# No E. General Information. Averaging Pitot Tube Flow Sensor HITROL CO., LTD.

Save this PDF as:

Size: px
Start display at page:

Download "No E. General Information. Averaging Pitot Tube Flow Sensor HITROL CO., LTD."

## Transcription

1 No E Averaging Pitot Tube Flow Sensor HITROL CO., LTD.

2 M e a s u r e m e n t Pr i n c i p l e Measurement principle of Hitbar sensor is explained below. When a Hitbar is installed in the pipeline through which the fluid is Ptot=Pstat+Pdyn P=Ptot-Pstat flowing, the flow velocity at the front side of Hitbar becomes zero for a moment, the pressure at this location is called as a stagnation pressure, and the diff e r e n c e between the stagnation pressure and the static pressure inside the pipeline is termed V= 2p/ as a dynamic pressure. This dynamic pressure has a certain relationship with the flow velocity in the pipeline, and therefore, once stagnation pressure and static pressure are measured, then the dynamic pressure can be obtained. Design of Hitbar S e n s o r There is almost no vibration since the configuration of cone-type Hitbar sensor is fluid-mechanically designed so that the fluid drag force becomes the minimum, and the constant pressure distribution is generated. Hitbar flow sensor consists of one body, and averaged dynamic pressure over the entire cross section of the pipeline is transferred through two sensing ports and pressure measuring tubes. The pressure difference between stagnation pressure and static pressure, so called dynamic pressure is proportional to the square of flow rate according to Bernoulli equation. Maximum pressure difference and flow ranges(hapt-1f) Pipe Sensor Size : 1 Size P GPM SCFM SCFM : Air Psia, 60F GPM : Water 60F, SG=1.0 P : Inch of water column Maximum pressure difference and flow ranges(hapt-2f) Pipe Size Sensor Size 1 1 1/2 DP GPM SCFM DP GPM SCFM DP GPM SCFM SCFM : Air Psia, and 60F GPM : Water 60F, SG=1.0 DP : Inch of water column 2

3 Measuring fluid M a t e r i a l Pipe size Flow ranges Pressure difference ranges Operating temperature Operating pressure A c c u r a c y R e p r o d u c i b i l i t y Turndown ratio liquid, gas, steam and etc. 304 SS option : 316 SS, 316L SS, Monel, and etc. from 50 mm to 1800 mm(2to 72) refer above table refer above table 300 option : maximum kgf/cm 2 Option : 150 kgf/cm % FS 0.1 % FS 10:1(extensible according to request) S p e c i f i c a t i o n s It is easy to install since it is inserted into the pipeline. Welding of mounting hardware is small compared with the other differential pressure flow meter. Operating energy cost is reduced since it has a less permanent pressure loss compared with other differential pressure flow meters such as an orifice plate. It has a very high measurement accuracy since the discharge coefficient is obtained using HITROL s flow standard facilities which is traceable to and a part of Korean National Calibration System. It has a high measurement accuracy since it measures the averaging velocity in the pipe. A d v a n t a g e s When measuring the incompressible fluids H A P T-1(2)F flow sensor 3-way or 5-way valve d i fferential pressure transmitter flow computer or flow indicator When measuring the compressible fluids H A P T-1(2)F flow sensor 3-way or 5-way valve d i fferential pressure transmitter flow computer or flow indicator pressure transmitter temperature transmitter(pt 100 ) S y s t e m A r r a n g e m e n t 3

4 Assembly Pa r t s Schematic diagram HAPT Instrument Valve Description Material Ball valve 1/2 NPT CS max. 85 bar at bar at max SS Ball valve 1/4 NPT CS max. 85 bar at bar at max SS Shut-off valve 1/4 NPT CS max. 85 bar at bar at max SS Shut-off valve 1/2 NPT CS max. 400 bar at bar at max SS HAPT Mounting Hardware HAPT-1F HAPT-1F Without assembly parts MH 01 Without assembly parts MH 11 With assembly parts MH 02 With assembly parts MH 12 With assembly parts MH 03 With assembly parts MH 13 material 316 material 316 Flange DN 25 Flange DN 32 HAPT-2F Without assembly parts MH 21 With assembly parts MH 22 With assembly parts MH 23 material 316 HAPT-1F HAPT-2F Without assembly parts MH 31 Without assembly parts MH 41 With assembly parts MH 32 With assembly parts MH 42 With assembly parts MH 33 With assembly parts MH 43 material 316 material 316 Flange DN 65 Flange DN 32 Flange DN 65 4

