# Product Catalogue. P-SERIES Pressure Gauges

Save this PDF as:

Size: px
Start display at page:

## Transcription

1 Product Catalogue P-SERIES Pressure Gauges

2 2

3 3 CONTENT INTRODUCTION PRESSURE GAUGES...4 P-SERIES (Gauges for Pressure and Vacuum) P-SERIES Model P1 (Stainless Steel Bourdon-Tube Pressure Gauge)...6 P-SERIES Model P1K (Contact Bourdon-Tube Pressure Gauge)...10 P-SERIES Model P1TM (Transmitter Bourdon-Tube Pressure Gauge)...14 P-SERIES Model P2 (Stainless Steel Capsule Pressure Gauge)...18 P-SERIES Model P3 (Stainless Steel Diahragm Pressure Gauge) CONVERSION TABLE FOR PRESSURE UNITS...26 INQUIRY CHECKLISTE FOR PRESSURE TRANSMITTERS...27 FURTHER PRODUCTS FROM PINTER...30 IMPRINT...31

4 4 DEFINITION OF PRESSURE A force applied uniformly over a certain area is called pressure: p = F / A (pressure = force / area) Pressure (P) besides temperature is one of the most frequently measured physical units. The unit Pascal (Pa) is the SI unit of pressure within the metric unit system. In Europe bar is the most commonly used (SI) unit. It roughly equals with the magnitude of the atmospheric pressure. 1 bar = 0,1 MPa = 0,1 N/m 2 = 10 5 Pa Particularly in the anglo-american influenced region psi (pounds per square inch) is the most common unit. The general term pressure is not always very clear: In technical usage several types of pressure are differentiated, mainly differences between two pressure points, which in general linguistic usage all are called pressure. To avoid confusion, the various types of pressure are distinguished according to their point of reference: Absolute Pressure (Pabs) Absolute pressure always refers to the absolute vacuum, i.e. the zero-point is the absolute vacuum. A pressure gauge with measuring range 0-10 bar absolute shows the current ambient pressure (Pamb) when in nonoperating state/not installed. Ambient Pressure (Pamb) The atmospheric pressure is the ambient pressure. Atmospheric Pressure Difference (Pe) The atmospheric pressure difference, also called positive pressure (Pe+) respectively negative pressure (Pe-) is the most commonly measured type of pressure in the technical field. It refers to atmospheric pressure (Pamb) and is the difference between the atmospheric pressure (Pamb) and absolute pressure (Pabs). Pe = Pabs - Pamb Pe becomes positive when the absolute pressure is higher than the athmospheric pressure; Pe becomes negative when the absolute pressure is lower than the atmospheric pressure. A pressure gauge with measuring range 0-10 bar relative shows 0 bar when in nonoperating state/ not installed. Differential Pressure (DP) Differential pressure is the pressure difference ( P) between to measured pressures (P1, P2). P = P1 - P2 Differential pressure instruments are universal, as they can be used to as a relative pressure instrument or for hydrostatic level measurement.

5 5 WHAT IS A PRESSURE GAUGE? Pressure Gauges are mechanical pressure indicators. The measuring element is made of metal (stainless steels, copper alloy). Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase in pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached on this axle, which makes this movement readable on the dial. Capsule Pressure Gauges Type P2 Capsule Pressure Gauges are mainly used for measuring lowest pressure and vacuum. They are suitable for dry gasous, not crystallizing media in applications such as gas production lines, medical applications, analytical applications, filter technology, pneumatic fuel gauges, instrument calibration Based on the measuring system pressure gauge are devide in different 3 types: Bourdon-Tube Pressure Gauge Type P1 Picture 2: Capsule Pressure Gauge Diaphragm Pressure Gauges Type P3 Picture 1: Bourdon-Tube Pressure Gauge Bourdon-Tube Presure Gauges are available for vacuum, low pressure, pressure and high pressure. They are suitable for gasous and liquid, not highly viscous media in almost any industry and application, such as steam- und power technology, chemical and petrochemical industry, pharmaceutical and food industry, general mechanical engineering, pneumatics, hydraulics. Abbildung 3: Plattenfeder-Manometer Diaphragm Pressure Gauges are suitable for measuring lowest pressure and pressure of gasous and liquid, with flanged process connections also for highly viscous and crystallizing and polluted media. Applications include chemical and petrochemical industry, water and sewage industry

6 6 Bourdon-Tube Pressure Gauge Model P1 nominal size 63/100/160 mm pressure ranges from bar up to bar accuracy class 1.0 / 1.6 Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

7 7 Technical Data Standard Options Function Nominal size Pressure ranges High pressure ranges Vacuum ranges pressure gauge with bourdon-tube measuring element 63; 100; 160 mm 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) dial 100 / 160 with pressure relief sheet steel black Material process connection stainless steel (AISI 316Ti) brass Material bourdon-tube stainless steel (AISI 316Ti) brass Material movement Material dial Material pointer Material window stainless steel aluminium aluminium safety laminated glass Damping without glycerin; damped axis Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.6 (dial 63) ; class 1.0 (dial 100 / 160) Process connection Mounting Protection (EN 60529) Further Options see dimensional drawings see dimensional drawings IP54 throttle screw dual scales; customer specific dials micro adjustment; reference pointer; min/max pointer silicone-free version; cleaned for O2 service Accessories connection accessories, e.g. shut-off valve see catalogue Accessories

