# Product Catalogue. P-SERIES Pressure Gauges

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Product Catalogue P-SERIES Pressure Gauges

2 2

3 3 CONTENT INTRODUCTION PRESSURE GAUGES...4 P-SERIES (Gauges for Pressure and Vacuum) P-SERIES Model P1 (Stainless Steel Bourdon-Tube Pressure Gauge)...6 P-SERIES Model P1K (Contact Bourdon-Tube Pressure Gauge)...10 P-SERIES Model P1TM (Transmitter Bourdon-Tube Pressure Gauge)...14 P-SERIES Model P2 (Stainless Steel Capsule Pressure Gauge)...18 P-SERIES Model P3 (Stainless Steel Diahragm Pressure Gauge) CONVERSION TABLE FOR PRESSURE UNITS...26 INQUIRY CHECKLISTE FOR PRESSURE TRANSMITTERS...27 FURTHER PRODUCTS FROM PINTER...30 IMPRINT...31

4 4 DEFINITION OF PRESSURE A force applied uniformly over a certain area is called pressure: p = F / A (pressure = force / area) Pressure (P) besides temperature is one of the most frequently measured physical units. The unit Pascal (Pa) is the SI unit of pressure within the metric unit system. In Europe bar is the most commonly used (SI) unit. It roughly equals with the magnitude of the atmospheric pressure. 1 bar = 0,1 MPa = 0,1 N/m 2 = 10 5 Pa Particularly in the anglo-american influenced region psi (pounds per square inch) is the most common unit. The general term pressure is not always very clear: In technical usage several types of pressure are differentiated, mainly differences between two pressure points, which in general linguistic usage all are called pressure. To avoid confusion, the various types of pressure are distinguished according to their point of reference: Absolute Pressure (Pabs) Absolute pressure always refers to the absolute vacuum, i.e. the zero-point is the absolute vacuum. A pressure gauge with measuring range 0-10 bar absolute shows the current ambient pressure (Pamb) when in nonoperating state/not installed. Ambient Pressure (Pamb) The atmospheric pressure is the ambient pressure. Atmospheric Pressure Difference (Pe) The atmospheric pressure difference, also called positive pressure (Pe+) respectively negative pressure (Pe-) is the most commonly measured type of pressure in the technical field. It refers to atmospheric pressure (Pamb) and is the difference between the atmospheric pressure (Pamb) and absolute pressure (Pabs). Pe = Pabs - Pamb Pe becomes positive when the absolute pressure is higher than the athmospheric pressure; Pe becomes negative when the absolute pressure is lower than the atmospheric pressure. A pressure gauge with measuring range 0-10 bar relative shows 0 bar when in nonoperating state/ not installed. Differential Pressure (DP) Differential pressure is the pressure difference ( P) between to measured pressures (P1, P2). P = P1 - P2 Differential pressure instruments are universal, as they can be used to as a relative pressure instrument or for hydrostatic level measurement.

5 5 WHAT IS A PRESSURE GAUGE? Pressure Gauges are mechanical pressure indicators. The measuring element is made of metal (stainless steels, copper alloy). Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase in pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached on this axle, which makes this movement readable on the dial. Capsule Pressure Gauges Type P2 Capsule Pressure Gauges are mainly used for measuring lowest pressure and vacuum. They are suitable for dry gasous, not crystallizing media in applications such as gas production lines, medical applications, analytical applications, filter technology, pneumatic fuel gauges, instrument calibration Based on the measuring system pressure gauge are devide in different 3 types: Bourdon-Tube Pressure Gauge Type P1 Picture 2: Capsule Pressure Gauge Diaphragm Pressure Gauges Type P3 Picture 1: Bourdon-Tube Pressure Gauge Bourdon-Tube Presure Gauges are available for vacuum, low pressure, pressure and high pressure. They are suitable for gasous and liquid, not highly viscous media in almost any industry and application, such as steam- und power technology, chemical and petrochemical industry, pharmaceutical and food industry, general mechanical engineering, pneumatics, hydraulics. Abbildung 3: Plattenfeder-Manometer Diaphragm Pressure Gauges are suitable for measuring lowest pressure and pressure of gasous and liquid, with flanged process connections also for highly viscous and crystallizing and polluted media. Applications include chemical and petrochemical industry, water and sewage industry

6 6 Bourdon-Tube Pressure Gauge Model P1 nominal size 63/100/160 mm pressure ranges from bar up to bar accuracy class 1.0 / 1.6 Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

7 7 Technical Data Standard Options Function Nominal size Pressure ranges High pressure ranges Vacuum ranges pressure gauge with bourdon-tube measuring element 63; 100; 160 mm 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) dial 100 / 160 with pressure relief sheet steel black Material process connection stainless steel (AISI 316Ti) brass Material bourdon-tube stainless steel (AISI 316Ti) brass Material movement Material dial Material pointer Material window stainless steel aluminium aluminium safety laminated glass Damping without glycerin; damped axis Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.6 (dial 63) ; class 1.0 (dial 100 / 160) Process connection Mounting Protection (EN 60529) Further Options see dimensional drawings see dimensional drawings IP54 throttle screw dual scales; customer specific dials micro adjustment; reference pointer; min/max pointer silicone-free version; cleaned for O2 service Accessories connection accessories, e.g. shut-off valve see catalogue Accessories