5 Upstream dimension(a) Downstream dimension(b) Length of Straight Pi p e 5

6 I n s t a l l a t i o n D r a w i n g Steam Liquid Liquid and steam Horizontal piping Fill tee 5 Valve m a n i f o l d 45 Preferred Recommended orientation DP transmitter Liquid and steam Liquid Steam Vertical piping Fill Cross Orientation can be anywhere around pipe 5 Valve manifold DP transmitter Gases Gases Horizontal piping Recommended orientation DP transmitter 5 Valve m a n i f o l d Vertical piping Orientation can be anywhere around pipe 6

7 Hitbar can be designed and the differential pressure can be calculated only when the flow parameters shown below should be known Flow parameter Tag No. Max. Flow Rate(m 3 /h) Min. Flow Rate(m 3 /h) Max. Temperature() Max. Pressure(psia) Pipe Inside Diameter(mm) Density(kg/m 3 ) Viscosity(cP) When the fluid is incompressible When the fluid is compressible or steam Tag No. Flow parameter Max. Flow Rate(m 3 /h) Min. Flow Rate(m 3 /h) Max. Temperature() Max. Pressure(psia) Pipe Inside Diameter(mm) Density(kg/m 3 ) Viscosity(cP) 7

8 ORDER CODES 1F : Single Support 2F : Double Support Process Connection Size(Flange) A(Rating) : 1 ANSI(150#, 300#, 600#, 900#) B(Rating) : 1 1/2 ANSI(150#, 300#, 600#, 900#) C(Rating) : 2 ANSI(150#, 300#, 600#, 900#) D(Rating) : 3 ANSI(150#, 300#, 600#, 900#) E(Rating) : 4 ANSI(150#, 300#, 600#, 900#) O(Rating) : Others Sensor Size A : 1 B : 1 1/2 C : 2 O : Others Process Pipe Size 02 : 2 03 : 3 04 : 4 06 : 6 72 : : Others HAPT-1(2)F - Sensor Materials A : STS 304 B : STS 316 C : STS 316L D : Monel E : Inconel O : Others Valve Mat l 1 : C.S 2 : 304 SS 3 : 316 SS 4 : 316L SS 5 : Monel Valve Type A : Ball Valve B : Shut-off Valve HITROL CO., LT D. HEAD OFFICE F A C T O R Y R & D INSTITUTE BONGILCHEON-RI CHORI-MYUN PAJU CITY KYUNGGI-DO KOREA TEL : (031) ~7 (02) FAX : (031) ,

### The Discussion of this exercise covers the following points:

Exercise 3-2 Orifice Plates EXERCISE OBJECTIVE In this exercise, you will study how differential pressure flowmeters operate. You will describe the relationship between the flow rate and the pressure drop

### Vortex Flow Meter Wafer or Flange Connection. - Steam - Liquid - Gas

Vortex Flow Meter Wafer or Flange Connection - Steam - Liquid - Gas Working Principle & Circuit Diagram Working Principle When a column body placed in flowing fluids in pipe, a series of vortices will

### For Multi-Parameter Meters see mvx

Design Features VORTEX IN-LINE FLOW METERS For Multi-Parameter Meters see m Principles of Operation n No moving parts to wear or fail. n Electronics can be remotely mounted up to 30.5 m (0 ft). n No fluid

### For Multi-Parameter Meters see mvx

VORTEX IN-LINE FLOW METERS Design Features For Multi-Parameter Meters see mvx Principles of Operation VX No moving parts to wear or fail. Electronics can be remotely mounted up to 30.5 m (0 ft). No fluid

### Over 20,000 Strain Gage Target flowmeters installed since 1952.

Over 20,000 Strain Gage Target flowmeters installed since 1952. Liquid, gases, superheated and saturated steam from -320 to 500 O F and up to 15,000 psi. No frictional moving parts. Designed to withstand

### SINGLE VALVE WITH LOW-FLOW BYPASS

CONTROL VALVES Pressure Reducing Valve Sizing Guide Sizing pilot operated reducing valves is not a complicated process. It starts with determining requirements and following these guidelines in valve size

