# Unit 2 Pressure Measurement Techniques 1. State importance of pressure measurement in process industries.

Save this PDF as:

Size: px
Start display at page:

Download "Unit 2 Pressure Measurement Techniques 1. State importance of pressure measurement in process industries."

## Transcription

1 Unit 2 Pressure Measurement Techniques 1. State importance of pressure measurement in process industries. The measurement of pressure is one of the most important measurements, as it is used in almost all industries. Some important applications of pressure measurement is listed. 1. The pressure of steam in a boiler is measured for ensuring safe operating condition of the boiler. 2. Pressure measurement is done in continuous processing industries such as manufacturing and chemical industries. 3. Pressure measurement helps in determining the liquid level in tanks and containers. 4. Pressure measurement helps in determining the density of liquids. 5. In many flow meter (such as venturimeter, orifice meter, flow nozzle, etc.,) pressure measurement serves as an indication of flow rate. 6. Measurement of pressure change becomes an indication of temperature (as used in pressure thermometers-fluid expansion type). 7. Apart from this, pressure measurement is also required in day-to-day situations such as maintaining optimal pressure in tubes of vehicle tires. 2. Define Pressure. Pressure (P) is defined as the amount of force (F) acting per unit area (A). The mathematical equation for pressure can be written as: P= F/A = mg/a Where P is pressure F is the normal force (g is acceleration) and A is the area of the surface. Although the normal force is a vector quantity, pressure is a scalar quantity. The SI unit for pressure is the Pascal (Pa), equal to one newton per square meter (N/m2 or 1kg/(m-s2). 3. List units of pressure for low and high pressure measurement. 1 P a g e

2 4. Give conversion of 1 atm to bars and psi and torr (mmhg) to psi. 1 atm = bar = psi. 1 torr = 1 mmhg = x 10-3 psi. 5. List and define different types of pressure. Atmospheric pressure: The pressure due to air surrounding the earth s surface is called as atmospheric pressure. Absolute pressure: The pressure intensity measured from a state of prefect vacuum is called as absolute pressure. Gauge Pressure: A pressure measuring instrument generally measures the difference between the unknown pressure (p) and the atmospheric pressure (pa). When the unknown pressure (P) is greater than the atmospheric pressure (Pa), the pressure measured by the instrument is called as the gauge pressure. Vacuum pressure: A Pressure measuring instrument generally measures the difference between the unknown pressure (P) and the atmospheric pressure (Pa). When the atmospheric pressure (Pa) is greater than the unknown pressure (P), the pressure measured by the instrument is called as the vacuum pressure. Pressure relation 2 P a g e

3 Static Pressure: The pressure caused on the walls of the pipe due to a fluid at rest inside the pipe or due to the flow of a fluid parallel to the walls of the pipe is called as static pressure. This static pressure is measured by inserting a pressure measuring tube into the pipe carrying the fluid, so that the tube is at right angle to the fluid flow path. Dynamic or Impact or Velocity pressure. The pressure due to fluid velocity (flow speed) is called as impact pressure. Total or Stagnation pressure: 3 P a g e Impact pressure = Total pressure static pressure. The pressure which is obtained by bringing the flowing fluid to rest is called as total or stagnation pressure. Hence the total pressure will be a sum of static pressure and impact pressure. 6. Explain principle of working for Manometers. The term manometer is derived from the ancient Greek words 'manós', meaning thin or rare, and 'métron'. A manometer works on the principle of hydrostatic equilibrium and is used for measuring the pressure (static pressure) exerted by a still liquid or gas. Hydrostatic equilibrium states that the pressure at any point in a fluid at rest is equal, and its value is just the weight of the overlying fluid. For example, the weight of a column of mercury at 0 deg C that is one inch high and one inch in cross sectional area is.4892 pounds. Thus we can say that a column of mercury one inch high imposes a force of.4892 pounds per square inch or.4892 PSI. P=h ρ g where, ρ = density of the liquid used in manometer. Hence, ρg = specific weight of the liquid. In its simplest form, a manometer is a U-shaped tube consisting of an incompressible fluid like water or mercury. It is inexpensive and does not need calibration. 7. Explain Construction and Working of U tube manometer. Construction: It consists of a glass tube bent like the letter 'U'. In this type of manometer, balancing a column of liquid is done by another column of same or other liquid. One end of the U-tube is attached to the point where pressure is to be measured, while the other end is open to atmospheric pressure or connected to lower pressure. The pressure at point B in the figure is given by: P = ρ2 g h2 - ρ1g h1

4 where, ρ2 = density of heavy liquid h2 = height of heavy liquid above reference line ρ1 = density of light liquid h1 = height of light liquid above reference line. The manometer consists of a steel, brass and aluminum material. It has a glass tube made up of parallax glass. The graduations are made on the tube in terms of mm or in some condition it is graduated in kilo Pascal. Working: The unknown pressure is applied in the one arm of the tube and the mercury in the tube or manometeric liquid filled in the tube moves in the tube or rises to the constant region and then the movement is stopped. The height of the liquid is measured and pressure is calculated as per the equation given above. Limitations: - In the U-tube manometer, the application of pressure causes the liquid in one leg to go down while that in the other leg goes up, so there is no fixed reference. This tends to make the measurement of the height more difficult than it would be if one surface could be maintained at some fixed level. 8. Explain Construction and Working of Well-type manometer. The principles of manometric measurements have been discussed in reference to the U-type manometer. However, the manometer has been arranged in other forms to provide greater convenience and to meet varying service requirements. The well type manometer is one of these. Construction: As illustrated in Figure, if one leg of the manometer is increased many times in area to that of the other, the volume of fluid displaced will represent very little change of height in the smaller area leg. This condition results in an ideal arrangement whereby it is necessary to read only one convenient scale adjacent to a single indicating tube rather than two in the U-type. The larger area leg is called the "well". 4 P a g e

5 Working: The higher pressure source being measured must always be connected to the well connection "P". A lower pressure source must always be connected to the top of the tube, and a differential pressure must always have the higher pressure source connected at the well connection "P". In any measurement the source of pressure must be connected in a manner that will cause the indicating fluid to rise in the indicating tube. The true pressure still follows the principles previously outlined and is measured by the difference between the fluid surfaces. It is apparent that there must be some drop in the well level. 9. Explain Construction and Working of Barometer. A barometer is used to measure atmospheric pressure. Atmospheric pressure is a measure of the amount of force air exerts onto the earth as it is pushed down from the atmosphere. Another term for atmospheric pressure is barometric pressure. Construction: A classic mercury barometer is constructed from a glass tube that is open at one end and sealed at the opposite end. The glass tube is filled with liquid mercury and rests upside-down in a reservoir of mercury. As the mercury moves down the glass tube, a vacuum is created. Working: When the atmospheric pressure above the reservoir increases, the mercury inside the tube rises. As atmospheric pressure decreases the mercury moves down the tube, into the reservoir. This type of barometer was first constructed in 1643 by Evangelista Torricelli. The change in the mercury level in the glass tube is equal to the pressure exerted by the air above the reservoir and is measured using the scale marked on the glass tube. Changes in atmospheric pressure occur prior to weather changes. Sudden drops in atmospheric pressure indicate stormy weather, while sudden rises in pressure predicate brief periods of fair weather. Gradual increases and decreases in atmospheric pressure point to a more sustained weather pattern. 10. Explain Construction and Working of Inclined leg type manometer. Construction: It is similar to a well type manometer in construction. The only difference being that the vertical column limb is inclined at an angle θ. Inclined manometers are used for accurate measurement of small pressure. 5 P a g e