8 8 DIMENSIONAL DRAWINGS Bottom connection dial 63 Bottom connection dial 100/60 Bottom connection, rear flange dial 63 Bottom connection, rear flange dial 100/160 Back connection dial 63 Back connection dial 100/160 Back connection, front flange dial 63 Back connection, front flange dial 100/160 Back connection bracket mounting dial 63 Back connection bracket mounting dial 100/160 measurement table NG a a1 b b1 b2 b3 c c1 c2 d1 d2 d3 d4 d5 d6 D e G g g1 h m n 63 9, , , G1/4B , , , ,5 G1/2B ,5 20, , ,5 G1/2B NG s s1 s2 s3 s4 SW , , ,5 22

9 9 ORDER CODES (most common options) order code P 1 x x x x - x - x x x x brass 1 Material stainless steel (dial 63) 2 (wetted parts) stainless steel (dial 100/160) 3 stainless steel (O2 service)) Dial size stainless steel 3 Enclosure stainless steel, front flange 6 stainless steel, bracket mounting 7 stainless steel, rear flange 9 Windows safety laminated glass C bar bar bar bar ,6 bar ,5 bar bar bar 025 Pressure range 0-10 bar bar bar bar bar bar bar bar bar 035 without A Damping glycerin filled D damped axis C Accuracy class 1.6 (dial 63) A class 1.0 (dial 100/160) B G 1/4 B bottom, brass A G 1/4 B bottom, stainless steel B G 1/2 B bottom, brass C Process connection G 1/2 B bottom, stainless steel D G 1/4 B back, brass H G 1/4 B back, stainless steel I G 1/2 B back, brass J G 1/2 B back, stainless steel K without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

10 10 Contact-Bourdon-Tube Pressure Gauge Model P1K Nominal size 100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from bar up to bar Accuracy class 1.0 / 1.6 with either 1 or 2 magnetic spring contact(s) or with 1 or 2 inductive contact(s) Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial. The built-in contact device is available with either 1 or 2 magnetic snap-action contacts or inductive contacts. A fixed pointer at the contact opens or closes the contact when it reaches the set value.

11 11 Technical Data Standard Options Function Nominal size Pressure ranges High pressure ranges Vacuum ranges contact pressure gauge with bourdon-tube measuring element 100; 160 mm 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) dial 100 / 160 with pressure relief sheet steel black Material process connection stainless steel (AISI 316Ti) brass Material bourdon-tube stainless steel (AISI 316Ti) brass Material movement Material dial Material pointer Material window stainless steel aluminium aluminium plastic with set point adjustment Damping without silicone oil Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.0 Process connection Mounting Protection (EN 60529) see dimensional drawings see dimensional drawings IP54 Electrical connection 7-pin plug M12 connector Contacts magnetic spring contact (MK) inductive contact (IN) Accuracy approx. 5% FS approx. 0,5% FS Adjustment range Number of contacts Switching function Further Options approx. 5-95% FS 1 or 2 contact(s) normally closed or normally open throttle screw dual scales; customer specific dials silicone-free version; cleaned for O2 service

12 12 DIMENSIONAL DRAWINGS Bottom connection dial 100 Back connection dial 100 Bottom connection, front flange dial 100 Back connection, front flange dial 100 measurement table NG a a1 b b1 c c1 c2 d1 d2 d3 D e G g h k m n s s1 SW , , ,8 101,5 34,5 G1/2B

13 13 ORDER CODES (most common options) order code P 5 x x x D - x - A B D x Material (wetted parts) Dial size Enclosure Pressure range Contact(s) stainless steel (dial 100/160) 3 stainless steel (O2 service)) 5 NG NG stainless steel 3 stainless steel, rear flange bar (not with contacts MK) bar bar bar (not with contacts MK) ,6 bar (not with contacts MK) ,5 bar bar bar bar bar bar bar bar bar bar bar bar 035 1x MK 2x MK 1x IN 2x IN K L Q R

14 14 Transmitter-Bourdon-Tube Pressure Gauge Model P1TM Nominal size 100 Enclosure stainless steel Wetted parts stainless steel Pressure ranges from bar up to bar Accuracy class 1.0 with integrated pressure transmitter Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial. The integrated electronic pressure measurement, based on the measuring cell of the INDUSENS -500 series provides an analogue ouput signal for control or regulation.

15 15 Technical Data Standard Options Function Pressure ranges High pressure ranges Vacuum ranges pressure gauge with bourdon-tube measuring element and integrated pressure transmitter 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure Material process connection Material bourdon-tube Material movement Material dial Material pointer Material window stainless steel (AISI 304) with pressure relief stainless steel (AISI 316Ti) stainless steel (AISI 316Ti) stainless steel aluminium aluminium safety laminated glas Material sensor stainless steel (AISI 304L) Damping without parafine oil Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy of pressure gauge (EN 837) class 1.0 Analogue output 4-20 ma (2-wire) 0-10 V (3-wire) Accuracy of analogue output Supply for analogue output Process connection Mounting Protection (EN 60529) < 0,5% FS VDC see dimensional drawings see dimensional drawings IP54 Electrical connection 7-pin plug M12 connector Further Options throttle screw dual scales; customer specific dials