8 8 DIMENSIONAL DRAWINGS Bottom connection dial 63 Bottom connection dial 100/60 Bottom connection, rear flange dial 63 Bottom connection, rear flange dial 100/160 Back connection dial 63 Back connection dial 100/160 Back connection, front flange dial 63 Back connection, front flange dial 100/160 Back connection bracket mounting dial 63 Back connection bracket mounting dial 100/160 measurement table NG a a1 b b1 b2 b3 c c1 c2 d1 d2 d3 d4 d5 d6 D e G g g1 h m n 63 9, , , G1/4B , , , ,5 G1/2B ,5 20, , ,5 G1/2B NG s s1 s2 s3 s4 SW , , ,5 22

9 9 ORDER CODES (most common options) order code P 1 x x x x - x - x x x x brass 1 Material stainless steel (dial 63) 2 (wetted parts) stainless steel (dial 100/160) 3 stainless steel (O2 service)) Dial size stainless steel 3 Enclosure stainless steel, front flange 6 stainless steel, bracket mounting 7 stainless steel, rear flange 9 Windows safety laminated glass C bar bar bar bar ,6 bar ,5 bar bar bar 025 Pressure range 0-10 bar bar bar bar bar bar bar bar bar 035 without A Damping glycerin filled D damped axis C Accuracy class 1.6 (dial 63) A class 1.0 (dial 100/160) B G 1/4 B bottom, brass A G 1/4 B bottom, stainless steel B G 1/2 B bottom, brass C Process connection G 1/2 B bottom, stainless steel D G 1/4 B back, brass H G 1/4 B back, stainless steel I G 1/2 B back, brass J G 1/2 B back, stainless steel K without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

10 10 Contact-Bourdon-Tube Pressure Gauge Model P1K Nominal size 100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from bar up to bar Accuracy class 1.0 / 1.6 with either 1 or 2 magnetic spring contact(s) or with 1 or 2 inductive contact(s) Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial. The built-in contact device is available with either 1 or 2 magnetic snap-action contacts or inductive contacts. A fixed pointer at the contact opens or closes the contact when it reaches the set value.

11 11 Technical Data Standard Options Function Nominal size Pressure ranges High pressure ranges Vacuum ranges contact pressure gauge with bourdon-tube measuring element 100; 160 mm 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) dial 100 / 160 with pressure relief sheet steel black Material process connection stainless steel (AISI 316Ti) brass Material bourdon-tube stainless steel (AISI 316Ti) brass Material movement Material dial Material pointer Material window stainless steel aluminium aluminium plastic with set point adjustment Damping without silicone oil Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.0 Process connection Mounting Protection (EN 60529) see dimensional drawings see dimensional drawings IP54 Electrical connection 7-pin plug M12 connector Contacts magnetic spring contact (MK) inductive contact (IN) Accuracy approx. 5% FS approx. 0,5% FS Adjustment range Number of contacts Switching function Further Options approx. 5-95% FS 1 or 2 contact(s) normally closed or normally open throttle screw dual scales; customer specific dials silicone-free version; cleaned for O2 service

12 12 DIMENSIONAL DRAWINGS Bottom connection dial 100 Back connection dial 100 Bottom connection, front flange dial 100 Back connection, front flange dial 100 measurement table NG a a1 b b1 c c1 c2 d1 d2 d3 D e G g h k m n s s1 SW , , ,8 101,5 34,5 G1/2B

13 13 ORDER CODES (most common options) order code P 5 x x x D - x - A B D x Material (wetted parts) Dial size Enclosure Pressure range Contact(s) stainless steel (dial 100/160) 3 stainless steel (O2 service)) 5 NG NG stainless steel 3 stainless steel, rear flange bar (not with contacts MK) bar bar bar (not with contacts MK) ,6 bar (not with contacts MK) ,5 bar bar bar bar bar bar bar bar bar bar bar bar 035 1x MK 2x MK 1x IN 2x IN K L Q R

14 14 Transmitter-Bourdon-Tube Pressure Gauge Model P1TM Nominal size 100 Enclosure stainless steel Wetted parts stainless steel Pressure ranges from bar up to bar Accuracy class 1.0 with integrated pressure transmitter Description Bourdon-tube pressure gauge model P1 vacuum, low pressure, pressure and high pressure applications with aggressive, gasous and liquid, non-crystallizing and not highly viscous media. Operating Principle P1 pressure gauges are pressure instruments with a bourdon-tube measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial. The integrated electronic pressure measurement, based on the measuring cell of the INDUSENS -500 series provides an analogue ouput signal for control or regulation.

15 15 Technical Data Standard Options Function Pressure ranges High pressure ranges Vacuum ranges pressure gauge with bourdon-tube measuring element and integrated pressure transmitter 0-0,6 bar; 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar; 0-40 bar 0-60 bar; bar; bar; bar; bar bar; ,6 bar; ,5 bar; bar; bar; bar; bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure Material process connection Material bourdon-tube Material movement Material dial Material pointer Material window stainless steel (AISI 304) with pressure relief stainless steel (AISI 316Ti) stainless steel (AISI 316Ti) stainless steel aluminium aluminium safety laminated glas Material sensor stainless steel (AISI 304L) Damping without parafine oil Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy of pressure gauge (EN 837) class 1.0 Analogue output 4-20 ma (2-wire) 0-10 V (3-wire) Accuracy of analogue output Supply for analogue output Process connection Mounting Protection (EN 60529) < 0,5% FS VDC see dimensional drawings see dimensional drawings IP54 Electrical connection 7-pin plug M12 connector Further Options throttle screw dual scales; customer specific dials

16 16 DIMENSIONAL DRAWINGS Bottom connection

17 17 ORDER CODES (most common options) order code P C - x - x B x x Pressure range Damping Process connection Output signal bar bar bar bar ,6 bar ,5 bar bar bar bar bar bar bar bar bar bar bar bar 035 without A parafine oil B 1/2 BSP male bottom, stainless steel D 1/2 BSP male back, stainless steel K 4-20 ma A 0-10 V B