6 Working: Many applications require accurate measurement of low pressure such as drafts and very low differentials, primarily in air and gas installations. In these applications the manometer is arranged with the indicating tube inclined, as shown in Figure, therefore providing an expanded scale. This arrangement can allow 12" of scale length to represent 1" of vertical liquid height. With scale subdivisions to.01 inches of liquid height, the equivalent pressure of PSI per division can be read using water as the indicating fluid. 11. Explain Construction and working of float type manometer. Construction: Let us now look at float-type manometers. This is a variation of well type manometer. We have a well with much larger diameter, and we have another tube another leg whose diameter is less than this. These two this well and this leg is connected by flexible connection, and this well diameter is larger than such that you can put a float inside. Now, this is connected to the high pressure source this is connected to the low pressure source. Working: OR Now as the manometer liquid changes its position the flow inside the well also changes its position, and the position of the float can be taken as a measure of pressure. We can choose a large flow, such that it generates enough force to move a pointer against the scale. So, the scale 6 P a g e

7 can be calibrated in the unit of pressure and from the pointer under scale the pressure can be indicated. By replacing the pointer by a plain, and by replacing the scale by a charge this float-type manometer can be converted to a recording type manometer. It is also easy to change the span of the measurement by changing the diameter of this leg. So, this is frequently called range tube since these two are connected by a flexible connection, if I change the diameter of this leg the span of the instrument can be easily changed. So, this is essentially a variation of well-type manometer, where replace a float in the well. 12. Explain Construction and Working of micro manometer. The micro manometer is another variation of liquid column manometers that is based on the principle of inclined tube manometer and is used for the measurement of extremely small differences of pressure. Construction: It is an extension of inclined lag manometer with flexible tube attached with the magnifier. There is a well where higher pressure (P2) is applied and lower pressure (P1) is applied at the tube section. Micrometer screw is attached with well to measure the minute change in level of well. Working: The meniscus of the inclined tube is at a reference level as shown in the figure below, viewing through a magnifier provided with cross hair line. This is done for the condition, p1=p2. The adjustment is done by moving the well up and down a micrometer. For the condition p1 not equal to p2, the shift in the meniscus position is restored to zero by raising or lowering the well as before and the difference between these two readings gives the pressure difference in terms of height. 13. Explain Errors in Manometers. Effect of Temperature Manometers indicate the correct pressure at only one temperature. Another factor governing manometer s accuracy is the scale. As with indicating fluids, temperature changes affect the scale. At higher temperatures the scale will expand and graduations will be further apart. The opposite effect will occur at lower temperatures. Also the indicating fluid density changes with temperature. If water is the indicating fluid, an inch scale indicates one inch of water at 4 C only. On the same scale mercury indicates one inch of 7 P a g e

9 16. Explain Construction and Working of C-type Bourdon tube. Principle: When an elastic transducer (bourdon tube in this case) is subjected to a pressure, it deflects. This deflection is proportional to the applied pressure when calibrated. Construction: A C-type Bourdon tube consists of a long thin-walled cylinder of non-circular cross-section, sealed at one end, made from materials such as phosphor bronze, steel and beryllium copper, and attached by a light line work to the mechanism which operates the pointer. The other end of the tube is fixed and is open for the application of the pressure which is to be measured. The tube is soldered or welded to a socket at the base, through which pressure connection is made. Working: As the fluid under pressure enters the Bourdon tube, it tries to change the section of the tube from oval to circular, and this tends to straighten out the tube. The resulting movement of the free end of the tube causes the pointer to move over the scale. The tip of the Bourdon tube is connected to a segmental lever through an adjustable length link. The segmental lever end on the segment side is provided with a rack which meshes to a suitable pinion mounted on a spindle. The segmental lever is suitably pivoted and the spindle holds the pointer. Bourdon tubes are made of a number of materials, depending upon the fluid and the pressure for which they are used, such as phosphor bronze, alloy steel, stainless steel, Monel metal, and beryllium copper. Bourdon tubes are generally made in three shapes: C-type, Helical type and Spiral type 17. Which adjustments need to perform on bourdon tube? Explain each in brief. 9 P a g e

11 Construction: It is made up of rubber or other flexible material. Making a diaphragm slack rather than tight allows it to move large distance in response to a small pressure. A pointer is attached with the diaphragm via linkage. Pressure is applied at the input and is indicated on the scale. Working The diagram of a diaphragm pressure gauge is shown below. Unknown pressure is applied to the input (P1) of the gauge which will exerts force on the slack diaphragm. When a force acts against a thin stretched diaphragm, it causes a deflection of the diaphragm with its center deflecting the most. This movement is transferred to the pointer mechanism via leaf spring as shown in figure. Figure 1 Non Metallic Diaphragm Non-metallic or slack diaphragms are used for measuring very small pressures. The commonly used materials for making the diaphragm are polythene, neoprene, animal membrane, silk, and synthetic materials. Due to their non-elastic characteristics, the device will have to be opposed with external springs for calibration and precise operation. The common range for pressure measurement varies between 50 Pa to 0.1 MPa. Metallic Diaphragm Construction: Since the elastic limit has to be maintained, the deflection of the diaphragm must be kept in a restricted manner. This can be done by cascading many diaphragm capsules as shown in the figure 2. A main capsule is designed by joining two diaphragms at the periphery. A pressure inlet line is provided at the central position. Figure 2 Metallic Diaphragm Working: 11 P a g e

12 When the pressure enters the capsule, the deflection will be the sum of deflections of all the individual capsules. This will also deflect the lever connected to the diaphragm. Through pivot, this movement of lever is transferred to the pointer and scale mechanism. As shown in figure, corrugated diaphragms are also used instead of the conventional ones. 20. List advantages and disadvantages of Diaphragm pressure gauge. Advantages of Elastic diaphragm gauges: 1. Best advantage is they cost less 2. They have a linear scale for a wide range 3. They can withstand over pressure and hence they are safe to be used. 4. No permanent zero shift. 5. They can measure both absolute and gauge pressure, that is, differential pressure. Disadvantages of Elastic diaphragm gauges: 1. Shocks and vibrations affects their performance and hence they are to be protected. 2. When used for high pressure measurement, the diaphragm gets damaged. 3. These gauges are difficult to be repaired. 21. Explain Construction and Working of Bellows type pressure gauge. Like a diaphragm, bellows are also used for pressure measurement, and can be made of cascaded capsules. The basic way of manufacturing bellows is by fastening together many individual diaphragms. The bellows element, basically, is a one-piece expansible, collapsible and axially flexible member. It has many convolutions or fold. It can be manufactured form a single piece of thin metal. 12 P a g e

