Pressure Sensing Platform

GE Measurement & Control UNIK 5600/5700 Marine Certified Pressure Sensing Platform The new UNIK 5600/5700 carries marine certification for most zones on-board ship, as well as Intrinsically Safe certifications.

### Model 4000 Pressure Controller

FEATURES Multiple Configurations The 4000 series pressure controller can be configured into either proportional only or proportional plus reset mode with a minimum of parts. Rugged Design Die cast aluminum

### ETd 04. Battery Powered Precision Digital Gauge. Stainless Steel Sensor. Type: ETd 04. class 0.05

Battery Powered Stainless Steel Sensor class 0.05 Nominal pressure from 0 100 mbar up to 0... 400 bar Special characteristics modular sensor concept data logger graphic display stainless steel housing

### DeltaSpan General Purpose Water Small Bore or Wastewater Pressure Level Transmitter LD34 and LD35 Series Manual

DeltaSpan General Purpose Water Small Bore or Wastewater Pressure Level Transmitter LD34 and LD35 Series Manual LD34 Series Shown LD35 Series Shown Flowline, Inc. 10500 Humbolt Street Los Alamitos, CA

### Gas Measurement Fundamentals Certification. Curriculum

Gas Certification Institute, LLC P.O. Box 131525 Houston, Texas 77219-1525 281-598-7200 Phone 281-598-7199 Fax contact@gascertification.com www.gascertification.com Curriculum Copyright 2007-2012 Gas Certification

### Ceramic Pressure Sensor

MARCH 2006 SENSORS FOR FOOD AND BIOPHARMA. Product Information DAC-341 FOOD Ceramic Pressure Sensor Application / Specified Usage Precise pressure measurement in pipes Authorizations Application Examples

### L 100. Bubble-Tube Level System. Installation, Operation and Maintenance Instructions

L 100 Bubble-Tube Level System Installation, Operation and Maintenance Instructions Figure 1 Contents Section Description Page 1.0 Introduction 2 2.0 Specifications 3 3.0 Installation 3 4.0 Warranty 6

### The Discussion of this exercise covers the following points: Range with an elevated or suppressed zero Suppressed-zero range Elevated-zero range

Exercise 4-3 Zero Suppression and Zero Elevation EXERCISE OBJECTIVE In this exercise, you will learn the effect that mounting a pressure transmitter above or below the reference level has on the hydrostatic

### ED 701 Hygenic Pressure Transmitter

ED 701 Hygenic Pressure Transmitter Hygienic process connections Complete range of electrical connections 4... 20 ma and Voltage outputs Accuracy: 0.1%, 0.2% and 0.4% FS Quick response time < 25 ms (Option

### DM01. Battery Powered Precision Digital Gauge. Stainless Steel Sensor. class 0.05

Battery Powered Stainless Steel Sensor class 0.05 Nominal pressure from 0 00 mbar up to 0... 00 bar Special characteristics modular sensor concept data logger graphic display stainless steel housing Ø

### Climatic Independent Level Sensor LAR

3 JAHRE GEWÄHRLEISTUNG SENSORS FOR FOOD AND BIOPHARMA. Product Information LAR-361 LAR-761 FOOD Climatic Independent Level Sensor LAR Application / Specified usage Hydrostatic level measurement in humid

### SITRANS P measuring instruments for pressure

Remote seals for transmitters and pressure gauges Technical description Overview In many cases the pressure transmitter and the measured medium have to be physically separated. It is then necessary to

### ADVANTAGES DESCRIPTION MECHANICAL FITTINGS AND DIMENSIONS

03304 Float Gauge: Level measurement Application: LNG tanks Level 03304 E 1 ADVANTAGES - High level accuracy - Local mechanical counter read out. - High reliability - Suitable for 12 and 8 still well.

### COSMO INSTRUMENTS CO., LTD.

Specification Sheet COSMO AIR LEAK TESTER ModelLS-1861 Meets SI Unit Requirement Improved accuracy with the new Master-Preset compensation feature Compensates for environmental changes Improves cycle time

### LEVEL Transmitters. ) of the liquid column above it and the atmospheric pressure (P atm

Hydrostatic level transmitters nivopress N GENERAL DESCRITION The NIVORESS N hydrostatic level transmitters are designed to measure the level of clean or contaminated liquids. The pressure sensor at the

### Technical Explanation

Technical Explanation for ESI Pressure Sensors Technical Explanation What is a pressure transmitter? A pressure transmitter (also called pressure transducer or pressure converter) is a component used to