### The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions

The M-Series Eletta Flow Meter High accuracy DP Flow Meter with multiple functions Flow Meter with multiple functions for gases and liquids M3 The M-series Flow Meter, with its versatile and user-friendly

### Pressure Regulators Pressure regulation to maximize steam system efficiency

Pressure Regulators Pressure regulation to maximize steam system efficiency The density of steam increases as the pressure rises. Because of this characteristic, more pounds of steam can be stored in a

### BULLETIN FOR ACCURATE INDICATION AND CONTROL OF GAS AND LIQUID FLOW IN LOW RANGES.

BULLETIN 20-7000 ROTAMETERS PURGE FOR ACCURATE INDICATION AND CONTROL OF GAS AND LIQUID FLOW IN LOW RANGES. INTRODUCTION McCrometer purge rotameters are precision-made instruments designed to accurately

### PanaFlow MV82. Insertion Style Multivariable Flowmeter. GE Measurement & Control. Key Benefits. Applications

GE Measurement & Control PanaFlow MV82 Insertion Style Multivariable Flowmeter Key Benefits Multivariable vortex flowmeter for measuring volumetric flow, temperature, pressure, density, and mass flow using

### NORDCALC Introduction... 2 Registration... 2 Flow Calculations tab Torque Calculation & Actuator Mounting Data tab... 21

NORDCALC User Guide Table of Contents NORDCALC Introduction... 2 Registration... 2 Flow Calculations tab... 5 Calculating the gas flow rate for gaseous medium... 6 Calculating the pressure drop for gaseous

### Wedge Type Flow Meter Installation and Operation Manual

Wedge Type Flow Meter Installation and Operation Manual 358-EN Please read and save these instructions. Wedge Type Flow Meter PLEASE READ THIS FIRST! The purpose of this document is to provide guidance

### Gas Pressure Regulator HON 200

Product information serving the gas industry worldwide Applications, characteristics, technical data Application Gas supply to municipal, industrial and individual consumers Regulator for low-load rails

### Turbine Flowmeters for Gas Applications

Turbine Flowmeters for Gas Applications Description Flow Technology s FT Series turbine flowmeters utilize a proven flow measurement technology to provide exceptionally reliable digital outputs. Because

### Orifice Plate Flow Meter

Orifice Plate Flow Meter FOFT Measuring Principle The principle of measuring the differential pressure is based on the fact that a differential pressure is created across that section of a pipe where its

### THERMAL MASS FLOW MEASUREMENT FOR GASES

THERMAL MASS MEASUREMENT FOR GASES Make the Wise Choice. Choose Sage Flow Meters. A Commitment to Higher Performance SAGE METERING is a manufacturer of high performance Thermal Mass Flow Meters which measure

### STANDARD SPECIFICATION FOR FLOW INSTRUMENTS DOCUMENT NO : 44-LK /J.02/0006-A4

Page 1 of 9 ` STANDARD SPECIFICATION FOR FLOW INSTRUMENTS DOCUMENT NO : 44-LK-5102-00/J.02/0006-A4 Rev No. Issue Date Pages Rev Description Prepared A 12.08.08 9 Issued for Comments / Approval By Checked

### Pressure and Temperature Controls

Pressure and Temperature s Pressure Reducing s pressure reducing valves (PRVs) and temperature regulators help you manage steam, air and liquid systems safely and efficiently. And assure uninterrupted

### Ratio Regulator Zero Governor. FRG/6 Series

Ratio Regulator Zero Governor FRG/6 Series CSA Certified ANSI Z21.18 / CSA 6.3 CSA Requirement # 5.01 Zero Governor and Gas/Air Regulator File # 157406 US and Canadian Models FRG 705/6 FRG 707/6 FRG 710/6

### Pressure and/or Temperature Pilot Operated Steam Regulators Series 2000

Hoffman Specialty Regulators Regulators Pressure and/or Temperature Operated Regulators Series 2000 The Hoffman Specialty Series 2000 consists of main valves, pilot valves, wells and hardware kits. They

### Mass Flow Controller (MFC) for Gases

Mass Flow Controller (MFC) for Gases Bypass MFC with capillary technology for nominal flow rates from 5 ml N /min to 15 l N /min Applicable for aggressive gases Compact design and digital communication

### Liquid Level Measurement

Technical Article Liquid Level Measurement A pressure transmitter can be used to determine the liquid level in a tank, well, river or other body of liquid. The pressure at the bottom of a liquid filled