16 16 DIMENSIONAL DRAWINGS Bottom connection

17 17 ORDER CODES (most common options) order code P C - x - x B x x Pressure range Damping Process connection Output signal bar bar bar bar ,6 bar ,5 bar bar bar bar bar bar bar bar bar bar bar bar 035 without A parafine oil B 1/2 BSP male bottom, stainless steel D 1/2 BSP male back, stainless steel K 4-20 ma A 0-10 V B

18 18 Stainless Steel Capsule Pressure Gauge Model P2 Nominal size 63/100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from mbar to mbar Accuracy class 1.0 / 1.6 Description Capsule pressure gauge model P2 for vacuum and low pressure applications with aggressive, gasous dry media. Operating Principle P2 pressure gauges are pressure instruments with a capsule measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

19 19 Technical Data Standard Options Function Nominal size Low pressure ranges Vacuum ranges pressure gauge with capsule measuring element 63; 100; 160 mm 0-25 mbar; 0-40 mbar; 0-60 mbar; mbar; mbar; mbar; mbar; mbar mbar; mbar; mbar; mbar; mbar; mbar; mbar; mbar; Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) Material process connection Material capsule Material seal Material movement Material dial Material pointer Material window stainless steel (AISI 316Ti) stainless steel (AISI 316Ti) FKM stainless steel aluminium aluminium safety laminated glass Damping without glycerin (pressure range > 0-60 mbar) Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.6 Zero adjustment Process connection Mounting Protection (EN 60529) Further Options front facing see dimensional drawings see dimensional drawings IP32 throttle screw dual scales; customer specific dials reference pointer; min/max pointer (pressure range > mbar) silicone-free version; cleaned for O2 service

20 20 DIMENSIONAL DRAWINGS Bottom connection dial 63 Bottom connection dial 100/60 Bottom connection, rear flange dial 63 Bottom connection, rear flange dial 100/160 Back connection dial 63 Back connection dial 100/160 Back connection, front flange dial 63 Back connection, front flange dial 100/160 Back connection bracket mounting dial 63 Back connection bracket mounting dial 100/160 measurement table NG a a1 b b1 b2 b3 c c1 c2 d1 d2 d3 d4 d5 d6 D e G g g1 h m n 63 9, , , G1/4B , , , ,5 G1/2B ,5 20, , ,5 G1/2B NG s s1 s2 s3 s4 SW , , ,5 22

21 21 ORDER CODES (most common options) order code P 2 x x x x - x - A x x x stainless steel (dial 63) 2 Material stainless steel (dial 100/160) 3 (wetted parts) stainless steel (O2 service)) 5 NG 63 3 Dial size NG NG stainless steel 3 Enclosure stainless steel, front flange 6 stainless steel, bracket mounting 7 stainless steel, rear flange 9 Window safety laminated glass C mbar mbar mbar mbar mbar mbar mbar 00E Pressure range mbar 00D 0-25 mbar 00I 0-40 mbar 00J 0-60 mbar mbar mbar mbar mbar mbar 015 Accuracy class 1.6 A class 1.0 (dial 100/160) B Process connection 1/2 BSP male bottom, stainless steel D 1/2 BSP male back, stainless steel K without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

22 22 Stainless Steel Diaphragm Pressure Gauge Model P3 Nominal size 100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from 0-10 mbar to 0-25 bar Accuracy class 1.6 Description Diaphragm pressure gauge model P3 for low pressure and normal pressure applications with aggressive, gasous, highly viscous not chrystallizing media. Operating Principle P3 pressure gauges are pressure instruments with a diaphragm measuring element, made of stainless steel or Duratherm. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

23 23 Technical Data Standard Options Function Nominal size Low pressure ranges Pressure ranges pressure gauge with diaphragm measuring element 100; 160 mm 0-10 mbar; 0-16 mbar; 0-25 mbar; 0-40 mbar; 0-60 mbar; mbar; mbar; mbar; mbar; mbar 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above Skalenendwert Skalenendwert all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure Material process connection stainless steel (AISI 304) with pressure relief stainless steel (AISI 316Ti) Material diaphragm Duratherm (flange 100) / stainless steel (AISI 316Ti) (flange 160) Material seal Material movement Material dial Material pointer Material window FKM stainless steel aluminium aluminium safety laminated glass Damping without glycerin (pressure range > 0-60 mbar) Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,75% per 20 C on request Accuracy (EN 837) class 1.6 Process connection Protection (EN 60529) Further Options see dimensional drawings IP54 throttle screw dual scales; customer specific dials reference pointer; min/max pointer (pressure range > mbar) silicone-free version; cleaned for O2 service over pressure safety 10x FS (max. 40 bar) open process flange

24 24 DIMENSIONAL DRAWINGS bottom connection flange 100 bottom connection flange 160 bottom connection - DN 25 EN flange 100 bottom connection - DN 25 EN flange 160 bottom connection - DN 50 EN flange 100 bottom connection - DN 50 EN flange 160 measurement table NG a b c c1 c2 d1 d2 d3 d4 d5 d6 d7 d8 D D1 DN DN1 G G1 h h1 h2 h3 h4 s s1 s2 s3 s4 s5 s6 s7 t SW x18 4x G1/2B 4xM x18 4x G1/2B 4xM