13 Construction: A bellows gauge contains an elastic element that is a convoluted unit that expands and contracts axially with changes in pressure. Most bellows gauges are spring-loaded; that is, a spring opposes the bellows, thus preventing full expansion of the bellows. Limiting the expansion of the bellows in this way protects the bellows and prolongs its life. In a spring-loaded bellows element, the deflection is the result of the force acting on the bellows and the opposing force of the spring. The movement of bellows is transferred to a pointer though a linkage. Bellows can also be used to measure differential pressure as shown in figure. Here two different pressure are applied to the two different pressure connection. Scale and Pointer is attached with gauge movement linkage at the center of the force bar. The bellows are connected between the input pressure connection and force bar. Working: The pressure to be measured is applied to the outside or inside of the bellows. However, in practice, most bellows measuring devices have the pressure applied to the outside of the bellows (see fig). As the inlet pressure varies, the bellows will expand or contract. This will move the linkage assembly and pointer will shows the applied pressure on the scale. For differential pressure measurement using bellows, applied differential pressure will try to imbalance the force bar and accordingly this movement is transferred to scale via gauge movement and pointer. Like Bourdon-tube elements, the elastic elements in bellows gauges are made of brass, phosphor bronze, stainless steel, beryllium-copper, or other metal that is suitable for the intended purpose of the gauge. Although some bellows instruments can be designed for measuring pressures up to 800 psig, their primary application is in the measurement of low pressures or small pressure differentials. 22. List advantages and disadvantages of Bellows type pressure gauge. Advantages Bellow joints do not require access; i.e. They can be direct buried, however a telltale is recommended No maintenance is required. Low cost Can be used to measured differential pressure Disadvantages Bellows joints can fail catastrophically. No in place maintenance or repair can be performed - they must be replaced if damaged. 13 P a g e

14 Require that the system to be shut down when a failure occurs. Smaller pressure range of application. Temperature compensation needed. 23. Explain principle of working for Strain gauge pressure Transducer. When force is applied to any metallic wire its length increases due to the strain. The more is the applied force, more is the strain and more is the increase in length of the wire. If L1 is the initial length of the wire and L2 is the final length after application of the force, the strain is given as: ε = (L2-L1)/L1 Further, as the length of the stretched wire increases, its diameter decreases. Now, we know that resistance of the conductor is the inverse function of the length. As the length of the conductor increases its resistance decreases. This change in resistance of the conductor can be measured easily and calibrated against the applied force. Thus strain gauges can be used to measure force and related parameters like pressure, displacement and stress. The input and output relationship of the strain gauges can be expressed by the term gauge factor or gauge gradient, which is defined as the change in resistance R for the given value of applied strain ε. 24. Explain Construction and Working of Strain gauge pressure transducer. Construction: Figure shows a bridge circuit with four strain gauges, Rsg1, R sg2, Rsg3 and R sg4. Two strain gauges, Rsg1 and R sg4 are mounted so that increasing pressure increases their resistance. Strain gauges Rsg2 and Rsg3, are mounted so that increasing pressure decreases their resistance. A change in temperature affects all the four strain gauges in the same way, resulting in no change in the pressure indication. Working: At balance, when there is no pressure, no current flows through the galvanometer G, and hence there will be no deflection in the galvanometer. As soon as the pressure is applied, the strain gauge stretches or compresses accordingly and the bridge circuit is unbalanced due to the change in resistance of the strain gauges. Thus, a current flows in the galvanometer, which is measured by the deflection of galvanometer. These changes affect the output of the bridge circuit, which indicates a change in measured pressure. Now, this change in output voltage may be calibrated for the pressure change. 14 P a g e

15 25. List advantages and disadvantages of Strain gauge pressure transducer. Advantages: No moving parts Small and inexpensive Fast response time Ease of compensation for temperature effects Relative freedom from acceleration effects Disadvantages: Non linear Needs calibration Inability to provide lower ranges (all) Low level outputs (all) Sensitive to environmental vibration (unbonded) Long term drift (all) Creep due to adhesive agents (semiconductor) 26. Explain principle of working for capacitive type pressure transducer. The principle of operation of capacitive transducers is based upon the equation for capacitance of a parallel plate capacitor as shown below. where: C is the capacitance of the Capacitor in farad C = ε 0ε r A D 15 P a g e

16 A = Overlapping area of plates; m 2, D = Distance between two plates; m, ε 0 = farad/m 2 ε r = Relative Permittivity (dielectric constant); F/m. The capacitive transducers work on the principle of change in capacitance of the capacitor. This change in capacitance could be caused by change in overlapping area A of the plates, change in the distance D between the plates and change in dielectric constant. 27. Explain Construction and Working of capacitive type pressure transducer. Construction: As shown in the figure below, a capacitive transducer has a static plate and a deflected flexible diaphragm with a dielectric in between. Diaphragm will expands and contracts due to change in pressure. These plats forms parallel plats capacitor which is connected as one arm of Wheatstone bridge circuit. Voltage output will give corresponding pressure measurement. Working: When a force is exerted to the outer side of the diaphragm the distance between the diaphragm and the static plate changes. This produces a capacitance change. This will unbalance the bridge and voltage output is measured in terms of applied pressure. 16 P a g e

17 28. List advantages and disadvantages capacitive type pressure transducer. Advantages: 1. Construction is very simple. 2. Cost of the transducer is low. 3. Has very high sensitivity. Disadvantages: 1. An increase or decrease in temperature to a high level will change the accuracy of the device. 2. Performance is affected by dirt and contaminations. 3. As the lead is lengthy, it can cause errors or distortion in signals. 29. Explain principle of working for LVDT. Definition of LVDT The term LVDT stands for the linear variable differential transformer. It is the most widely used inductive transducer that covert the linear motion into the electrical signals. It is called differential as the output across secondary of this transformer is the differential. They are very accurate inductive transducers as compared to other inductive transducers. Principle of LVDT: LVDT works under the principle of mutual induction, and the displacement which is a nonelectrical energy is converted into an electrical energy. 30. Explain Construction and Working of LVDT. Construction of LVDT Main Features of Construction are as, The transformer consists of a primary winding P and two secondary winding S1 and S2 wound on a cylindrical former (which is hollow in nature and will contain core). Both the secondary windings have equal number of turns and are identically placed on the either side of primary winding The primary winding is connected to an AC source which produces a flux in the air gap and voltages are induced in secondary windings. A movable soft iron core is placed inside the former and displacement to be measured is connected to the iron core. The iron core is generally of high permeability which helps in reducing harmonics and high sensitivity of LVDT. 17 P a g e

18 The LVDT is placed inside a stainless steel housing because it will provide electrostatic and electromagnetic shielding. The both the secondary windings are connected in such a way that resulted output is the difference of the voltages of two windings. Working: As the primary is connected to an AC source so alternating current and voltages are produced in the secondary of the LVDT. The output in secondary S1 is e1 and in the secondary S2 is e2. So the differential output is, eout = e1 - e2 This equation explains the principle of Operation of LVDT. Now three cases arise according to the locations of core which explains the working of LVDT are discussed below as, CASE I When the core is at null position (for no displacement) 18 P a g e

19 When the core is at null position then the flux linking with both the secondary windings is equal so the induced emf is equal in both the windings. So for no displacement the value of output eout is zero as e1 and e2 both are equal. So it shows that no displacement took place. CASE II When the core is moved to upward of null position (For displacement to the upward of reference point) In this case the flux linking with secondary winding S1 is more as compared to flux linking with S2. Due to this e1 will be more as that of e2. Due to this output voltage eout is positive. CASE III When the core is moved to downward of Null position (for displacement to the downward of reference point) In this case magnitude of e2 will be more as that of e1. Due to this output eout will be negative and shows the output to downward of reference point. Output VS Core Displacement A linear curve shows that output voltage varies linearly with displacement of core. Some important points about magnitude and sign of voltage induced in LVDT The amount of change in voltage either negative or positive is proportional to the amount of movement of core and indicates amount of linear motion. By noting the output voltage increasing or decreasing the direction of motion can be determined The output voltage of an LVDT is linear function of core displacement. 19 P a g e