### Coriolis Mass Flow Meter

Coriolis Mass Flow Meter TMFW Series Mini Type Coriolis Mass Flow Meter Xi an Tosilon Automation Co., Ltd No.299, Daqing Rd, Lianhu District, Xi'an Shaanxi, China Tel: +86-29-8823 8550 info@tosilon.com;

### Product Overview A DIVISION OF MARSH BELLOFRAM

Product Overview P627 & P630 P627 Available with internal relief device - R627 version P627 & P630 - High Flow Regulators The P627 and P630 are spring loaded, direct-operated regulators for both low and

### Variable Area Flow Meters

Variable Area Flow Meters Over sixty years of variable area meter expertise Brooks started designing and manufacturing variable area (VA) meters, also referred to as rotameters, in 1946. Now, with more

### Experiment 8: Minor Losses

Experiment 8: Minor Losses Purpose: To determine the loss factors for flow through a range of pipe fittings including bends, a contraction, an enlargement and a gate-valve. Introduction: Energy losses

### SERIES 10. Pilot Operated Tank Blanketing Valve, 2" PROTECTOSEAL.

SERIES 10 PROTECTOSEAL 2" NPT inlet and outlet standard Pilot operated design provides tight operating band Optional inlet and outlet connections available Inlet gas pressures from 40 PSIG to 130 PSIG

### CVEN 311 Fluid Dynamics Fall Semester 2011 Dr. Kelly Brumbelow, Texas A&M University. Final Exam

CVEN 311 Fluid Dynamics Fall Semester 2011 Dr. Kelly Brumbelow, Texas A&M University Final Exam 8 pages, front & back, not including reference sheets; 21 questions An excerpt from the NCEES Fundamentals

### The Discussion of this exercise covers the following points:

Exercise 5-3 Wet Reference Leg EXERCISE OBJECTIVE Learn to measure the level in a vessel using a wet reference leg. DISCUSSION OUTLINE The Discussion of this exercise covers the following points: Measuring

### L 100. Bubble-Tube Level System. Installation, Operation and Maintenance Instructions

L 100 Bubble-Tube Level System Installation, Operation and Maintenance Instructions Figure 1 Contents Section Description Page 1.0 Introduction 2 2.0 Specifications 3 3.0 Installation 3 4.0 Warranty 6

November 1993 Prepared By: Ken A. Steward. P.E. Linco-Electromatic, Inc. 4580 West Wall Street Midland, Texas 78703 LPG DRIVER ATTENDED TRANSPORT LOADING INTRODUCTION The safest and most accurate method

### Porter Lab & Analytical Products

Porter Lab & Analytical Products Porter Lab & Analytical Products Engineered to Excel in Critical Applications The Porter Instrument Division of Parker Hannifin Corporation is a leading manufacturer of

### Type 299 Pressure Reducing Regulators

Type 299 Pressure Reducing Regulators Fisher, Fisher-Rosemount, and Managing The Process Better are marks owned by Fisher Controls International, Inc. or Fisher-Rosemount Systems, Inc. All other marks

### International Journal of Technical Research and Applications e-issn: , Volume 4, Issue 3 (May-June, 2016), PP.

DESIGN AND ANALYSIS OF FEED CHECK VALVE AS CONTROL VALVE USING CFD SOFTWARE R.Nikhil M.Tech Student Industrial & Production Engineering National Institute of Engineering Mysuru, Karnataka, India -570008

### Development of Gasdynamic Probe for Total Temperature Measurement in gases

ASET 20- National Conference on Emerging Trends in Propulsion Technology Development of Gasdynamic Probe for Total Temperature Measurement in gases K Sathiyamoorthy, Souren Misra 2, Srinivas J, Baskaran

### Fisher DSA Steam-Atomized Desuperheater

Instruction Manual DSA Desuperheater Fisher DSA Steam-Atomized Desuperheater Contents Introduction... 1 Scope of Manual... 1 Description... 1 Principle of Operation... 2 Installation... 2 Operating Instructions...

### Averaging Pitot Tubes

Data Sheet DS/TORBAR-EN Issue 1 Averaging Pitot Tubes Torbar Unique profile shape offers high flow turndown No drift in co-efficient ensures long term stability One-piece outer tube for pipes up to 5000mm

### Fisher DVI Desuperheater Venturi Inline

Instruction Manual DVI Desuperheater Fisher DVI Desuperheater Venturi Inline Contents Introduction... 1 Scope of Manual... 1 Description... 1 Principle of Operation... 2 Installation... 3 Operating Instructions...