25 25 ORDER CODES (most common options) order code P 3 x x 3 C - x - A A x x stainless steel (flange 160) 2 Material (wetted parts) stainless steel, Duratherm 8 (flange 100) Dial size NG NG bar bar bar bar ,6 bar ,5 bar bar bar 025 Pressure range 0-10 bar bar bar bar bar bar bar bar bar 035 Process connection 1/2 BSP male bottom, stainless steel D without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

26 26 CONVERSION TABLE FOR PRESSURE UNITS

27 27 INQUIRY CHECKLIST PRESSURE GAUGES (BY FAX TO )

28 Distributore esclusivo per l Italia DELTA STRUMENTI S.r.l Via Mattei GEMONIO (VA) Tel Fax All given information and/or technical data in this document have been prepared very carefully and reflect the state of the art when printed. Information and/or technical data may change without prior notice. All given information and/or technical data in this document are not binding and for information purposes only. Binding information and/or technical data can be obtained from our quotations and/or order confirmations. Please understand that we cannot be held responsible for the correctness of any given information and/or technical data in this document. When installing or maintaining PINTER products always refer to the corresponding operating manual and technical data sheet. All mentioned product names, product designations, product descriptions and logos are trademarks and property of their respective owners. CHEMSEAL, DIMIO, INDUSENS, INDUSWITCH, INTELLICOMB, MA- NOCOMB, MINICOMB are trademarks and/or registered trademarks of the PINTER Mess- und Regeltechnik GmbH and/or their affiliated companies in Germany, the European Union, Switzerland and/or other countries. The use of PINTER trademarks is prohibited if not clearly agreed otherwise. CATALOGUE.MANOMETER.14R1.EN

### Product Catalogue MINICOMB. Switches for Pressure and Vacuum

Product Catalogue MINICOMB Switches for Pressure and Vacuum PL ATEX 2 DEFINITION OF PRESSURE A force applied uniformly over a certain area is called pressure: p = F / A (pressure = force / area) Pressure

### Product Catalogue INDUSENS Pressure Transmitters

SIL Product Catalogue INDUSENS Pressure Transmitters 2 3 CONTENT Introduction Pressure Transmitters... 4 Product Matrix / Products...5-41 INDUSENS Model 100 - Universal Pressure Transmitter for Pressure/High

### Bourdon tube pressure gauges with glycerine filling, for chemical applications EN 837-1

Application For aggressive gaseous and liquid media which are not highly viscous and do not crystallize. Suitable for use in corrosive atmospheres. For measurement in areas subject to high vibration levels

### Bourdon tube pressure gauge for diaphragm seals

Bourdon tube pressure gauge for diaphragm seals and switch function, Type series BR42.. Features Bourdon tube pressure gauge for diaphragm seal and switch function Nominal range -1 3 bar to -1 15 bar,

### Mechanical Pressure Measurement

Mechanical Pressure Measurement Differential Pressure Gauges Model GDP, Universal Version, with Diaphragm Element High Overpressure Safety PN 40, 100, 250 or 400 Applications Differential pressure measurement

### Bourdon tube pressure gauges for chemical applications EN 837-1

Application For aggressive gaseous and liquid media which are not highly viscous and do not crystallize. Suitable for use in corrosive atmospheres.! When measuring gas or vapour, the instruments must be

### Differential pressure gauge For the process industry Model , all-metal media chamber

Pressure Differential pressure gauge For the process industry Model 732.51, all-metal media chamber WIKA data sheet PM 07.05 for further approvals see page 4 Applications For gaseous and liquid aggressive

### PRESSURE & TEMPERATURE GAUGES PRESSURE TEMPERATURE. Catalogue H20200a

PRESSURE & TEMPERATURE GAUGES PRESSURE Catalogue TEMPERATURE Applications The label Trafag Industrial Components extends the Trafag brand name to instruments manufactured by qualified partner companies.

### Bourdon tube pressure gauge, stainless steel For extremely low ambient temperatures down to -70 C Model PG23LT, with case filling

Pressure Bourdon tube pressure gauge, stainless steel For extremely low ambient temperatures down to -70 C Model PG23LT, with case filling WIKA data sheet PM 02.22 for further approvals see page 3 Applications

### Instruction Manual Contact Pressure Vacuum Gauge

MS10 Instruction Manual Contact Pressure Vacuum Gauge Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance

### Gas density monitor With integrated transmitter Model GDM-100-TI

SF 6 gas solutions Gas density monitor With integrated transmitter Model GDM-100-TI grid Products WIKA data sheet SP 60.05 for further approvals see page 5 Applications Gas density monitoring of closed

### Expansion Thermometers Stainless Steel Series, Model 70

Mechanical Temperature Measurement Expansion Thermometers Stainless Steel Series, Model 70 WIKA Data Sheet TM 81.01 Applications General-purpose temperature measuring instruments for gaseous, liquid and

### Pressure Transmitters 36 F-27839

Pressure Transmitters 36 F-27839 SPD310/SPD360 Air Differential schneider-electric.com 37 SPD310/SPD360 SPD310/SPD360 differential pressure transmitters are intended for use in air handling systems for

### Pressure gauges EN All stainless steel construction PBX100, PBX150 (1.18) Functional and constructive characteristics

Pressure gauges FT PBX 100-150-E-9.02 All stainless steel construction PBX100, PBX150 (1.18) EN 837-1 These instruments are designed for use in food, beverage, pharmaceutical, cryogenics, chemical and

### SPD310/SPD360 Air Differential

SPD310/SPD360 Air Differential Pressure SPD310 / SPD360 SPD310 / SPD360 differential pressure transmitters are intended for use in air handling systems for the monitoring of air ducts, filters and fans.