18 18 Stainless Steel Capsule Pressure Gauge Model P2 Nominal size 63/100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from mbar to mbar Accuracy class 1.0 / 1.6 Description Capsule pressure gauge model P2 for vacuum and low pressure applications with aggressive, gasous dry media. Operating Principle P2 pressure gauges are pressure instruments with a capsule measuring element, made of stainless steel. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

19 19 Technical Data Standard Options Function Nominal size Low pressure ranges Vacuum ranges pressure gauge with capsule measuring element 63; 100; 160 mm 0-25 mbar; 0-40 mbar; 0-60 mbar; mbar; mbar; mbar; mbar; mbar mbar; mbar; mbar; mbar; mbar; mbar; mbar; mbar; Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above 3/4x full scale 2/3x full scale all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure stainless steel (AISI 304) Material process connection Material capsule Material seal Material movement Material dial Material pointer Material window stainless steel (AISI 316Ti) stainless steel (AISI 316Ti) FKM stainless steel aluminium aluminium safety laminated glass Damping without glycerin (pressure range > 0-60 mbar) Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,5% per 20 C on request Accuracy (EN 837) class 1.6 Zero adjustment Process connection Mounting Protection (EN 60529) Further Options front facing see dimensional drawings see dimensional drawings IP32 throttle screw dual scales; customer specific dials reference pointer; min/max pointer (pressure range > mbar) silicone-free version; cleaned for O2 service

20 20 DIMENSIONAL DRAWINGS Bottom connection dial 63 Bottom connection dial 100/60 Bottom connection, rear flange dial 63 Bottom connection, rear flange dial 100/160 Back connection dial 63 Back connection dial 100/160 Back connection, front flange dial 63 Back connection, front flange dial 100/160 Back connection bracket mounting dial 63 Back connection bracket mounting dial 100/160 measurement table NG a a1 b b1 b2 b3 c c1 c2 d1 d2 d3 d4 d5 d6 D e G g g1 h m n 63 9, , , G1/4B , , , ,5 G1/2B ,5 20, , ,5 G1/2B NG s s1 s2 s3 s4 SW , , ,5 22

21 21 ORDER CODES (most common options) order code P 2 x x x x - x - A x x x stainless steel (dial 63) 2 Material stainless steel (dial 100/160) 3 (wetted parts) stainless steel (O2 service)) 5 NG 63 3 Dial size NG NG stainless steel 3 Enclosure stainless steel, front flange 6 stainless steel, bracket mounting 7 stainless steel, rear flange 9 Window safety laminated glass C mbar mbar mbar mbar mbar mbar mbar 00E Pressure range mbar 00D 0-25 mbar 00I 0-40 mbar 00J 0-60 mbar mbar mbar mbar mbar mbar 015 Accuracy class 1.6 A class 1.0 (dial 100/160) B Process connection 1/2 BSP male bottom, stainless steel D 1/2 BSP male back, stainless steel K without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

22 22 Stainless Steel Diaphragm Pressure Gauge Model P3 Nominal size 100/160 mm Enclosure stainless steel Wetted parts stainless steel Pressure ranges from 0-10 mbar to 0-25 bar Accuracy class 1.6 Description Diaphragm pressure gauge model P3 for low pressure and normal pressure applications with aggressive, gasous, highly viscous not chrystallizing media. Operating Principle P3 pressure gauges are pressure instruments with a diaphragm measuring element, made of stainless steel or Duratherm. Once the measuring element is pressurized, it deforms proportionally to the increase of pressure ( expand during increase of pressure and/or, pull together with negative pressure). With the controlled deformation of the measuring element, it travels its way, which then is transmitted to an axle. A pointer is attached to this axle, which makes this movement readable on the dial.

23 23 Technical Data Standard Options Function Nominal size Low pressure ranges Pressure ranges pressure gauge with diaphragm measuring element 100; 160 mm 0-10 mbar; 0-16 mbar; 0-25 mbar; 0-40 mbar; 0-60 mbar; mbar; mbar; mbar; mbar; mbar 0-1 bar; 0-1,6 bar; 0-2,5 bar; 0-4 bar; 0-6 bar; 0-10 bar; 0-16 bar; 0-25 bar Pressure units Operative range (stat. load) Operative range (dyn. load) (m)bar units as above Skalenendwert Skalenendwert all common pressure/vacuum units such as psi, kpa, MPa,... Material enclosure Material process connection stainless steel (AISI 304) with pressure relief stainless steel (AISI 316Ti) Material diaphragm Duratherm (flange 100) / stainless steel (AISI 316Ti) (flange 160) Material seal Material movement Material dial Material pointer Material window FKM stainless steel aluminium aluminium safety laminated glass Damping without glycerin (pressure range > 0-60 mbar) Permissable media temperature C on request Permissable ambient temperature C on request Temperature deviation approx. 0,75% per 20 C on request Accuracy (EN 837) class 1.6 Process connection Protection (EN 60529) Further Options see dimensional drawings IP54 throttle screw dual scales; customer specific dials reference pointer; min/max pointer (pressure range > mbar) silicone-free version; cleaned for O2 service over pressure safety 10x FS (max. 40 bar) open process flange

24 24 DIMENSIONAL DRAWINGS bottom connection flange 100 bottom connection flange 160 bottom connection - DN 25 EN flange 100 bottom connection - DN 25 EN flange 160 bottom connection - DN 50 EN flange 100 bottom connection - DN 50 EN flange 160 measurement table NG a b c c1 c2 d1 d2 d3 d4 d5 d6 d7 d8 D D1 DN DN1 G G1 h h1 h2 h3 h4 s s1 s2 s3 s4 s5 s6 s7 t SW x18 4x G1/2B 4xM x18 4x G1/2B 4xM