### North American Orifice Metering System

Combustion North American Orifice Metering System 8697 and M8697 Metering Orifices accurately measure gas flows to industrial burners. They are compact, inexpensive, and can be installed in individual

### Aspects of Flow Control in Metering Henry Gomes SGS Kuwait WLL

Aspects of Flow Control in Metering Henry Gomes SGS Kuwait WLL HENRY GOMES Metering Consultant SGS Kuwait Born on Aruba (Carribean) in 1963 School and College in the Netherlands Resident in Romania - Bucharest

### Un-Pressurized Orefice Fittings FIO EZ. Parts List and Operation Instructions TECHNICAL MANUAL. Dn 2-6 Class Lbs

Un-Pressurized Orefice Fittings FIO EZ Parts List and Operation Instructions TECHNICAL MANUAL Dn 2-6 Class 150-600 Lbs US US 2 FIO EZ - MT 108-US - 05-2016 FIO EZ Important Instructions US Pietro Fiorentini

### TWO PHASE FLOW METER UTILIZING A SLOTTED PLATE. Acadiana Flow Measurement Society

TWO PHASE FLOW METER UTILIZING A SLOTTED PLATE Acadiana Flow Measurement Society Gerald L. Morrison Presented by: Mechanical Engineering Department Daniel J. Rudroff 323 Texas A&M University Flowline Meters

### 94270 Vapor Guard Tank Blanketing Valve Vapor Guard Tank Blanketing Valve What is Tank Blanketing? Features How does it work?

94270 What is Tank Blanketing? blanketing systems are used to prevent the escape of liquid vapors into the atmosphere or to prevent moisture from entering a tank and contaminating its contents. A tank

### CONTROL VALVE TESTING

The optimal functioning of the Control valve not only exists of sufficient body & seat tightness, but more important, the total "performance" of the valve and its controls! For an accurate and reliable

### Self-operated Pressure Regulators Type 2405 Pressure Reducing Valve

Self-operated Pressure Regulators Type 2405 Pressure Reducing Valve ANSI version Application Pressure reducing valve for set points from 0.075 to 150 psi (5 mbar to 10 bar) Valve size NPS ½ to 2 1) (DN

### Model 601CV. Mass Flow Controller. Product Features:

Model 601CV Mass Flow Controller Porter s Model 601CV Series II Mass Flow Controller (MFCs) is designed for precise control of virtually all conventional process gases. The MFC consists of a thermal mass

### CA10S, CA14 and CA14S Air and Gas Traps

1448150/9 IM-P148-13 ST Issue 9 CA10S, CA14 and CA14S Air and Gas Traps Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation 4. Commissioning 5.

### Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters

ABB MEASUREMENT & ANALYTICS WHITE PAPER Pressure measurement Applying remote seal, isolation diaphragms between process fluids and transmitters Remote seals protect transmitters when measuring difficult

### The Discussion of this exercise covers the following points: Range with an elevated or suppressed zero Suppressed-zero range Elevated-zero range

Exercise 4-3 Zero Suppression and Zero Elevation EXERCISE OBJECTIVE In this exercise, you will learn the effect that mounting a pressure transmitter above or below the reference level has on the hydrostatic

### PTF4 Pivotrol Pump (patented) version Dual Mechanism - Pressure Powered Pump

Local regulations may restrict the use of this product to below the conditions quoted. In the interests of development and improvement of the product, we reserve the right to change the specification without

### Gas Measurement Fundamentals Certification. Curriculum

Gas Certification Institute, LLC P.O. Box 131525 Houston, Texas 77219-1525 281-598-7200 Phone 281-598-7199 Fax contact@gascertification.com www.gascertification.com Curriculum Copyright 2007-2012 Gas Certification

### MANUAL FLOW CONTROL 2-1/2" - 18" QUICKSET FLOW CHART

This chart performs two functions: It allows you to calculate the permanent pressure drop for any size Quickset or. This figure is useful mostly in the design process, for system calculations. It allows

### High-Purity. Ball Valves

High-Purity s Components MANUFACTURING ASSEMBLY TESTING HP High-Purity s CFOS Applications Cost-effective systems where cleaning for oxygen-grade services are required, such as: Clean-Dry Air (CDA), Oil-Free