### Pressure transmitters EMP 2

Data sheet Pressure transmitters EMP 2 The high accuracy pressure transmitter EMP 2 is designed for monitoring and control in marine and industial applications and offers a reliable pressure measurement,

### Pressure switch with bourdon tube

Pressure switch with bourdon tube for plants requiring special supervision,type series BN4 Features Pressure switch with bourdon tube for plants requiring special supervision, Nominal ranges -1...0 bar

### SITRANS P measuring instruments for pressure

SITRANS P measuring instruments for pressure Z series for gage pressure Siemens AG 008 Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and

### Aluminium, white with black lettering. Bar. Option with unit bar/pa, bar/psi, kpa, MPa, psi

Ed.17.01 Application Industrial pressure gauge with liquid filling. Intented where pulsation or vibration exists. Suitable for media that will not attack copper alloy parts Design EN 837-1 Series Serie

### Oil Solutions. Pressure Gauges & Test Kits. Local solutions for individual customers worldwide

Oil Solutions Pressure Gauges & Test Kits Local solutions for individual customers worldwide www.oilsolutions.com.au sales@oilsolutions.com.au Stauff Pressure Gauges SPG 63mm and 100mm SPG-63-SB SPG-63-RB

### Model NDA Differential pressure transmitter. Deltapi N Series A complete range of pneumatic pressure transmitters

Data Sheet DS/NDA-EN Rev. H Model NDA Differential pressure transmitter Deltapi N Series A complete range of pneumatic pressure transmitters Introduction The blind type differential pressure transmitter

### BDT20 Safety process pressure gauge 100 & 160mm

BDT20 Safety process pressure gauge 100 & 160mm Product description Badotherm pressure gauge model BDT20 is the solid front, safety pattern gauge according the highest class of the EN 837-1 / 9.7.2 and

### Diaphragm monitoring system with threaded connection For the chemical and petrochemical industries Model DMS34

Pressure Diaphragm monitoring system with threaded connection For the chemical and petrochemical industries Model DMS34 WIKA data sheet DS 95.18 Applications Chemical/petrochemical industry, oil/gas industries

### OPTIBAR P 1010 C Technical Datasheet

Technical Datasheet Pressure transmitter with recessed diaphragm for general applications Wide variety thanks to modular design Measuring ranges up to 250 bar / 3750 psi High overload and temperature stability

### BSP Pressure Sensor Balluff GmbH, BA Marco Zorcic.

1 2 Standard Pressure Sensor with display Overview Pressure Range Relative nominal Overload Burst Permitted negative -1..2 bar 2 bar 4 bar 10 bar -1..10 bar 10 bar 20 bar 35 bar 0..2 bar 2 bar 4 bar 10

### ATM/GR. Electronic pressure switches CUSTOMER BENEFITS

Electronic pressure switches ATM/GR CUSTOMER BENEFITS Pressure sensor with integrated switching contacts Modular sensor structure (any process and electrical connections can be combined) Manual readjustment

### 2 Overview. SITRANS P measuring instruments for pressure. Transmitters for gage and absolute pressure. Z series for gage pressure

Z series for gage pressure Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and electronics plate Process connection Electrical connection

### Pressure gauges EN Safety pressure gauges solid-front PBXSF100, PBXSF150 (1.20) Functional and constructive characteristics

Pressure gauges FT PBXSF 100-150-E-9.02 Safety pressure gauges solid-front PBXSF100, PBXSF150 (1.20) EN 837-1 These Solid-Front instruments are built in accordance with safety specifications of EN 837.1

### Pressure Measurement. Transmitters for basic requirements SITRANS P Z for gauge pressure. 2/4 Siemens FI Overview

f gauge pressure Siemens AG 010 Overview Design The main components of the pressure transmitter are: Brass housing with silicon measuring cell and electronics plate Process connection Electrical connection

### MANUAL KPS Pressure Control Valve

TetraTec Instruments GmbH Gewerbestrasse 8 71144 Steinenbronn Deutschland E-Mail: info@tetratec.de Tel.: 07157/5387-0 Fax: 07157/5387-10 MANUAL Pressure Control Valve *** VERSION 1.0 *** Update: 17.11.2006

### SOLFRUNT Model /2" (63mm) and 4" (100mm) Stainless Steel Gauges

LFG - 14 LIQUID FILLED GAUGES SOLID FRONT STYLE SOLFRUNT Model 659 2-1/2" (63) and 4" (100) Stainless Steel Gauges DESCRIPTION When safety is your top priority, AMETEK U.S. Gauge offers the Model 659 Series

### Instruction Manual Contact Pressure Gauge

MS11 Instruction Manual Contact Pressure Gauge Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance 7.

### Hydrostatic. Continuous level measurement. Overview Page 188 VEGAWELL 52 Page 190 VEGABAR 86, 87 Page 192 Accessory Page 196 AL-EN

Hydrostatic Continuous level measurement 8 Overview Page 188 VEGAWELL 52 Page 190 VEGABAR 86, 87 Page 192 Accessory Page 196 187 Hydrostatic Level and pressure measurement Measuring principle The pressure

### Hydrostatic Level and Pressure Measurement

Hydrostatic Level and Pressure Measurement NIVUS- Your Partner In the fields of pressure measurement and hydrostatic level measurement technology NIVUS offers individual solutions for various applications.