25 25 ORDER CODES (most common options) order code P 3 x x 3 C - x - A A x x stainless steel (flange 160) 2 Material (wetted parts) stainless steel, Duratherm 8 (flange 100) Dial size NG NG bar bar bar bar ,6 bar ,5 bar bar bar 025 Pressure range 0-10 bar bar bar bar bar bar bar bar bar 035 Process connection 1/2 BSP male bottom, stainless steel D without further options O further options dual-scale bar / psi I dual-scale bar / kpa AU further options in text #

26 26 CONVERSION TABLE FOR PRESSURE UNITS

27 27 INQUIRY CHECKLIST PRESSURE GAUGES (BY FAX TO )

28 Distributore esclusivo per l Italia DELTA STRUMENTI S.r.l Via Mattei GEMONIO (VA) Tel Fax All given information and/or technical data in this document have been prepared very carefully and reflect the state of the art when printed. Information and/or technical data may change without prior notice. All given information and/or technical data in this document are not binding and for information purposes only. Binding information and/or technical data can be obtained from our quotations and/or order confirmations. Please understand that we cannot be held responsible for the correctness of any given information and/or technical data in this document. When installing or maintaining PINTER products always refer to the corresponding operating manual and technical data sheet. All mentioned product names, product designations, product descriptions and logos are trademarks and property of their respective owners. CHEMSEAL, DIMIO, INDUSENS, INDUSWITCH, INTELLICOMB, MA- NOCOMB, MINICOMB are trademarks and/or registered trademarks of the PINTER Mess- und Regeltechnik GmbH and/or their affiliated companies in Germany, the European Union, Switzerland and/or other countries. The use of PINTER trademarks is prohibited if not clearly agreed otherwise. CATALOGUE.MANOMETER.14R1.EN

### Pressure gauges EN All stainless steel construction PBX100, PBX150 (1.18) Functional and constructive characteristics

Pressure gauges FT PBX 100-150-E-9.02 All stainless steel construction PBX100, PBX150 (1.18) EN 837-1 These instruments are designed for use in food, beverage, pharmaceutical, cryogenics, chemical and

### SOLFRUNT Model /2" (63mm) and 4" (100mm) Stainless Steel Gauges

LFG - 14 LIQUID FILLED GAUGES SOLID FRONT STYLE SOLFRUNT Model 659 2-1/2" (63) and 4" (100) Stainless Steel Gauges DESCRIPTION When safety is your top priority, AMETEK U.S. Gauge offers the Model 659 Series

### Nominal size dim 63mm. Option with case dim 50 mm. Aluminium, white with black lettering

Ed.17.01 Application Industrial pressure gauge with liquid filling. Intented where pulsation or vibration exists. Suitable for media that will not attack copper alloy parts Design EN 837-1 Series Serie

### ED 701 Hygenic Pressure Transmitter

ED 701 Hygenic Pressure Transmitter Hygienic process connections Complete range of electrical connections 4... 20 ma and Voltage outputs Accuracy: 0.1%, 0.2% and 0.4% FS Quick response time < 25 ms (Option

### Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC

Mechatronic pressure measurement Compact differential pressure switch Flameproof enclosure Ex d Models DE, DEC WIKA data sheet PV 35.41 Process Compact Series Applications Differential pressure monitoring

### MANUAL KPS Pressure Control Valve

TetraTec Instruments GmbH Gewerbestrasse 8 71144 Steinenbronn Deutschland E-Mail: info@tetratec.de Tel.: 07157/5387-0 Fax: 07157/5387-10 MANUAL Pressure Control Valve *** VERSION 1.0 *** Update: 17.11.2006

### High Precision Gauge Pressure Indicators

Series 1500 8½" Dial High Precision Gauge Pressure Indicators Series 1500 Gauges have capsule-type pressure elements up to and including the 150-psig range. Ranges of 200 psig and above have Bourdon tubes.

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### Mounting and Operating Instructions EB 9520 EN. Differential Pressure and Flow Meter Media 05

Differential Pressure and Flow Meter Media 05 Fig. 1 Media 05, indicating unit with differential pressure cell, with valve block and pressure gauge (right) Mounting and Operating Instructions EB 9520 EN

### S265 Remote seals with capillary tube. 2600T Series Pressure Transmitters. Engineered solutions for all applications

Contents Data Sheet S265 2600T Series Pressure Transmitters Engineered solutions for all applications Wide range of remote seal types Allows optimum design for each application without compromising performance

### Parametrable / Programmable Pressure Transmitter

Parametrable / Programmable Pressure Transmitter PTM/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available

### Series INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries. Description ALL STAINLESS DESIGN

INTELLIGENT PRESSURE- AND LEVEL TRANSMITTERS For all industries Series 2000 ALL STAINLESS DESIGN EASY CALIBRATION WITHOUT TEST PRESSURE BY 3 PUSH BUTTONS ACCURACY 0,1% 4-20 ma AND HART PROTOCOL ATEX II

### Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases

Glass tube flowmeters Series 2000 Variable area flowmeter for low flows of liquids and gases Short mounting length and compact construction, especially indicated for control panels Easy installation. No

### Electropneumatic Positioner and Pneumatic Positioner Type 3760 JIS

Electropneumatic Positioner and Pneumatic Positioner Type 3760 Application Single-acting positioners for direct attachment to pneumatic control valves. Supplied with an electric input signal of 4 to 20