### Micro Motion H-Series Hygienic Coriolis Flow and Density Meters

Product Data Sheet PS-00599, Rev. E July 2009 Micro Motion H-Series Hygienic Coriolis Flow and Density Meters Micro Motion H-Series Coriolis meters offer highly accurate flow and density measurement for

### Mini-Pro CO2 Sensor User s Manual - ANALOG MODEL -

Mini-Pro CO2 Sensor User s Manual - ANALOG MODEL - Table of Contents 1. Introduction 3 2. Instrument Setup 4 2.1 Instrument Checklist 4 2.2 Optional Accessories 4 2.3 Gas Concentration Ranges Available

### PULSAR 5000 SERIES OPERATING & INSTALLATION INSTRUCTIONS SERIES 5000 PLEASE READ CAREFULLY BEFORE INSTALLING

PULSAR 5000 SERIES OPERATING & INSTALLATION INSTRUCTIONS SERIES 5000 PLEASE READ CAREFULLY BEFORE INSTALLING Please Note: Ranges above 500mbar are designed and manufactured in accordance with sound engineering

### TYPE APPROVAL CERTIFICATE

TYPE APPROVAL CERTIFICATE Certificate No: TAA0000123 This is to certify: That the Pressure Transmitter with type designation(s) SITRANS P series DS III, P310, P410 Issued to Siemens AG - Process Industries

### The Ins and Outs of I/P Transducers

The Ins and Outs of I/P Transducers By Mark B. Levine, ControlAir Inc. General description I/P transducers are versatile instruments that use an electrical control signal to proportionally regulate gas

### Hydraulic pressure balance Model CPB5800

Calibration technology Hydraulic pressure balance Model CPB5800 WIKA data sheet CT 31.11 for further approvals see page 8 Applications Primary standard for calibrating the pressure scale in a hydraulic

### HUMOR 20. High-precision Humidity Calibrator HUMOR 20. Operation

High-precision Humidity Calibrator The role of humidity calibrations that are accurate, reproducible, and documentable is becoming more and more important. ISO quality guidelines and regulations according

### 2600T Series Pressure Transmitters Plugged Impulse Line Detection Diagnostic. Pressure Measurement Engineered solutions for all applications

Application Description AG/266PILD-EN Rev. C 2600T Series Pressure Transmitters Plugged Impulse Line Detection Diagnostic Pressure Measurement Engineered solutions for all applications Increase plant productivity

### Flow Sensors & Hydrometers Specification Sheet

Baseline Flow s and Hydrometers Baseline s smart flow sensors and hydrometers are two-wire ready flow measurement and control devices that are pre-programmed with the correct K and Offset values. In addition,

Low Cost Liquid Filled Pressure Gauges Size: 1.5", 2", and 2.5" Accuracy: 2.5% of full scale Materials: Case and ring: 304 stainless steel; Window: Polycarbonate; Seals: Buna-N; Pointer: Black-finished

### Model 575 and 575P Submersible Level Transmitters. Model 675 Submersible Level Transmitters. Model SDT Submersible Level Transmitters

Model 575 and 575P Submersible Level Transmitters Class I, Div 1 Approvals Standard or flush diaphragms for easy of cleaning Measurement ranges up to 690 of water Protected flush option available with

### Colima TOR Magnetic Level Switches

3.793.5275.200 IM-P324-03 CH Issue 1 Colima TOR Magnetic Level Switches Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation and Maintenance 4.

### E8AA. Pressure Sensor of Stainless Steel Construction Is Ideal for a Wide Range of Applications. Pressure Sensor (Stainless Steel Diaphragm)

Pressure Sensor (Stainless Steel Diaphragm) CSM DS_E_3_1 Pressure Sensor of Stainless Steel Construction Is Ideal for a Wide Range of Applications Incorporates double diaphragms consisting of SUS316L stainless

### For Multi-Parameter Meters see mvx

Design Features VORTEX IN-LINE FLOW METERS For Multi-Parameter Meters see m Principles of Operation n No moving parts to wear or fail. n Electronics can be remotely mounted up to 30.5 m (0 ft). n No fluid

### Liquid level instruments BM 24 /BM 51

KROHNE 09/2001 7.02362.21.00 GR/PRINTO Liquid level instruments / Status: 10/99 Variable area flowmeters Vortex flowmeters Flow controllers Electromagnetic flowmeters Ultrasonic flowmeters Mass flowmeters

### BULLETIN FOR ACCURATE INDICATION AND CONTROL OF GAS AND LIQUID FLOW IN LOW RANGES.

BULLETIN 20-7000 ROTAMETERS PURGE FOR ACCURATE INDICATION AND CONTROL OF GAS AND LIQUID FLOW IN LOW RANGES. INTRODUCTION McCrometer purge rotameters are precision-made instruments designed to accurately