20 31. List advantages and disadvantages of LVDT. Advantages of LVDT High Range - The LVDTs have a very high range for measurement of displacement. They can used for measurement of displacements ranging from 1.25 mm to 250 mm No Frictional Losses - As the core moves inside a hollow former so there is no loss of displacement input as frictional loss so it makes LVDT as very accurate device. High Input and High Sensitivity - The output of LVDT is so high that it doesn t need any amplification. The transducer possesses a high sensitivity which is typically about 40V/mm. Low Hysteresis - LVDTs show a low hysteresis and hence repeatability is excellent under all conditions Low Power Consumption - The power is about 1W which is very as compared to other transducers. Direct Conversion to Electrical Signals - They convert the linear displacement to electrical voltage which are easy to process Disadvantages of LVDT LVDT is sensitive to stray magnetic fields so they always require a setup to protect them from stray magnetic fields. They are affected by vibrations and temperature. It is concluded that they are advantageous as compared than any other inductive transducers. Applications of LVDT 1. They are used in applications where displacements ranging from fraction of mm to few cm are to be measured. The LVDT acting as a primary transducer converts the displacement to electrical signal directly. 2. They can also acts as the secondary transducers. E.g. the Bourbon tube which acts as a primary transducer and covert pressure into linear displacement. Then LVDT coverts this displacement into electrical signal which after calibration gives the ideas of the pressure of fluid. 32. Explain principle of working for Piezo Electric pressure transducer. The main principle of a piezoelectric transducer is that a force, when applied on the quartz crystal, produces electric charges on the crystal surface. The charge thus produced can be called as piezoelectricity. 20 P a g e

21 Piezo electricity can be defined as the electrical polarization produced by mechanical strain on certain class of crystals. The rate of charge produced will be proportional to the rate of change of force applied as input. As the charge produced is very small, a charge amplifier is needed so as to produce an output voltage big enough to be measured. 33. Explain Construction and Working of Piezo Electric pressure transducer. In 1880, Pierre and Jacques Curie determined that a small amount of voltage could be produced by applying large amounts of pressure to certain crystals of elements. This phenomenon is called the piezoelectric effect. Construction: When the piezoelectric effect is used in a pressure sensor, the sensor uses a diaphragm that deflects slightly when pressure is applied. A rod transfers this small amount of movement directly to the piezoelectric crystal Y1. The pressure on the crystal causes a small voltage to be produced that is proportional to the pressure. The voltage is amplified to traditional voltage signal values (0-10 volts) using charge amplifier. Crystal Y2 is for the purpose of compensation due to acceleration of device during use. Vibration is the major source of acceleration. Differential amplifier will subtracts all effects of accelerations and gives pressure alone. Working: When pressure is applied to the diaphragm, it will deform the crystal Y1 and produce a small voltage. The amount of voltage is proportional to the amount of deformation. The amount of voltage that is produced is very small and the internal impedance of the crystal is very large, which makes the use of op amps a necessity to produce a usable signal. Charge amplifier will amplify the signals from both the crystals Y1 and Y2. Subtracted voltage at output of differential amplifier will be the calibrated in terms of the input pressure. The best crystals that are used for this type of sensor come from ammonium dihydrogen phosphate and sintered ceramics. 21 P a g e

22 34. List advantages and disadvantages of Piezo Electric pressure transducer. Advantages 1. Very high frequency response. 2. Self-generating, so no need of external source. 3. Simple to use as they have small dimensions and large measuring range. 4. Barium titanate and quartz can be made in any desired shape and form. It also has a large dielectric constant. The crystal axis is selectable by orienting the direction of orientation. Disadvantages 1. It is not suitable for measurement in static condition. 2. Since the device operates with the small electric charge, they need high impedance cable for electrical interface. 3. The output may vary according to the temperature variation of the crystal. 4. The relative humidity rises above 85% or falls below 35%, its output will be affected. If so, it has to be coated with wax or polymer material. 35. Explain Construction and Working of optical type pressure transducer. Construction: Optical type pressure measurement is receiving considerable attention in recent years where the movement of a diaphragm, a bellows element or such other primary sensors are detected by optical means. The principle is nothing new, but the technique of adaptation in commercialization is varied in nature. A typical case with a diaphragm and a vane attached to it that covers and uncovers an irradiated photo diode with changing pressure is shown in the figure below. Reference diode is also used for the compensation. 22 P a g e

23 Working: The movement of elastic pressure sensors can be used to operate optical sensors. As the process pressure moves a diaphragm sensor, which in turn lifts a vane in front of an infrared light beam, the amount of light impinging on the measuring diode varies (Figure). A reference diode is also provided to compensate for the aging of the light source (LED) or for dirt buildup on the optics. Calibration may be made directly in pressure from output voltage of photo diode. The ratio metric technique is often preferred for avoiding drift error in electronic components as they are likely to be equally affected and cancelled. The vane movement or the diaphragm movement is kept small for negligible hysteresis and good precision. The range may be adjusted from (0-400) MPa with an accuracy of 0.1 percent scan. Advantages: Disadvantages: Application: 1. This transducer is insensitive to temperature variations, as such variations affect the measuring and reference diodes in the same way. 2. Because the amount of movement in the sensor is very small (0.5 mm), both the hysteresis and the repeatability errors are negligible. 1. Diode signals have non-linearity which may also vary from unit to unit. 2. Signal conditioning circuit may require more attention. 3. Temperature, though compensated, affects measurement to a certain extent which, in zero scale may be compensated by auto-zeroing facility. This system is often used as a null detecting one in a force balance type pressure measurement, where the servo-system brings the sensor to the zero balance point. 36. Explain principle of working for Dead Weight Tester. Working Principle: Dead Weight Tester is based on the principle of Pascal's law. The law states that in a closed system of incompressible fluid, the pressure applied will exert equal amount of force in all the directions. In Dead Weight Tester system, silicon oil is used within the closed boundaries of the Piston cylinder arrangement, piping, pressurization chamber and in the head on which the gauge to be tested/ calibrated is fixed. The oil is taken in to the pressurization chamber from oil bowl and all the air entrapped is vented off. 23 P a g e

25 1. The valve of the apparatus is closed. 2. A known weight is placed on the platform. 3. Now by operating the plunger, fluid pressure is applied to the other side of the piston until enough force is developed to lift the piston-weight combination. When this happens, the piston weight combination floats freely within the cylinder between limit stops. 4. In this condition of equilibrium, the pressure force of fluid is balanced against the gravitational force of the weights plus the friction drag. Therefore, Hence: Where, P = pressure PA = Mg + F P = (Mg + F) / A M = Mass; Kg g = Acceleration due to gravity; m/s² F = Friction drag; N A = Equivalent area of piston cylinder combination; m² 5. Thus the pressure P which is caused due to the weights placed on the platform is calculated. 6. After calculating P, the plunger is released. 7. Now the pressure gauge to be calibrated is fitted at an appropriate place on the dead weight tester. The same known weight which was used to calculate P is placed on the platform. Due to the weight, the piston moves downwards and exerts a pressure P on the fluid. Now the valve in the apparatus is opened so that the fluid pressure P is transmitted to the gauge, which makes the gauge indicate a pressure value. This pressure value shown by the gauge should be equal to the known input pressure P. If the gauge indicates some other value other than p the gauge is adjusted so that it reads a value equal to P. Thus the gauge is calibrated. Applications: It is used to calibrate all kinds of pressure gauges such as industrial pressure gauges, engine indicators and piezoelectric transducers. 38. List advantages and disadvantages of Dead Weight Tester. Advantages: It is simple in construction and easy to use. It can be used to calibrate a wide range of pressure measuring devices. Fluid pressure can be easily varied by adding weights or by changing the piston cylinder combination. 25 P a g e