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 26.06.2014 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### Mounting and Operating Instructions EB EN. Type Supply Pressure Regulator. with increased air capacity

Type 4708-45 Supply Pressure Regulator with increased air capacity Translation of original instructions Mounting and Operating Instructions EB 8546-1 EN Edition March 2016 Note on these mounting and operating

Low Cost Liquid Filled Pressure Gauges Size: 1.5", 2", and 2.5" Accuracy: 2.5% of full scale Materials: Case and ring: 304 stainless steel; Window: Polycarbonate; Seals: Buna-N; Pointer: Black-finished

### High Purity & Ultra High Purity

Solar and Semiconductor High Purity & Ultra High Purity Pressure and Weight Measurement R Electronic Pressure Measurement High Purity & Ultra High Purity - Pressure and Weight Measurement WIKA offers a

### DM01. Battery Powered Precision Digital Gauge. Stainless Steel Sensor. class 0.05

Battery Powered Stainless Steel Sensor class 0.05 Nominal pressure from 0 00 mbar up to 0... 00 bar Special characteristics modular sensor concept data logger graphic display stainless steel housing Ø

### Expert Hydrostatic Level Transmitters

Expert Hydrostatic s General Features MJK Expert hydrostatic level transmitters are designed for level measurement by submerging the transmitter in open channels, drains and tanks. Expert hydrostatic level

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### Programmable pressure and temperature transmitters PTM/RS485

Programmable pressure and temperature transmitters PTM/RS485 Version: 03-28-2012 Technical Specifications Pressure measuring range (bar) 0.1... 0.5 > 0.5 2 > 2 25 Overpressure 3 bar 3 x FS ( 3 bar) 3 x

### Pressure Measurement

Pressure Measurement Manometers Sensors, Transducers Ashish J. Modi Lecturer, Dept. of Mech.Engg., Shri S.V.M. inst. Of Technology, Bharuch Pressure Pressure is a force per unit area exerted by a fluid

### Pressure Switches Application Guideline

Pressure Switches Application Guideline Product Overview The Fema Pressure Switch product portfolio provides devices suitable for many applications. The portfolio contains Special functions and equipment

### Pneumatic service kit Model CPG-KITP

Calibration technology Pneumatic service kit Model CPG-KITP WIKA data sheet CT 93.01 Applications Oil and gas industry Maintenance and service facilities Calibration service companies and service industry

### Air service units series K G1/8, G1/4

Air service units series K Series FK Filter 12.011 Series RK Pressure regulator 12.012 Series FRK Filter/regulator 12.013 Series OK Lubricator 12.014 Series FROK Two piece air service unit 12.015 Series

### Pressure & Vacuum Measurement. Series 370. Solutions. Stabil-Ion Vacuum gauge and Controller

w w w. m k s i n s t. c o m Pressure & Vacuum Measurement Solutions Series 370 Stabil-Ion Vacuum gauge and Controller Features & Benefits All-metal, rack-mount controller for Stabil-Ion and Convectron

### GA24 Technical Datasheet

GA24 Technical Datasheet Variable-area flowmeter Sturdy construction for several applications Local indication without auxiliary power Replaceable mounting parts KROHNE CONTENTS GA24 1 Product features

### Pneumatic dead-weight tester Model CPB3500

Calibration technology Pneumatic dead-weight tester Model CPB3500 WIKA data sheet CT 31.22 Applications Primary standard for calibrating the scale in a pneumatic range from -1... 120 bar / -14.5... 1,600

### Digital pressure gauge Model CPG500

Calibration technology Digital pressure gauge Model CPG500 WIKA data sheet CT 09.01 Applications Calibration service companies and service industry Measurement and control laboratories Quality assurance

### UNIK 5600/5700 Marine Certified Pressure Sensing Platform

GE Measurement & Control UNIK 5600/5700 Marine Certified Pressure Sensing Platform The new UNIK 5600/5700 carries marine certification for most zones on-board ship, as well as Intrinsically Safe certifications.

### Drum-Type Gas Meters Series TG (Wet-Test)

Drum-Type Gas Meters Series TG (Wet-Test) Precision Volumetric Gas Meters for Scientific Laboratories User Benefits Highest accuracy: ±0.5% across full flow rate range Calibration traceable to German National

### Powers Controls RC 195 Single Input Receiver-Controller

Document No. 155-119P25 RC 195-4 Powers Controls RC 195 Single Input Receiver-Controller Description The RC 195 Single lnput Receiver-Controller is a pneumatic instrument which receives one pneumatic input,

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Bypass MFC with capillary technology for nominal flow rates from 5 ml N /min to 15 l N /min Applicable for aggressive gases Compact design and digital communication

### Air service units series Y G1/4 to G1

Air service units series Y Series FY Filter, 12.071 Series RY+ RYP Pressure regulator, 12.076 Series FRY Filter/regulator, 12.080 Series OY Lubricator, 12.083 Series FROY Filter, Regulator, Lubricator

### Level transmitter LT 100

KROHNE 09/2001 7.02363.21.00 GR/PRINTO Level transmitter LT 100 Status: 10/99 Variable area flowmeters Vortex flowmeters Flow controllers Electromagnetic flowmeters Ultrasonic flowmeters Mass flowmeters

### SITRANS P measuring instruments for pressure

Remote seals for transmitters and pressure gauges Technical description Overview In many cases the pressure transmitter and the measured medium have to be physically separated. It is then necessary to

### Product Catalog. Pressure Switches. Pressure Transmitters. Thermostats. Temperature Transmitters. Solenoid Valves TÜV DVGW ATEX.

f Product Catalog Pressure Switches Pressure Transmitters Thermostats Temperature Transmitters Solenoid Valves TÜV DVGW ATEX Zulassung in Vorbereitung Flow Monitors Featuring New products H Electronic