### Flow Measurement SITRANS F C

Overview The flow measuring principle is based on the Coriolis Effect. The FCS400 sensor s measuring tubes are energized by an electromechanical driver circuit which oscillates them at their resonance

### Purgemaster Series 10A6100 SNAP-IN. Instruction Bulletin Replaceable Parts List. D184B003U35 Rev. 01 /

SNAP-IN Purgemaster Series 0A600 Instruction Bulletin Replaceable Parts List D84B003U35 Rev. 0 / 03.200 You have purchased a high quality, modern instrument from ABB Automation Products. We thank you for

### 44 (0) E:

FluidFlow Relief Valve Sizing Handbook Flite Software 2016 Flite Software N.I. Ltd, Block E, Balliniska Business Park, Springtown Rd, Derry, BT48 0LY, N. Ireland. T: 44 (0) 2871 279227 E: sales@fluidflowinfo.com

### Pumping Systems for Landscaping Pumps, Controls and Accessories. Mark Snyder, PE

October 21, 2010 Pumping Systems for Landscaping Pumps, Controls and Accessories Mark Snyder, PE Pump Station Design Purpose of Pump Stations Pump stations are designed to boost water pressure from a lower

### M111 Spiraflo Transducer

3309551/ 7 IM-P330-38 MI Issue 7 M111 Spiraflo Transducer Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Technical data 4. Mechanical installation: M111

### Turbine Flowmeters for Gas Applications (Metric)

Turbine Flowmeters for Gas pplications (Metric) Description Flow Technology s FT Series turbine flowmeters utilize a proven flow measurement technology to provide exceptionally reliable digital outputs.

### How to use the PBM Flow Calculator Spreadsheet (V2.04)

How to use the PBM Flow Calculator Spreadsheet (V2.04) o Navigate to the F:\Common\ENGINEERING\Common Design Calculations folder o o Open PBM FLOW CALC 2.04.xlsm On the title page, the version, version

### Modeling a Pressure Safety Valve

Modeling a Pressure Safety Valve Pressure Safety Valves (PSV), Pressure Relief Valves (PRV), and other pressure relieving devices offer protection against overpressure in many types of hydraulic systems.

### 776 Cryogenic Safety Valve

776 Cryogenic Safety Valve INTRODUCTION 776 Cryogenic Safety Valve The effects of exceeding safe pressure levels in an unprotected pressure vessel or system, can have catastrophic effects on both plant

### Mounting and Operating Instructions EB 3007 EN. Self-operated Pressure Regulators. Differential Pressure Regulators (opening) Type Type 42-25

Self-operated Pressure Regulators Differential Pressure Regulators (opening) Type 42-20 Type 42-25 Type 42-20 Differential Pressure Regulator Type 42-25 Differential Pressure Regulator Mounting and Operating

### COMBINATION AIR RELEASE DEGASSING (CARD) VALVES INSTALLATION AND MAINTENANCE MANUAL

COMBINATION AIR RELEASE DEGASSING (CARD) VALVES INSTALLATION AND MAINTENANCE MANUAL SPECIFICATIONS: The CARD series air valves are available in 3 pipe sizes, 1, 2 and 4 NPT or socket. Maximum inlet pressure

### Pressure reducing valves Index

Index General information page Introduction 506 General introduction 507 for a steam plant 509 Product information BSP thread page Brass 510 ; BSP female thread 511 ; BSP male thread 513 Stainless steel

### RD(H)20/25 Pressure-Reducing Regulator User Manual

RD(H)20/25 Pressure-Reducing Regulator User Manual Read the complete manual before installing and using the regulator. 2 Safe Product Selection When selecting a product, the total system design must be

### Model 1159 Remote Diaphragm Seals Manual

00809-0100-4504 English June 1999 Rev. AB Model 1159 Remote Diaphragm Seals Manual With Model 1152, Model 1153 Series B, D, and F, and Model 1154 Alphaline Nuclear Pressure Transmitters Product Discontinued

### Standard Test Method for Measuring Insertion Loss of Pneumatic Exhaust Silencers 1

Designation: E 1265 90 (Reapproved 1995) e1 An American National Standard Standard Test Method for Measuring Insertion Loss of Pneumatic Exhaust Silencers 1 This standard is issued under the fixed designation