### New! SOLFRUNT 659 Stainless Steel Pressure Gauges. Process Gauges A18

A18 New! DESCRIPTION When safety is your top priority, AMETEK U.S. Gauge offers the new Model 659 Series SOLFRUNT Gauges. The Model 659 provides users with a solid-front and pressure-relieving back design

### Nominal size dim 63mm. Option with case dim 50 mm. Aluminium, white with black lettering

Ed.17.01 Application Industrial pressure gauge with liquid filling. Intented where pulsation or vibration exists. Suitable for media that will not attack copper alloy parts Design EN 837-1 Series Serie

### Electro Controls. Pressure WattsIndustries.co.uk

Electro Controls Pressure - 07 WattsIndustries.co.uk PRESSURE AIR DIFFERENTIAL PRESSURE SWITCHES EDA.. EFS.. To monitor air flow, dirty filters, flue draught, frost on coils & level. For positive, negative,

### MEASURING INSTRUMENTS - STRUMENTI PER MISURARE

MEASURING INSTRUMENTS - STRUMENTI PER MISURARE SPECIAL PRESSURE GAUGE MN9 capsule pressure gauges DS 2.5 (63mm) PED 201/68/UE For measurement of low pressure and vacuum, for use on gas and dry air. 2.09.1

### Pressure transmitter PASCAL Ci4

Pressure transmitter PASCAL Ci4 for general application,type Series CI4100 Features Pressure transmitter for the measuring of relative and absolute pressures of gases, vapors and liquids High-resolution

### Instrumentation Catalogue

Hydraulic & Offshore Supplies Instrumentation Catalogue YOUR GATEWAY TO GLOBAL SUPPLY HOSE ASSEMBLIES PRESSURE TESTING WALFORM EQUIPMENT PACKAGES FLANGES PIPE & TUBE Hydraulic & Of f shore Supplies Specialist

### Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC

Mechatronic pressure measurement Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC WIKA data sheet PV 35.41 Process Compact Series Applications Differential pressure monitoring

### Pressure measurement. Pressure gauges

Pressure gauge Pressure measurement There are four types of pressure measurements: Measurement of absolute pressure : Measurements are made with respect to absolute pressure, for example as with atmospheric

### ED 701 Hygenic Pressure Transmitter

ED 701 Hygenic Pressure Transmitter Hygienic process connections Complete range of electrical connections 4... 20 ma and Voltage outputs Accuracy: 0.1%, 0.2% and 0.4% FS Quick response time < 25 ms (Option

### PKP Prozessmesstechnik GmbH. Borsigstrasse 24. D Wiesbaden-Nordenstadt. Tel: / Fax: / Operating Instructions

PKP Prozessmesstechnik GmbH Borsigstrasse 24 D-65205 Wiesbaden-Nordenstadt Tel: 06122 / 7055-0 Fax: 06122 / 7055 50 Operating Instructions PMR02 / PMR04 Bourdon Tube Pressure Gauge Contents 1 Introduction

### ATM - Analog Pressure Transmitter

Pressure Transmitter ATM - Analog Pressure Transmitter CUSTOMER BENEFITS Fast customization thanks to configurable product design Demountable electrical connector option to allow adjustment of zero and

### Operating Manual * * Pressure Transmitter. Table of Content. 1 Safety Guidelines. 1.3 Risks due to Non-Observance of Safety Instructions

*09005184* BA_EN_ME11 Rev.B 09/15 *09005184* d e v e l o p i n g s o l u t i o n s ME11 Operating Manual Pressure Transmitter Table of Content 1 Safety Guidelines 2 Application Purpose 3 Description of

### Instruction Manual Differential Pressure Transmitter

DE61 Instruction Manual Differential Pressure Transmitter Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance

### Parametrable / Programmable Pressure Transmitter

Parametrable / Programmable Pressure Transmitter PTM/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available

### Glass tube flowmeters Series 2000

Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases Short mounting length and compact construction, especially indicated for control panels Easy installation. No

### Series 44 Self-operated Pressure Regulators. Type 44-1 B Pressure Reducing Valve. Type 44-6 B Excess Pressure Valve. ANSI version

Series 44 Self-operated Pressure Regulators Type 44-1 B Pressure Reducing Valve Type 44-6 B Excess Pressure Valve ANSI version Application Set points from 3 to 290 psi (0.2 to 20 bar) with valves ½ NPT

### High-performance submersible pressure transmitter For level measurement Model LH-10

Electronic pressure measurement High-performance submersible pressure transmitter For level measurement Model LH-10 WIKA data sheet PE 81.09 Applications Level measurement in rivers and lakes Deep well

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### High Precision Gauge Pressure Indicators

Series 1500 8½" Dial High Precision Gauge Pressure Indicators Series 1500 Gauges have capsule-type pressure elements up to and including the 150-psig range. Ranges of 200 psig and above have Bourdon tubes.