### 9700 Transmitter Submersible hydrostatic level transmitter

Specification sheet IP0078 October 2007 Level 9700 Transmitter Submersible hydrostatic level transmitter Description The 9700 series hydrostatic level transmitter is designed to perform in the most arduous

### Technical Manual. Liquid Level Transmitter CT801-LB/S

CT801-LB/S Technical Manual SAS au Capital de 2 158 244-444 871 933 R.C.S. Bourges - APE : 2651B Headquarter : 9, rue Isaac Newton - 18000 Bourges - France Technical Manual CT801-LB/S 1 st Edition Released

### ETd 04. Battery Powered Precision Digital Gauge. Stainless Steel Sensor. Type: ETd 04. class 0.05

Battery Powered Stainless Steel Sensor class 0.05 Nominal pressure from 0 100 mbar up to 0... 400 bar Special characteristics modular sensor concept data logger graphic display stainless steel housing

### TRZ 03-K Volumeter. Proven Technology. Superior Performance. TRZ 03-K Volumeter is a Turbine Gas Flowmeter for Secondary Gas Metering.

TRZ 0-K Volumeter TRZ 0-K Volumeter is a Turbine Gas Flowmeter for Secondary Gas Metering. TRZ 0-K Turbine Gas Flowmeter continues the tradition of Honeywell s field-proven turbine meter technology, delivering

### Notes on equipment protection per EN/IEC and NEMA For Bourdon tube or diaphragm pressure gauges

Technical information Notes on equipment protection per EN/IEC 60529 and NEMA For Bourdon tube or pressure gauges WIKA data sheet IN 00.18 General information This technical information describes the measures

### Certified according to DIN EN ISO 9001 Technical Datasheet TRICOR Series Mass Flow Meters

www.kem-kueppers.com info@kem-kueppers.com Certified according to DIN EN ISO 9001 Technical Datasheet TRICOR Series Mass Flow Meters Description The Tricor Mass Flow Meters measure simultaneously mass

### Installation, operating and maintenance Instructions for Seemag bypass level indicator

Issue: S Date: 05-09-14 Type G35 General information The Seetru bypass magnetic level indicator, abbreviate SEEMAG, serves to show the filling level of fluids in tanks, basins, tubes etc. The Seemag operates

### 3.0 Pressure Transmitter Selection

3.0 Pressure Transmitter Selection Each Tronic Line pressure transmitter has different features to meet specific performance, environmental, and price requirements. It is not possible to describe every

### MJK Expert 700 / 800 / 900 / 7070 / 7060 MJK Expert 1400 / 3400

MJK Expert 700 / 800 / 900 / 7070 / 7060 MJK Expert 1400 / 3400 Submersible Hydrostatic s Rugged and versatile 2.75 GB 2.75 Expert 090311 We reserve the right to continuously improve our products and make

### Notes on equipment protection to DIN EN and NEMA

Technical information Notes on equipment protection to DIN EN 60529 and NEMA WIKA data sheet IN 00.18 General information This technical information describes the measures to prevent both the formation

### SITRANS. Pressure transmitter SITRANS P, Z series for gauge and absolute pressure. Introduction. Safety instructions 2.

Introduction 1 Safety instructions 2 SITRANS Pressure transmitter SITRANS P, Z series for gauge and absolute pressure 7MF1564 Description 3 Assembly and connection 4 Technical data 5 Dimensional drawings

### Additel 761 Automated Pressure Calibrators Selection Guide

/ Process Calibration Equipment Automated Calibrators Selection Guide Model Features Range Module 1 761-LLP 761-D 761-L 761-LA 761-M 761-MA 761-H 761-HA 761-BP -2.5 to 2.5 mbar (-1 to 1 inh 2 O) -0.95

### Model /2" (63mm), 4" (100mm) and 6" (160mm) Open Front Stainless Steel Gauges

LFG - 8 LIQUI FILL GAUGS OPN FRONT STYL Model 656 (63), 4" (100) and 6" (160) Open Front Stainless Steel Gauges SCRIPTION The 656 Series are suitable for corrosive environments in chemical, petrochemical,

### CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES

SENSORS FOR MASS FLOW AND PRESSURE CORI-FLOW PRECISION MASS FLOW METER / CONTROLLER FOR LIQUIDS AND GASES INTRODUCTION Bronkhorst High-Tech BV is the worldwide pioneer in the field of low flow liquid metering

### KTE6000 / KTU6000 Series OEM pressure transmitters for industrial media

FEATURES 50 mbar to 400 bar, 5 to 6000 psi gage or absolute pressure For many industrial gases and liquids 00, 0545, 05, 6 or 40 ma output Field interchangeable EMC according to EN 66-8 MEDIA COMPATIBILITY

Document No. 129-082 RC 195 Receiver-Controller Product Description The POWERS RC 195 Receiver-Controller is a pneumatic instrument that receives one, two or three pneumatic inputs. It produces a pneumatic

### Volume Booster. Volume Booster Description Supply pressure Pressure range Connection Device Page max. bar bar thread

Volume ooster Supply pressure Pressure range onnection Device Page max. bar bar thread precise differential pressure also 17 0 1 / 10 and G 3 8 R50.02 differential pressure also 10 0.2 1 / 8 ¼ NPT 53.19

### Discontinued. Powers Controls. Technical Instructions Document No P25 RV Rev. 1, May, RV 201 Pressure Reducing Valves.