26 Disadvantages: The accuracy of the dead weight tester is affected due to the friction between the piston and cylinder, and due to the uncertainty of the value of gravitational constant 'g'. 39. Explain Construction and Working of Ring balance type pressure gauge or manometer. This device cannot be actually called a manometer, but it is often considered so. Construction: The tube is made of polythene or other light and transparent material. This tube is bent into in to the form of a ring and is supported at the centre by a suitable pivot. The tubular chamber is divided in to two parts by spilling, sealing, and filling with a suitable light liquid like kerosene or paraffin oil for isolating the two pressures. Pressure taps are made with two flexible tubings. Pressures p1 and p2 act against the sealed walls as shown in the figure below, and rotate the ring which is balanced by the counter weight w. Working: The ring balance contains a sensitive element in the form of a hollow ring with a partition. A compensating weight is attached to the lower part of the ring, which is filled with a liquid (water, oil, or mercury). For p1 = p2, the liquid level in both sections of the ring is the same and the center of gravity of the weight is located on the vertical axis, which passes through the center of the ring. For p1 > p2, the liquid level in the left-hand part is lowered, and the liquid level in the righthand part is raised. The force created by the action of the pressure difference on the partition generates a moment that tends to turn the ring clockwise. This angle θ is used to measure the pressure. 26 P a g e

27 40. Explain construction and working of McLeod gauge. Principal: A known volume gas is compressed to a smaller volume whose final value provides an indication of the applied pressure. The gas used must obey Boyle s law given by; P1V1=P2V2 Where, P1 = Pressure of gas at initial condition (applied pressure). P2 = Pressure of gas at final condition. V1 = Volume of gas at initial Condition. V2 = Volume of gas at final Condition. Initial Condition = Before Compression. And Final Condition = After Compression. A known volume gas (with low pressure) is compressed to a smaller volume (with high pressure), and using the resulting volume and pressure, the initial pressure can be calculated. This is the principle behind the McLeod gauge operation. Construction: P 1 = P 2V 2 V 1 27 P a g e

28 A reference column is opened to unknown pressure B with reference capillary tube. The reference capillary tube has a point called zero reference point on open capillary A. This reference column is connected to a bulb and closed capillary and the place of connection of the bulb with reference column is called as cut off point. It is called the cut off point, since if the mercury level is raised above this point, it will cut off the entry of the applied pressure to the bulb and measuring capillary. Below the reference column and the bulb, there is a mercury reservoir operated by a flexible tube. Working: The gauge is used to compress a small quantity of low pressure gas to produce a readable large pressure. Bulb H of the gauge is attached to closed capillary. The mercury level in the gauge is lowered up to cut off by lowering the reservoir, thereby allowing a little process fluid to enter H. By raising the reservoir, the gas is now compressed in the closed capillary till mercury rises to the zero mark in the side tube and open capillary A. The capillary A is required to avoid any error due to capillary. The McLeod gauge is independent of gas composition. If, however, the gas contains condensable material and during compression it condenses, the reading of the gauge is faulty. The gauge is not capable of continuous reading and the scale is of square law type. The compression of the gas in a closed capillary makes the pressure of the trapped gas higher than the measured pressure. This pressure difference causes a difference in the mercury levels in the two tube. The difference in the height is used to calculate the pressure. The pressure can also be calculated using following equitation: Where P= Measured pressure P = KH H 0 (1 KH) K= a constant, determined by the geometry of the gauge H= difference in heights of the two mercury column H0= height of the top of the closed capillary tube above the zero line. Advantages of the McLeod Gauge: It is independent of the gas composition. It serves as a reference standard to calibrate other low pressure gauges. A linear relationship exists between the applied pressure and h There is no need to apply corrections to the McLeod Gauge readings. Limitations of McLeod Gauge: The gas whose pressure is to be measured should obey the Boyle s law 28 P a g e

29 Moisture traps must be provided to avoid any considerable vapor into the gauge. It measure only on a sampling basis. It cannot give a continuous output. 41. Explain principle of working for thermal conductivity gauge. Principle: A hot wire, placed within an envelope, will transfer thermal energy from the wire to any gas molecules that come into contact with it, and that energy will be again transferred to the walls of the envelope. With continual motion of the gas molecules, a thermal equilibrium will be reached as long as the number of gas molecules (pressure) remains constant. If, though, the pressure changes and the wire is resistively heated by current from a constant power source, a new thermal equilibrium will be reached, and the temperature of the wire will change to reflect the new number of gas molecules that can carry heat away from the wire. This means that the temperature of the wire can be used as an indication of the pressure within the envelope. This is the basic principle of all thermal conductivity gauges. The change in pressure vs. wire temperature remains fairly linear over a pressure range of about torr. Below this range, heat transfer is mostly by radiation from the wire s surface and mostly by thermal convection above it. Thermal conductivity gauges covering this range have been in use for many years that fall into two main groups: thermocouple gauges and Pirani gauges. 42. Explain Construction and Working of Pirani gauge. 29 P a g e

30 Construction: The main parts of the arrangement are: A pirani gauge chamber which encloses a platinum filament. A compensating cell to minimize variation caused due to ambient temperature changes. The pirani gauge chamber and the compensating cell is housed on a wheat stone bridge circuit as shown in diagram. Working: 1. A constant current is passed through the filament in the pirani gauge chamber. Due to this current, the filament gets heated and assumes a resistance which is measured using the bridge. 2. Now the pressure to be measured (applied pressure) is connected to the pirani gauge chamber. Due to the applied pressure the density of the surrounding of the pirani gauge filament changes. Due to this change in density of the surrounding of the filament its conductivity changes causing the temperature of the filament to change. 3. When the temperature of the filament changes, the resistance of the filament also changes. 4. Now the change in resistance of the filament is determined using the bridge. 5. This change in resistance of the pirani gauge filament becomes a measure of the applied pressure when calibrated. Note: [higher pressure higher density higher conductivity reduced filament temperature less resistance of filament] and vice versa. Applications of Pirani gauge Used to measure low vacuum and ultra-high vacuum pressures. Advantages of Pirani gauge They are rugged and inexpensive Give accurate results Good response to pressure changes. Relation between pressure and resistance is linear for the range of use. Readings can be taken from a distance. Limitations of Pirani gauge Pirani gauge must be checked frequently. Pirani gauge must be calibrated from different gases. Electric power is a must for its operation. 30 P a g e

31 43. Explain working of Ionization gauge. Principle: Boyle's law or the pressure-volume law states that the volume of a given amount of gas held at constant temperature varies inversely with the applied pressure when the temperature and mass are constant. V 1 P Another way to describing it is saying that their products are constant. When pressure goes up, volume goes down. When volume goes up, pressure goes down. From the equation above, this can be derived: The density (d) of a gas is defined as where m= mass of gas V= Volume of Gas P1V1 = P2V2 = P3V3 etc. d = m / V So now from above equation of pressure and volume, we can write for pressure and density as P1 d1 = P2 d2 Ionization gage measure vacuum by measuring the current produced by ionized gas molecules. It is known as ion current. The gas molecules are ionized as a stream of electrons collide with them. This current will be proportional to the density and pressure of the gas There are two method to produce gas ions. Based on this method, there are two types of ionization gauge. Alphatron Gauge: 1. Alphatron gauge 2. Hot filament ionization gauge The device uses alpha particles in order to ionize the gas in the vacuum chamber. The number of ions formed in the chamber is directly proportional to the gas density and pressure, if the chamber dimensions are shorter than the range of alpha particles. The figure below shows the schematic diagram of an alphatron. 31 P a g e