### Liquid level measurement using hydrostatic pressure and buoyancy

iquid level measurement using hydrostatic pressure and buoyancy This worksheet and all related files are licensed under the Creative Commons Attribution icense, version 1.0. To view a copy of this license,

### PENBERTHY SERIES LM, ELL, FL, GL, GH, U, L AND 2NC FOR PUMPING GASES

Practical, simple and cost-effective alternatives for process industries to purge gases from chambers, exhaust, evacuate or prime using liquid, steam or air operating media FEATURES Simple design with

### INSTALLATION, OPERATION

IOM RE1 110622 en INSTALLATION, OPERATION and MAINTENANCE MANUAL Direct Acting Pressure Reducing Valve Model: RE 1 MIYAWAKI INC. Osaka, Japan The MIYAWAKI-Pressure Reducing Valve RE 1 is a direct acting

### 3.0 Pressure Transmitter Selection

3.0 Pressure Transmitter Selection Each Tronic Line pressure transmitter has different features to meet specific performance, environmental, and price requirements. It is not possible to describe every

Document No. 129-082 RC 195 Receiver-Controller Product Description The POWERS RC 195 Receiver-Controller is a pneumatic instrument that receives one, two or three pneumatic inputs. It produces a pneumatic

### Series MG Mixing Tubes

Series MG Mixing Tubes Fuel gas / air mixers 30-60.8-1 Air/fuel gas mixing devices for use with all MAXON premix burner systems to provide thorough blending of air/gas mixture. Mixing tubes available for

### 3-way control valves VK 3

Data sheet 3-way control valves VK 3 Description / Application VK 3 valves are a range of 3 port high specification flanged valves for chilled water, LPHW, MPHW, HPHW (low, medium or high pressure hot

### CONTROL VALVE WHAT YOU NEED TO LEARN?

CONTROL VALVE WHAT YOU NEED TO LEARN? i) The control valve characteristics refers to the relationship between the volumetric flowrate F (Y-axis) through the valve AND the valve travel or opening position

### Write important assumptions used in derivation of Bernoulli s equation. Apart from an airplane wing, give an example based on Bernoulli s principle

HW#3 Sum07 #1. Answer in 4 to 5 lines in the space provided for each question: (a) A tank partially filled with water has a balloon well below the free surface and anchored to the bottom by a string. The

### DTG - LCD Digital Temperature Gauge USER MANUAL

DTG - LCD USER MANUAL Page 1 of 7 USERS GUIDE page 1. Description 3 2. Function 3 3. Safety Instruction 3 3.1. Safety Conventions 3 3.2. Proper Use 3 4. Installation 4 4.1. Unpacking 4 4.2. Storage 4 4.3.

### Rosemount Integral Orifice Flow Meter Series. Reference Manual , Rev KA December 2017

Rosemount Integral Orifice Flow Meter Series Reference Manual Reference Manual Contents Contents 1Section 1: Introduction 1.1 Using this manual...................................................... 1

### Mounting and Operating Instructions EB 9520 EN. Differential Pressure and Flow Meter Media 05

Differential Pressure and Flow Meter Media 05 Fig. 1 Media 05, indicating unit with differential pressure cell, with valve block and pressure gauge (right) Mounting and Operating Instructions EB 9520 EN

### BERMAD Waterworks. Level Control Valve with Altitude Pilot. 700 Series. Model X. Features and Benefits. Major Additional Features

Level Control Valve with Altitude Pilot High level reservoirs & water towers Energy cost critical systems Systems with poor water quality Inherent refreshing Level sustaining at reservoir outlet The Level

### 299H Series Pressure Reducing Regulators

September 12 299H Series Pressure Reducing Regulators Inlet Pressure up to psig / 1 bar Compact ±1% Accuracy for Fixed Factor Billing (PFM) Rugged Construction Integral Pilot Easy to Maintain W713 Figure

### S265 Remote seals with capillary tube. 2600T Series Pressure Transmitters. Engineered solutions for all applications

Contents Data Sheet S265 2600T Series Pressure Transmitters Engineered solutions for all applications Wide range of remote seal types Allows optimum design for each application without compromising performance

### INSTRUCTION MANUAL. process. Ultrasonic Flowmeter Range

INSTRUCTION MANUAL process TM Ultrasonic Flowmeter Range process TM CONTENTS Page No 1 General 4 2 Order Codes 4 3 Installation 5 & 6 4 Specifications 7 & 8 1 General 3 Installation The Process Atrato

### General Specifications

General Specifications GS 34P02P32-01E Gas Flow Calculation Portfolio (FCN-500) GENERAL This general specification document describes the Gas Flow Calculation Portfolio for FCN-500 on FCN autonomous controllers.