Powers Controls RV 201 Pressure Reducing Valves Description Features Product Numbers Dual Pressure PRV Technical Instructions Document No. 155-049P25 RV 201-1 Single Pressure PRV The RV 201 Pressure Reducing

### JUMO dtrans p30 Pressure Transmitter

609 Fulda, Germany Postal address: 605 Fulda, Germany Phone: +49 661 600-0 Fax: +49 661 600-607 Phone: +44 179 6 55 Data Sheet 40466 Page 1/6 JUMO dtrans p0 Pressure Transmitter General application Pressure

### H-LEVEL _ H-LEVEL LIQUID LEVEL SYSTEM SETTLEMENT GAUGES

_ LIQUID LEVEL SYSTEM SETTLEMENT GAUGES LIQUID LEVEL SYSTEM The H-Level is an automatic system for long-term monitoring of differential settlements in buildings, tunnels, and other civil structures. The

### Gas Pressure Regulator HON 200

Product information serving the gas industry worldwide Applications, characteristics, technical data Application Gas supply to municipal, industrial and individual consumers Regulator for low-load rails

### Programmable Submersible Level Transmitters

Programmable Submersible Level Transmitters PTM/N/RS485 CUSTOMER BENEFITS High flexibility due to scalalbe pressure range Digital (RS485) and analogue (4-20mA) output signal in one sensor Available as

### Orifice Plate Flow Meter

Orifice Plate Flow Meter FOFT Measuring Principle The principle of measuring the differential pressure is based on the fact that a differential pressure is created across that section of a pipe where its

### Hydraulic pressure balance Model CPB5800

Calibration technology Hydraulic pressure balance Model CPB5800 WIKA data sheet CT 31.11 for further approvals see page 8 Applications Primary standard for calibrating the pressure scale in a hydraulic

### GRUNDFOS alldos product information. Vaccuperm 111 / 113. Vacuum gas dosing systems for chlorine gas, ammonia, sulphur dioxide and carbon dioxide

GRUNDFOS alldos product information Vacuum gas dosing systems for chlorine gas, ammonia, sulphur dioxide and carbon dioxide Construction and function Modular system Vacuum regulator for direct mounting

### Additel 761 Automated Pressure Calibrators

Automated Calibrators UPDATED Fully automated pressure calibrator with built-in pressure generator / controller to as high as 600 psi (40 bar) or as low as 0.01 Pa (0.00004 inh 2 O) accuracy Dual pressure

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### Liquid level instruments BM 24 /BM 51

KROHNE 09/2001 7.02362.21.00 GR/PRINTO Liquid level instruments / Status: 10/99 Variable area flowmeters Vortex flowmeters Flow controllers Electromagnetic flowmeters Ultrasonic flowmeters Mass flowmeters

### Precision Mass Flow Meters / Controllers for Liquids and Gases

CORI-FLOW TM Precision Mass Flow Meters / Controllers for Liquids and Gases Introduction Bronkhorst Cori-Tech B.V. manufactures precise, compact Mass Flow Meters and Controllers for gases and liquids,

### Pressure Measurement Single-range transmitters for general applications

Siemens AG 07 Overview Function U const. p U I EM Sensor Connection for auxiliary power supply Vent pipe with humidity filter Protective conductor connection/ Equipotential bonding Diaphragm The pressure

### KMT. Thermal Mass Flowmeter. for compressed air and gases. Measuring range: Nm³/h Nm³/h

Thermal Mass Flowmeter for compressed air and gases measuring monitoring analysing KMT Measuring range: 0.32... 63 Nm³/h - 263.4... 263350 Nm³/h Accuracy: ± 1.5% of reading + 0.5% of full scale ( DN 50)

### PRODUCT CATALOGUE DIGITAL GAUGES

PRODUCT CATALOGUE DIGITAL GAUGES Pressure AT THE HIGHEST LEVEL Successful medium-sized companies are not successful because they are active in many areas, but rather because they concentrate on one area

### 1/2 to 2-1/2 NPT; 0 to 230 PSI Advantages / Benefits

/-Way Angle-Seat Valve Type 000 to - NPT; 0 to 30 Advantages / Benefits Spring return closed or open, double-acting Materials: bronze, stainless steel High flow rate Self-adjusting packing glands with

### The Discussion of this exercise covers the following points:

Exercise 5-3 Wet Reference Leg EXERCISE OBJECTIVE Learn to measure the level in a vessel using a wet reference leg. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Measuring

### Model 4000 Pressure Controller

FEATURES Multiple Configurations The 4000 series pressure controller can be configured into either proportional only or proportional plus reset mode with a minimum of parts. Rugged Design Die cast aluminum

### Primary standard differential pressure balance Model CPB6000DP

Calibration technology Primary standard differential pressure balance Model CPB6000DP WIKA data sheet CT 32.02 Applications High-level primary standard Test and calibration of differential sensors and

### GasMultiBloc Combined regulating and safety valve Infinitely variable air/gas ratio control mode MB-VEF B01

GasMultiBloc Combined regulating and safety valve Infinitely variable air/gas ratio control mode MB-VEF 407-42 B0 7.27 Flange, line sockets, pulse lines and screw unions are not contained in the scope

### Gas Pressure Regulator HON 300

Product information serving the gas industry worldwide Applications, characteristics, technical data Applications direct acting gas pressure regulator, for systems in accordance with DVGW working instruction

### Botlek Studiegroep Badotherm. Solving measurements issues with Diaphragm seals 3 Case Studies

Botlek Studiegroep 9-4-2015 Badotherm Solving measurements issues with Diaphragm seals 3 Case Studies Prepared: 10/04/2015 Badotherm Company Summary Badotherm introduction Family owned company Founded