### High Purity & Ultra High Purity

Solar and Semiconductor High Purity & Ultra High Purity Pressure and Weight Measurement R Electronic Pressure Measurement High Purity & Ultra High Purity - Pressure and Weight Measurement WIKA offers a

### Electro-Magnetic Flowmeters Electromagnetic Sensor

Data Sheet (US) Electro-Magnetic Flowmeters Electromagnetic Sensor MagMaster TM Plus MFF Unrivalled, fully traceable flow performance 0.15% accuracy Operable flow range: 1500:1 ensures more precise revenue

### CONTROL VALVE TESTING

The optimal functioning of the Control valve not only exists of sufficient body & seat tightness, but more important, the total "performance" of the valve and its controls! For an accurate and reliable

### Vortex flowmeters. Product family introduction Principle of operation Product review Applications Key product features

Vortex flowmeters introduction Product review s Key product features This document should not be duplicated, used, distributed, or disclosed for any purpose unless authorized by Siemens. Page 1 Vortex

### 2600T Series Pressure Transmitters

Installation manual IM/S264_S364-EN Rev.B 2600T Series Pressure Transmitters Models S264 and S364 Remote Seals for Smart Pressure Transmitters - 1 - ABB The Company We are an established world force in

### For Multi-Parameter Meters see mvx

VORTEX IN-LINE FLOW METERS Design Features For Multi-Parameter Meters see mvx Principles of Operation VX No moving parts to wear or fail. Electronics can be remotely mounted up to 30.5 m (0 ft). No fluid

### GasSense NDIR User Manual

INDEX INDEX... 1 1. OVERVIEW... 2 2. TECHNICAL DATA... 3 3. SPECIFICATIONS... 4 4. PRODUCT DESCRIPTION... 5 4.1 Mechanical details... 5 4.2 Piping... 7 4.3 Connections... 7 5. INSTALLATION... 10 6. CALIBRATION

### Engineering and Design Rules for Hopper Scales

Engineering and Design Rules for Hopper Scales Defining hopper scale Dimensioning hopper scale Selecting load cell mounts Ambient influences and shunt forces Hopper scale accuracy we make processes work

### Variable Area Flow Meters

Variable Area Flow Meters Over sixty years of variable area meter expertise Brooks started designing and manufacturing variable area (VA) meters, also referred to as rotameters, in 1946. Now, with more

### Pressure reducing valves Index

Index General information page Introduction 506 General introduction 507 for a steam plant 509 Product information BSP thread page Brass 510 ; BSP female thread 511 ; BSP male thread 513 Stainless steel

### PHYS 101 Previous Exam Problems

PHYS 101 Previous Exam Problems CHAPTER 14 Fluids Fluids at rest pressure vs. depth Pascal s principle Archimedes s principle Buoynat forces Fluids in motion: Continuity & Bernoulli equations 1. How deep

### Technical Data Sheet. OPTIMASS 7000 Custody Transfer Mass Flowmeter

Technical Data Sheet KROHNE 12/2004 7.02516.21.00 CH/GR OPTIMASS 7000 Custody Transfer Mass Flowmeter with single straight measuring tube OIML R117 Pattern Approval Accuracy classes 0.3 / 0.5 / 1.0 / 1.5

### 1 Overview. Pressure Measurement Single-range transmitters for general applications. 1/16 Siemens FI US Edition

Siemens AG 206 Overview Design Device structure without explosion protection The pressure transmitter consists of a piezoresistive measuring cell with a diaphragm installed in a stainless steel enclosure.

### Orifice Plate Flow Meter

Orifice Plate Flow Meter FOFT Measuring Principle The principle of measuring the differential pressure is based on the fact that a differential pressure is created across that section of a pipe where its

### Transmitter CS 21 Operation Manual

Transmitter CS 21 Operation Manual Content Page For your Safety 3 General Description 3 Detection Principle 4 Operational Notes 4 Design 4 Mounting Position of CS21 5 Mounting 6 Installation of Electrical

### AIR-X COMPACT AIR-X ON-LINE LUBRICANT AERATION MEASUREMENT. Technical Brochure ON RUNNING ENGINES

AIR-X COMPACT AIR-X ON-LINE LUBRICANT AERATION MEASUREMENT ON RUNNING ENGINES Technical Brochure 2015 Introduction The presence of air in the fluid of a working hydraulic system can cause significant performance

### Dynisco SPX Industrial (2280/2281) Series

From lab to production, providing a window into the process Dynisco SPX Industrial (2280/2281) Series Smart General Purpose Pressure Sensors Description The Dynisco SPX228X is an all welded construction