32 32 P a g e

33 The ions produced by the alpha particles are collected by the collector electrode and a current between Amperes will flow though the resistor R. The output voltage e0 is measured using a high input impedance output meter. The device has a range between 10 3 to 10-3 Torr. Hot Filament Ionization Gauge: In the hot Filament type, a column of gas is introduced into which, a potential difference V is applied to heater to create free electron in the space. This causes the electron with a charge to acquire a kinetic energy. This energy may be high enough to initiate ionization, and positive ions will be produced when the electrons collide with the gas molecules. The grid is maintained at a large positive potential with respect to the cathode and the plate. The plate is at a negative potential with respect to the cathode. The positive ions available between the grid and the cathode will be drawn by the cathode, and those between the grid and the plate will be collected by the plate. These ions creates a currents I1 and I2 which is proportional to density and pressure of the gas. 44. Explain Construction and Working of Pressure switch. A pressure switch turns an electric circuit ON or OFF at a preset pressure. This pressure is called the set point of the switch. A pressure switch is used in some form of control, e.g. to operate a solenoid valve at a given pressure, or start up a pump. Construction: The pressure switch is usually a micro switch or a mercury switch. A Burdon tube, a diaphragm or a bellows can be used to actuate the switch. Figure shows the simplest form of a pressure switch is used to actuate relay. Working: The pressure is fed to the inside of a bellows which carries a contact plate B. When pressure reaches a sufficient (or preset) value, the contact plate touches contact points C1 and C2, thus 33 P a g e

34 closing an electrical circuit to an alarm or motor control gear. The flexibility of the bellow ensures that the plate makes adequate contact with both points and gives a slight rubbing or wiping action that keeps the contact area clean. The pressure switch can be modified so as to make a low pressure contact in addition to a high pressure contact. It is adjustable. The contact in a pressure switch may be normally closed when the pressure is below the set point. For example, the contacts in a normally open switch remains open until the pressure rises above the set point. Then the sensing element makes the contacts snap to the closed position. The contacts open again when the pressure falls below the set point. The contacts in a normally closed switch remain closed until the pressure rises above the set point. Then the contacts snap open and remain open until the pressure drops below the set point again. Most switches contains two sets of contacts, one normally open and the other normally closed. A pressure switch has a dead band, i.e. the pressure must fall below the set point before the switch resets to its normal position. The amount of dead band is the difference in pressure between the set point and the reset point. The pressure switch is used to operate safety valve which vents steam when the pressure exceeds the upper limit. Uses of Pressure Switches: Following are the uses of pressure switches: One of the most important use of the pressure switch is in limiting pressure, e.g. in steam power plants. The pressure of the steam entering a turbine must not exceed an upper limit. The pressure switch is used to operate safety valve which vents steam when the pressure exceeds the upper limit. An important use of the pressure switch is in the computer panel. In the computer panel, blowers are used for cooling purposes. Whenever the blower fails due to any reason, a pressure switch is actuated which cuts off the power supply of the panel. Thus, the computer panel components are protected from the high temperature which can occur due to failure of the blowers. 45. Explain Pneumatic Differential pressure transmitter. Principle: It works on the principle of force balance. In the case of pressure instruments, pressure is easily converted into force by acting on the surface area of a sensing element such as a diaphragm or a bellows. A balancing force may be generated to exactly cancel the process pressure s force, making a force-balance pressure instrument. Like the laboratory balance scale, an industrial instrument built on the principle of balancing a sensed quantity with an adjustable quantity will be inherently linear, which is a tremendous advantage for measurement purposes. 34 P a g e

35 Here, we see a diagram of a force-balance pneumatic pressure transmitter, balancing a sensed differential pressure with an adjustable air pressure which becomes a pneumatic output signal: Force-Balance Pneumatic Pressure Transmitter balancing a sensed differential pressure transmitter with an adjustable air pressure which becomes a pneumatic output signal. Differential pressure is sensed by a liquid-filled diaphragm capsule, which transmits force to a force bar. If the force bar moves out of position due to this applied force, a highly sensitive baffle and nozzle mechanism senses it and causes a pneumatic amplifier (called a relay ) to send a different amount of air pressure to a bellows unit. The bellows presses against the range bar which pivots to counter-act the initial motion of the force bar. When the system returns to equilibrium, the air pressure inside the bellows will be a direct, linear representation of the process fluid pressure applied to the diaphragm capsule. 35 P a g e

36 46. Explain Electronic Differential pressure transmitter. With minor modifications to the design of this pressure transmitter2, we may convert it from pneumatic to electronic force-balancing: Differential pressure is sensed by the same type of liquid-filled diaphragm capsule, which transmits force to the force bar. If the force bar moves out of position due to this applied force, a highly sensitive electromagnetic sensor detects it and causes an electronic amplifier to send a different amount of electric current to a force coil. The force coil presses against the range bar which pivots to counteract the initial motion of the force bar. When the system returns to equilibrium, the mill ampere current through the force coil will be a direct, linear representation of the process fluid pressure applied to the diaphragm capsule. 36 P a g e

37 Based on the input pressure, diaphragm of the capsule deflects. This deflection is converted into an electrical signal. This is normally done by the sensors. The commonly used sensors are (a) Strain Gauge (b) Differential Capacitance (c) Vibrating wire. The sensor output is proportional to the applied pressure. Capacitance type: Strain Gauge Type: The electrical signal generated at the lower chamber by the sensor is in the range of milli-volt only. This signal is to be amplified to 0-5V or 0-10V range or is to be converted to 4-20mA for onward transmission to a remote instrument. This upper housing is the Transmitter portion of the DP Transmitter which houses the Electronic Unit. 2-Wire 4-20mA Current Transmitter: A DC output current is generated which is directly proportional to the pressure range of the Differential Pressure Transmitter. The lower range is 4mA, and the upper range is 20mA. This controlled current output is not affected by load impedance variation and supply voltage 37 P a g e

### Pressure Measurement

Pressure Measurement Manometers Sensors, Transducers Ashish J. Modi Lecturer, Dept. of Mech.Engg., Shri S.V.M. inst. Of Technology, Bharuch Pressure Pressure is a force per unit area exerted by a fluid

### Pressure Measurement. Introduction. Engr325 Instrumentation. Dr Curtis Nelson 3/12/18

3/12/18 Pressure Measurement Engr325 Instrumentation Dr Curtis Nelson Introduction A cluster of 72 helium-filled balloons over Temecula, California in April of 2003. The helium balloons displace approximately

### Instrumentation & Data Acquisition Systems

Instrumentation & Data Acquisition Systems Section 4 - Pressure Robert W. Harrison, PE Bob@TheHarrisonHouse.com Made in USA 1 Definition of Pressure Pressure is the amount of force applied perpendicular

### Pressure Measurement

Pressure Measurement Absolute and Gage Pressure P abs = P gage + P atm where P abs = Absolute pressure P abs = Gage pressure P abs = atmospheric pressure A perfect vacuum is the lowest possible pressure.