### P3000 Series. Pneumatic Deadweight Testers Model P3000. Technical Data. Features

P3000 Series Pneumatic Deadweight Testers P3000 Technical Data Features Pressure ranges from Vacuum through 2 000 psi (140 bar) Accuracy better than 0.015 % of reading. (Increased accuracy option of 0.008

### STANDARD SPECIFICATION FOR LEVEL INSTRUMENTS DOCUMENT NO : 44-LK /J.02/0007-A4

Page 1 of 8 STANDARD SPECIFICATION FOR LEVEL INSTRUMENTS DOCUMENT NO : 44-LK-5100-00/J.02/0007-A4 Rev No. Issue Date Pages Rev Description Prepared A 12.08.08 8 Issued for Comments / Approval By Checked

### LEVEL Transmitters. ) of the liquid column above it and the atmospheric pressure (P atm

Hydrostatic level transmitters nivopress N GENERAL DESCRITION The NIVORESS N hydrostatic level transmitters are designed to measure the level of clean or contaminated liquids. The pressure sensor at the

### MASS-STREAM TM Instruction Manual. D-6200 Analog Mass Flow Meters / Controllers

MASS-STREAM TM Instruction Manual D-6200 Analog Mass Flow Meters / Controllers Doc. No.: 9.17.112A Date: 17-03-2017 ATTENTION Before installing and operating the instrument it is strongly recommended that

### Air Operated Hydraulic Pumping Systems to 50,000 psi

High Pressure Equipment Air Operated Hydraulic Pumping Systems to 50,000 psi PS-10: 10,000 psi PS-20: 20,000 psi PS-30: 30,000 psi PS-40: 40,000 psi PS-50: 50,000 psi PS-90: 90,000 psi High Pressure air

### Pilot valves for pilot operated servo valves

Data sheet Pilot valves for pilot operated servo valves The range of pilot valves consists of: Constant-pressure pilot valve, type CVP (LP) and CVP (HP) Differential-pressure pilot valve, type CVPP (LP)

### Heading 10 NEW. Calibration Technology Pressure. ARMATURENBAU GmbH MANOTHERM Beierfeld GmbH

Heading 10 Calibration Technology Pressure NEW 4/17 ARMATURENBAU GmbH MANOTHERM Beierfeld GmbH 10 000 ARMATURENBAU GmbH & MANOTHERM Beierfeld GmbH What is barotec Kalibriertechnik? The instruments of barotec

### SCH. Mechanical Pressure Switches. for overpressure, vacuum pressure and differential pressure

Mechanical Pressure Switches for overpressure, vacuum pressure and differential pressure measuring monitoring analysing SCH P3 OO Switching range: -1... + 0.1 bar - 250... +100 mbar, 1... 16 mbar a 16...

### Types ACE95 and ACE95Sr Tank Blanketing Valves

Types ACE95 and ACE95Sr Tank Blanketing Valves Bulletin 74.1:ACE95 May 2010 W9133 W8155 Figure 1. Type ACE95 Tank Blanketing Valve Figure 2. Type ACE95Sr Tank Blanketing Valve Features Fully Balanced Pilot

### GM Series Dual-Block Multi-Function Gas Control Valves

Installation Sheets Manual 121 Gas Combustion Combination Controls and Systems Section G Technical Bulletin GM Issue Date 0297 GM Series Dual-Block Multi-Function Gas Control Valves Figure 1: GM Series

### Mounting and Operating Instructions EB 2172 EN. Series 43 Temperature Regulators. Type 43-5 and Type Type Type 43-6

Series 43 Temperature Regulators Type 43-5 and Type 43-7 Type 43-6 Type 43-6 Type 43-7 with flanged valve body, DN 32 to 50 Type 43-5 Type 43-7 with welding ends Mounting and Operating Instructions EB

### Climatic Independent Level Sensor LAR

3 JAHRE GEWÄHRLEISTUNG SENSORS FOR FOOD AND BIOPHARMA. Product Information LAR-361 LAR-761 FOOD Climatic Independent Level Sensor LAR Application / Specified usage Hydrostatic level measurement in humid

### MULTIFUNCTIONAL CONTROL FOR GAS BURNING APPLIANCE

SIT Group SIT -3- SIGMA MULTIFUNCTIONAL CONTROL FOR GAS BURNING APPLIANCE 9.955. Subject to change without notice Application Domestic gas appliances: central heating boilers, combi boilers, instantaneous

### L3 Pressure and Level transmitter

SENSORS FOR AND BIOPHARMA. Product Information L3 L3 Pressure and Level transmitter Range of applications Hydrostatic Level measurement in dynamic temperature environments Pressure measurement in pipes

### sitrans p Operating Instructions (de/en) Edition 03/2004

(de/en) Edition 03/2004 sitrans p Messumformer für Food, Pharma und Biotechnik Transmitters for food, pharmaceuticals and biotechnologgy 7FM8010 Contents Information for the operator...27 Important security

### Hand lever valves VHER

Hand lever valves VHER Hand lever valves VHER Key features Powerful Flexible Practical -M- Flow 170 3800 l/min 4/3-way valve mid-position closed mid-position exhausted mid-position pressurised Connections:

### Pressure gauges Product range overview

Pressure gauges Pressure gauges Product range overview Type Pneumatic connection Nominal size Display unit Red/ Outer scale Inner scale green bar MPa psi bar MPa psi range Page/ Internet Pressure gauge

### AMT-Ex Dewpoint Transmitter

AMT-Ex Dewpoint Transmitter Instruction Manual Alpha Moisture Systems Alpha House 96 City Road Bradford BD8 8ES England Tel: +44 1274 733100 Fax: +44 1274 733200 email: mail@amsytems.co.uk web: www.amsystems.co.uk