### Mounting and Operating Instructions EB 9520 EN. Differential Pressure and Flow Meter Media 05

Differential Pressure and Flow Meter Media 05 Fig. 1 Media 05, indicating unit with differential pressure cell, with valve block and pressure gauge (right) Mounting and Operating Instructions EB 9520 EN

### Differential Pressure Transmiter

Differential Pressure Transmiter Description The is an economical alternative to established differential pressure transmitters. It combines state of the art electronics and a high performance sensor;

### Instrumentation & Data Acquisition Systems

Instrumentation & Data Acquisition Systems Section 4 - Pressure Robert W. Harrison, PE Bob@TheHarrisonHouse.com Made in USA 1 Definition of Pressure Pressure is the amount of force applied perpendicular

### Mounting and Operating Instructions EB EN. Type Supply Pressure Regulator. with increased air capacity

Type 4708-45 Supply Pressure Regulator with increased air capacity Translation of original instructions Mounting and Operating Instructions EB 8546-1 EN Edition March 2016 Note on these mounting and operating

### Bourdon Tube Pressure Gauges Ultra High Purity (UHP) Series Type

Mechanical Pressure Measurement Bourdon Tube Pressure Gauges Ultra High Purity (UHP) Series Type 230.25 WIKA Datasheet 230.25 Applications Semiconductor and electronic industry, medical industry, gene

### Test pumps. Pressure sources. Good reasons for proper and reliable calibration

Test pumps Pressure is one of the most commonly measured quantities in engineering, which is why exact and reliable pressure measurement is especially important. However, the characteristics of the even

### Electropneumatic Positioner and Pneumatic Positioner Type 3760 JIS

Electropneumatic Positioner and Pneumatic Positioner Type 3760 Application Single-acting positioners for direct attachment to pneumatic control valves. Supplied with an electric input signal of 4 to 20

### Series INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries. Description ALL STAINLESS DESIGN

INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries Series 2000 ALL STAINLESS DESIGN EASY CALIBRATION WITHOUT TEST PRESSURE BY 3 PUSH BUTTONS ACCURACY 0,1% 4-20 ma AND HART PROTOCOL ATEX II

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 26.06.2014 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### 1 Overview. Pressure Measurement Transmitters for basic requirements. 1/16 Siemens FI SITRANS P220 for gauge pressure

Siemens AG 204 Overview The pressure transmitter SITRANS P220 measures the gauge pressure of liquids, gases and vapors. Stainless steel measuring cell, fully welded Measuring ranges 2.5 to 600 bar (36.3

### Model 130M Pneumatic Controller

Instruction MI 017-450 May 1978 Model 130M Pneumatic Controller Installation and Operation Manual Control Unit Controller Model 130M Controller is a pneumatic, shelf-mounted instrument with a separate

### Series INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries. Description ALL STAINLESS DESIGN

INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries Series 2000 ALL STAINLESS DESIGN EASY CALIBRATION WITHOUT TEST PRESSURE BY 3 PUSH BUTTONS ACCURACY 0,1% 4-20 ma AND HART PROTOCOL ATEX II

### Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases

Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases Short mounting length and compact construction, especially indicated for control panels Easy installation. No

### BOSHART INDUSTRIES Telephone: Toll Free: Fax:

BOSHART INDUSTRIES 6-1 March 21/11 Steel Case Application: General Purpose - not for use with oxygen Accuracy: 3%-2%-3% Grade B Operating Temperature: 32 F-104 F (0 C-40 C) Window: Acrylic Connection:

### 2600T Series Pressure Transmitter Models 264DC Differential and 264HC Gage Level Transmitters. Kent-Taylor

INDUSTRIAL INSTRUMENTS AND CONTROLS SPECIALIST Kent-Taylor 2600T Series Pressure Transmitter Models 264DC Differential and 264HC Gage Level Transmitters Features Include Base accuracy : ±0.075% Span limits

### Electropneumatic Positioner and Pneumatic Positioner Type 3760 JIS

Electropneumatic Positioner and Pneumatic Positioner Type 3760 Application Single-acting positioners for direct attachment to pneumatic control valves. An electric standardized signal of 4 to 20 ma or

### Differential pressure

Differential pressure 9 Overview Page 200 VEGADIF 65 Page 202 Chemical seal Page 204 Accessory Page 208 199 VEGADIF For differential pressure, level and flow measurements Measuring principle The sensor

### ABB measurement & analytics data sheet. JAA, JAF and JAG Pneumatic indicating controllers

ABB measurement & analytics data sheet JAA, JAF and JAG Pneumatic indicating controllers 2 JA A, JAF an d JAG P N EUMATI C I ndicating cont rollers DS/JAx_DBT_DTE-EN Rev. H Measurement made easy Engineered

### Torque Tube TB300 Digital Transmitters

Page 1 of 7 7E.300-E Issue 4-2009 Description Series TB300 torque tube liquid level instruments utilize the buoyancy exerted on a displacer when immersed in a liquid. The buoyancy on the displacer is proportional

### Expert Hydrostatic Level Transmitters

Expert Hydrostatic s General Features MJK Expert hydrostatic level transmitters are designed for level measurement by submerging the transmitter in open channels, drains and tanks. Expert hydrostatic level