### Pressure Measurements

ME 22.302 Mechanical Lab I Pressure Measurements Dr. Peter Avitabile University of Massachusetts Lowell Pressure - 122601-1 Copyright 2001 A transducer is a device that converts some mechanical quantity

### Applied Thermodynamics. Experiment_01_MT_234. Instructor: Mr. Adnan Qamar. Mechanical Engineering Department

Applied Thermodynamics Experiment_01_MT_234 Instructor: Mr. Adnan Qamar Mechanical Engineering Department 1 Experiment No. 01: To test that Pressure is an intensive property. Apparatus: Nozzle Distribution

### Third measurement MEASUREMENT OF PRESSURE

1. Pressure gauges using liquids Third measurement MEASUREMENT OF PRESSURE U tube manometers are the simplest instruments to measure pressure with. In Fig.22 there can be seen three kinds of U tube manometers

### Pressure Measurement. Introduction. Engr325 Instrumentation. Dr Curtis Nelson 3/1/17

3/1/17 Pressure Measurement Engr325 Instrumentation Dr Curtis Nelson Introduction A cluster of 72 helium-filled balloons over Temecula, California in April of 2003. The helium balloons displace approximately

### Pressure measurement. Absolute pressure D Gauge pressure C Atmospheric pressure

Pressure measurement Pressure measurement is a very common requirement for most industrial process control systems and many different types of -sensing and -measurement systems are available. However,

### 3 1 PRESSURE. This is illustrated in Fig. 3 3.

P = 3 psi 66 FLUID MECHANICS 150 pounds A feet = 50 in P = 6 psi P = s W 150 lbf n = = 50 in = 3 psi A feet FIGURE 3 1 The normal stress (or pressure ) on the feet of a chubby person is much greater than

### Chapter 3 PRESSURE AND FLUID STATICS

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Chapter 3 PRESSURE AND FLUID STATICS Lecture slides by Hasan Hacışevki Copyright The McGraw-Hill

### Fluid Machinery Introduction to the laboratory measurements

Fluid Machinery Introduction to the laboratory measurements Csaba H s (csaba.hos@hds.bme.hu) Ferenc Hegedus (hegedusf@hds.bme.hu) February 21, 2014 1 Requirements related to the measurement part of the

### Instrumentation & Data Acquisition Systems

Instrumentation & Data Acquisition Systems Section 3 -Level Robert W. Harrison, PE Bob@TheHarrisonHouse.com Made in USA 1 Level Section Question Which level measuring technology is the best solution when

### INSTRUMENTATION EQUIPMENT

INSTRUMENTATION EQUIPMENT 2.0 INTRODUCTION Instrumentation is the art of measuring the value of some plant parameter, pressure, flow, level or temperature to name a few and supplying a signal that is proportional

### Level MEASUREMENT 1/2016

Level MEASUREMENT 1/2016 AGENDA 2 A. Introduction B. Float method C. Displacer method D. Hydrostatic pressure method E. Capacitance method G. Ultrasonic method H. Radar method I. Laser method J. Level

### PHYS 101 Previous Exam Problems

PHYS 101 Previous Exam Problems CHAPTER 14 Fluids Fluids at rest pressure vs. depth Pascal s principle Archimedes s principle Buoynat forces Fluids in motion: Continuity & Bernoulli equations 1. How deep

Hydrostatic pressure Consider a tank of fluid which contains a very thin plate of (neutrally buoyant) material with area A. This situation is shown in Figure below. If the plate is in equilibrium (it does

### DEVICES FOR FIELD DETERMINATION OF WATER VAPOR IN NATURAL GAS Betsy Murphy MNM Enterprises 801 N. Riverside Drive Fort Worth, Texas 76111

INTRODUCTION Water vapor in natural gas has more than a substantial effect on the quality of the gas stream. Without quality measurement of water vapor the gas is basically not saleable. Contracts are

### Irrigation &Hydraulics Department lb / ft to kg/lit.

CAIRO UNIVERSITY FLUID MECHANICS Faculty of Engineering nd Year CIVIL ENG. Irrigation &Hydraulics Department 010-011 1. FLUID PROPERTIES 1. Identify the dimensions and units for the following engineering

### PHYSICS - CLUTCH CH 17: FLUID MECHANICS.

!! www.clutchprep.com INTRO TO DENSITY LIQUIDS and GASES are types of. So we use the term to refer generally to both Liquids AND Gases. The DENSITY of a material is a measure of how tight the molecules

### The Experts in Vacuum Solutions

By Woodrow Farrow - Reprinted with permission from Specialty Gas Report Vacuum: a space that is relatively empty of matter, especially when that void has been created through artificial means. The earth

### Pressure measurement. Pressure gauges

Pressure gauge Pressure measurement There are four types of pressure measurements: Measurement of absolute pressure : Measurements are made with respect to absolute pressure, for example as with atmospheric

### This educational seminar discusses creating, measuring, and troubleshooting Rough Vacuum.

This educational seminar discusses creating, measuring, and troubleshooting Rough Vacuum. Specifically, today s talk will cover: Brief review of Vacuum Fundamentals Applications Using Rough Vacuum Rough

### Exercise 2-3. Flow Rate and Velocity EXERCISE OBJECTIVE C C C

Exercise 2-3 EXERCISE OBJECTIVE C C C To describe the operation of a flow control valve; To establish the relationship between flow rate and velocity; To operate meter-in, meter-out, and bypass flow control

### Phys101 Lectures Fluids I. Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle. Ref: 10-1,2,3,4,5,6,7.

Phys101 Lectures 21-22 Fluids I Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle Ref: 10-1,2,3,4,5,6,7. Page 1 10-1 Phases of Matter The three common phases of matter are solid,

### Objectives deals with forces applied by fluids at rest or in rigid-body motion.

Objectives deals with forces applied by fluids at rest or in rigid-body motion. The fluid property responsible for those forces is pressure, which is a normal force exerted by a fluid per unit area. discussion

### Module 2, Add on Lesson Depth Sensor. Teacher. 90 minutes

Module 2, Add on Lesson Depth Sensor 90 minutes Teacher Purpose of this lesson Investigate the relationship between pressure and depth Construct a sensor to measure the depth of water Graph data and reason

### A Journal of Practical and Useful Vacuum Technology. By Phil Danielson

A Journal of Practical and Useful Vacuum Technology From By Phil Danielson Thermal Conductivity Gauges Thermal conductivity pressure gauges are extremely common in vacuum technology, but an understanding

### ACCURACY, PERFORMANCE, AND HANDLING OF OIL-FILLED DIGIQUARTZ PRESSURE INSTRUMENTATION

Application Note Doc. G8108-001 Rev. A - 23-Jul-02 ACCURACY, PERFORMANCE, AND HANDLING OF OIL-FILLED DIGIQUARTZ PRESSURE INSTRUMENTATION For more information regarding Digiquartz products contact: Paroscientific,

### Chapter 13 Fluids. Copyright 2009 Pearson Education, Inc.

Chapter 13 Fluids Phases of Matter Density and Specific Gravity Pressure in Fluids Atmospheric Pressure and Gauge Pressure Pascal s Principle Units of Chapter 13 Measurement of Pressure; Gauges and the

### PRESSURE. 7. Fluids 2

DENSITY Fluids can flow, change shape, split into smaller portions and combine into a larger system One of the best ways to quantify a fluid is in terms of its density The density, ρ, of a material (or

### Chapter 13 Fluids. Copyright 2009 Pearson Education, Inc.

Chapter 13 Fluids Phases of Matter Density and Specific Gravity Pressure in Fluids Atmospheric Pressure and Gauge Pressure Pascal s Principle Units of Chapter 13 Measurement of Pressure; Gauges and the

### . In an elevator accelerating upward (A) both the elevator accelerating upward (B) the first is equations are valid

IIT JEE Achiever 2014 Ist Year Physics-2: Worksheet-1 Date: 2014-06-26 Hydrostatics 1. A liquid can easily change its shape but a solid cannot because (A) the density of a liquid is smaller than that of

### WHITEPAPER. The flowplus 16 Pressure Sensor. Physical Principles and Function

The flowplus 16 Pressure Sensor Physical Principles and Function Sensors (from the Latin "Sensire") are devices by means of which it is possible to detect physical or chemical properties of materials.