### MAINTENANCE AND OPERATING MANUAL

MAINTENANCE AND OPERATING MANUAL Cyclone condensate separator Type ASA EN EN Maintenance and operating manual Purpose and appropriate use Dear Customer, thank you for choosing our product. In order to

### 1/2 to 2-1/2 NPT; 0 to 230 PSI Advantages / Benefits

/-Way Angle-Seat Valve Type 000 to - NPT; 0 to 30 Advantages / Benefits Spring return closed or open, double-acting Materials: bronze, stainless steel High flow rate Self-adjusting packing glands with

### SB AXIAL FLOW VALVES

SB9509.3 AXIAL FLOW VALVES The improved technology for pressure regulation The American Axial Flow Valve provides pressure and flow control in high capacity pipelines. It can be used for pressure regulation,

### Reduce pressure zone device suitable for high and medium hazard rated applications BSP screwed connections

Reduce pressure zone device suitable for high and medium hazard rated applications BSP screwed connections Features General application The RP03 provides protection from both backsiphonage and backpressure

### SABERINDO PACIF SABERINDO PACIFIC CIFIC SABERINDO PA. A Tyco International Company

CIF A Tyco International Company 1 Foam Concentrate CIF 3% AFFF -UL Listed -UL Canada Listed 6% AFFF 6 parts AFFF concentrate to 94 parts water -UL Listed- Foam Liquid -UL Canada Listed 3% FLUOROPROTEIN

### HON C210 Series Control Valves. High Pressure Control valves

HON C210 Series Control Valves High Pressure Control valves technical specifications General Type indication HON C210 series Pressure class ANSI 300 and ANSI 600(*) Inlet pressure range 0 to 50 bar ANSI

### Differential pressure controller (PN 16) AHP - return mounting, adjustable setting

Data sheet Differential pressure controller (PN 16) AHP - return mounting, adjustable setting Description DN 15-40 DN 50 DN 65-100 AHP is a self-acting differential pressure controller primarily for use

### SITRANS F C MASSFLO. MASS flowmeters Sensor type MASS MC1 DN 50 - DN 100. Manual, supplement *083R9505* Order no.: FDK:521H1145

s Manual, supplement SITRANS F C MASSFLO MASS flowmeters [ ] Order no.: FDK:521H1145 SFIDK.PS.28.P3.2 - A5E253681 *83R955* Contents Page Introduction... 2 Siemens Flow Instruments range of coriolis mass

### Model GP PRESSURE REDUCING VALVE Installation & Operation Manual

Model GP-2000 PRESSURE REDUCING VALVE Installation & Operation Manual Please read this bulletin thoroughly before using the pressure reducing valve, so that you may do so correctly and safely. Please carefully

### SKP Series SKP25 U.. Pressure Regulating Gas Valve Actuator with Safety Shut-off Function

SKP Series SKP25 U.. Pressure Regulating Gas Valve Actuator with Safety Shut-off Function Technical Instructions Document No. 155-752 SKP25 U.. Description Only when assembled to Series VG Gas valves SKP25

### Differential pressure control and flow limiter AVPB/AVPB-F, PN 16

Data sheet Differential pressure control and flow limiter AVPB/AVPB-F, PN 16 Application AVPB is a self-acting differential pressure control and flow limiter primarily for use in district heating systems.

### MAXON M-PAKT ULTRA LOW NOx BURNERS

ULTRA LOW NOx BURNERS TECHNICAL CATALOG 3M-0007-0 TABLE OF CONTENTS Product description... 3 Features and benefits... 3 Typical emissions... 3 Model Numbers... Specifications of M-PAKT Burners... Materials

### VPPL VARIABLE DISPLACEMENT AXIAL-PISTON PUMPS FOR INTERMEDIATE PRESSURE SERIES 10

/ ED VPPL VARIABLE DISPLACEMENT AXIAL-PISTON PUMPS FOR INTERMEDIATE PRESSURE SERIES OPERATING PRINCIPLE The VPPL are variable displacement axial-piston pumps with variable swash plate, suitable for applications