### Series 44 Self-operated Pressure Regulators. Type 44-0 B Pressure Reducing Valve. ANSI version

Series 44 Self-operated Pressure Regulators Type 44-0 B Pressure Reducing Valve ANSI version Application Set points from 3 to 290 psi (0.2 to 20 bar) with valves ½ NPT to 1 NPT as well as NPS ½ and 1 Pressure

### BOSHART INDUSTRIES Telephone: Toll Free: Fax:

6-1 Steel Case Application: General Purpose - not for use with oxygen Accuracy: 3%-2%-3% Grade B Operating Temperature: 32 F-104 F (0 C-40 C) Window: Acrylic Connection: Brass Bourdon Tube PRESSURE GAUGES

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### Instruction Manual Differential Pressure Gauge

DA09 Instruction Manual Differential Pressure Gauge Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance

### Technical Datasheet. D Series SMART Pressure Transmitter Model: DPC D-Series. Key Features. Series Overview. Product applications

Technical Datasheet D Series SMART Pressure Transmitter Model: DPC-2000 Key Features ATEX - Flameproof and Intrinsically Safe IECEx - Flameproof and Intrinsically Safe SIL 2 certificate Compliant to NAMUR

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 03-28-2012 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### High-performance submersible pressure transmitter For level measurement Model LH-10

Electronic pressure measurement High-performance submersible pressure transmitter For level measurement Model LH-10 WIKA data sheet PE 81.09 Applications Level measurement in rivers and lakes Deep well

### Pressure Transmitter P41

SENSORS FOR AND BIOPHARMA. Product Information P41 Pressure Transmitter P41 Application/Specified usage Pressure measurement in pipes and tanks For process temperatures up to 250 C (428 F) with optional

### High Purity & Ultra High Purity

Solar and Semiconductor High Purity & Ultra High Purity Pressure and Weight Measurement R Electronic Pressure Measurement High Purity & Ultra High Purity - Pressure and Weight Measurement WIKA offers a

### Air service units series K G1/8, G1/4

Air service units series K Series FK Filter 12.011 Series RK Pressure regulator 12.012 Series FRK Filter/regulator 12.013 Series OK Lubricator 12.014 Series FROK Two piece air service unit 12.015 Series

### better measurement Simply a question of SCHMIDT Flow Sensor SS The cost-effective alternative in pressurised systems up to 10 bars.

Simply a question of better measurement SCHMIDT Flow Sensor SS 20.261 The cost-effective alternative in pressurised systems up to 10 bars. Compressed air technology Industrial processes A cost analysis

### Series 44 Self-operated Pressure Regulators. Type 44-1 B Pressure Reducing Valve. Type 44-6 B Excess Pressure Valve

Series Self-operated Pressure Regulators Type -6 B Excess Pressure Valve Application Set points from 0.2 to 20 bar with valves G ½, G ¾, and G as well as DN, DN 2, DN 0, and DN 0 Nominal pressure PN 2

### Hydraulic hand test pump Models CPP700-H and CPP1000-H

Calibration technology Hydraulic hand test pump Models CPP700-H and CPP1000-H WIKA data sheet CT 91.07 for further approvals see page 2 Applications Simple test pressure generation on site, in the laboratory

### ATM.ECO - Analog Pressure Transmitter

Pressure Transmitter ATM.ECO - Analog Pressure Transmitter CUSTOMER BENEFITS Entry level series into precision pressure measurement High measurement accuracy is ensured by sophisticated digital temperature

### Series K Air service units G 1/8, G 1/4

Air service units G /8, G /4 Series FK Filter, G /8 G /4.02 Series RK Pressure regulator, G /8 G /4.04 Series FRK Filter/regulator, G /8 G /4.06 Series OK Lubricator, G /8 G /4.08 Series FROK Two piece

### Instruction Manual EMA 84 Digital Pressure Gauge

Instruction Manual EMA 84 Digital Pressure Gauge halstrup-walcher GmbH Stegener Straße 10 D-79199 Kirchzarten Phone: +49 (0) 76 61/39 63 0 Fax: +49 (0) 76 61/39 63 99 E-Mail: info@halstrup-walcher.com

### Pressure measurement explained

Rev A1, May 25 th, 2018 Sens4Knowledge Sens4 A/S Nordre Strandvej 119 G 3150 Hellebaek Denmark Phone: +45 8844 7044 Email: info@sens4.com www.sens4.com Introduction Pressure is defined as the force per

### Pressure and Temperature Gauges

Pressure and Temperature Gauges 8 10 12 14 Products Stock and Availability Pressure Gauges Pressure Gauges - Glycerine Filled Dial Thermometers Combination Gauges Cocks and Syphons Industrial Thermometers

### Integral type Differential pressure flowmeter VNT Series

Integral type Differential pressure flowmeter VNT Series OUTLINE VH series Wafer-Cone differential pressure flowmeter and high precision differential pressure transmitter are integrated into one flowmeter.

### Pressure Measurement

Pressure Measurement Manometers Sensors, Transducers Ashish J. Modi Lecturer, Dept. of Mech.Engg., Shri S.V.M. inst. Of Technology, Bharuch Pressure Pressure is a force per unit area exerted by a fluid

### Instruction Manual Differential Pressure Transmitter

DE13 Instruction Manual Differential Pressure Transmitter Table of Contents 1. Safety Instructions 2. Intended Applications 3. Product Description and Functions 4. Installation 5. Commissioning 6. Maintenance