### Another convenient term is gauge pressure, which is a pressure measured above barometric pressure.

VACUUM Theory and Applications Vacuum may be defined as the complete emptiness of a given volume. It is impossible to obtain a perfect vacuum, but it is possible to obtain a level of vacuum, defined as

### Lecture 19 Fluids: density, pressure, Pascal s principle and Buoyancy.

Lecture 19 Water tower Fluids: density, pressure, Pascal s principle and Buoyancy. Hydraulic press Pascal s vases Barometer What is a fluid? Fluids are substances that flow. substances that take the shape

### COURSE NUMBER: ME 321 Fluid Mechanics I Fluid statics. Course teacher Dr. M. Mahbubur Razzaque Professor Department of Mechanical Engineering BUET

COURSE NUMBER: ME 321 Fluid Mechanics I Fluid statics Course teacher Dr. M. Mahbubur Razzaque Professor Department of Mechanical Engineering BUET 1 Fluid statics Fluid statics is the study of fluids in

### PNEUMATIC PRESSURE CONTROLLERS

PNEUMATIC PRESSURE CONTROLLERS VARIABLE VOLUME PRESSURE CONTROLLER MODELS: V-1 R AND V-2R The 3D Variable Volume Pressure Controller is available for requirements of 0-1,000 psi and 0-6,000 psi in absolute

### Chapter 9 Fluids and Buoyant Force

Chapter 9 Fluids and Buoyant Force In Physics, liquids and gases are collectively called fluids. 3/0/018 8:56 AM 1 Fluids and Buoyant Force Formula for Mass Density density mass volume m V water 1000 kg

### High Precision Gauge Pressure Indicators

Series 1500 8½" Dial High Precision Gauge Pressure Indicators Series 1500 Gauges have capsule-type pressure elements up to and including the 150-psig range. Ranges of 200 psig and above have Bourdon tubes.

### Pressure measurement explained

Rev A1, May 25 th, 2018 Sens4Knowledge Sens4 A/S Nordre Strandvej 119 G 3150 Hellebaek Denmark Phone: +45 8844 7044 Email: info@sens4.com www.sens4.com Introduction Pressure is defined as the force per

### Phys101 Lectures Fluids I. Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle. Ref: 10-1,2,3,4,5,6,7.

Phys101 Lectures 24-25 luids I Key points: Pressure and Pascal s Principle Buoyancy and Archimedes Principle Ref: 10-1,2,3,4,5,6,7. Page 1 10-1 Phases of Matter The three common phases of matter are solid,

### Unit 2 Kinetic Theory, Heat, and Thermodynamics: 2.A.1 Problems Temperature and Heat Sections of your book.

Unit 2 Kinetic Theory, Heat, and Thermodynamics: 2.A.1 Problems Temperature and Heat Sections 10.1 10.2 of your book. Convert the following to Celsius and Kelvin temperatures: 1. 80.0 o F Early E. C.:

### Unit A-2: List of Subjects

ES312 Energy Transfer Fundamentals Unit A: Fundamental Concepts ROAD MAP... A-1: Introduction to Thermodynamics A-2: Engineering Properties Unit A-2: List of Subjects Basic Properties and Temperature Pressure

1- (a) A water tank has a rectangular base of dimensions 1.5m by 1.2m and contains 1440 kg of water. Calculate (i) the weight of the water, weight =...... [1] (ii) the pressure exerted by the water on

### Properties of Fluids SPH4C

Properties of Fluids SPH4C Fluids Liquids and gases are both fluids: a fluid is any substance that flows and takes the shape of its container. Fluids Liquids and gases are both fluids: a fluid is any substance

### 1. All fluids are: A. gases B. liquids C. gases or liquids D. non-metallic E. transparent ans: C

Chapter 14: FLUIDS 1 All fluids are: A gases B liquids C gases or liquids D non-metallic E transparent 2 Gases may be distinguished from other forms of matter by their: A lack of color B small atomic weights

### 1. The principle of fluid pressure that is used in hydraulic brakes or lifts is that:

University Physics (Prof. David Flory) Chapt_15 Thursday, November 15, 2007 Page 1 Name: Date: 1. The principle of fluid pressure that is used in hydraulic brakes or lifts is that: A) pressure is the same

### How to specify a product. Process Sensors and Mechanical Instruments

How to specify a product Process Sensors and Mechanical Instruments Keep the overview. Here is some guideline information on how to specify our products. Intended as supplementary help to specification

### Fluid Mechanics. Chapter Three: Fluid Statics. Dr. Amer Khalil Ababneh

Fluid Mechanics Chapter Three: Fluid Statics Dr. Amer Khalil Ababneh This chapter deals with mechanics of fluids by introducing concepts related to pressure and by describing how to calculate forces associated

### Fluid Mechanics. Liquids and gases have the ability to flow They are called fluids There are a variety of LAWS that fluids obey

Fluid Mechanics Fluid Mechanics Liquids and gases have the ability to flow They are called fluids There are a variety of LAWS that fluids obey Density Regardless of form (solid, liquid, gas) we can define

### Precision level sensing with low-pressure module MS

The task on hand Level sensing as it is understood in modern process automation is much more than simply "tank half full" or "tank a quarter full". Using suitable sensors, levels, inlets and outlets can

### THERMODYNAMICS, HEAT AND MASS TRANSFER TUTORIAL NO: 1 (SPECIFIC VOLUME, PRESSURE AND TEMPERATURE)

THERMODYNAMICS, HEAT AND MASS TRANSFER TUTORIAL NO: 1 (SPECIFIC VOLUME, PRESSURE AND TEMPERATURE) 1. A vacuum gauge mounted on a condenser reads 66 cm Hg. What is the absolute pressure in the condenser

### Gas Physics Pressure and Flow Topics Covered:

Gas Physics Pressure and Flow Topics Covered: Molecular Theory of Gases Definition of Pressure The Gas Laws Definition of Flow Definition of Pressure Drop Gas Physics Pressure and Flow Topics Covered:

### Pressure calibration equipment

Pressure calibration equipment Contents Introduction 2 Oil operated deadweight testers DOS0015 (accuracy 0.015% of reading) 5 DOS0015 Dual Piston (accuracy 0.015% of reading) 6 DOS001 (accuracy 0.01% of

### Chapter 3: Fluid Statics. 3-1 Pressure 3-2 Fluid Statics 3-3 Buoyancy and Stability 3-4 Rigid-Body Motion

3-1 Pressure 3-2 Fluid Statics 3-3 Buoyancy and Stability 3-4 Rigid-Body Motion Chapter 3 Fluid Statics 3-1 Pressure (1) Pressure is defined as a normal force exerted by a fluid per unit area. Units of

### and its weight (in newtons) when located on a planet with an acceleration of gravity equal to 4.0 ft/s 2.

1.26. A certain object weighs 300 N at the earth's surface. Determine the mass of the object (in kilograms) and its weight (in newtons) when located on a planet with an acceleration of gravity equal to

### Investigation of Boyle s Law: methods

Name: Teacher: Class: Investigation of Boyle s Law: methods Your task is to investigate the relationship between volume, pressure and temperature for a gas. You must write detailed methods and select appropriate

### Model 4000 Pressure Controller

FEATURES Multiple Configurations The 4000 series pressure controller can be configured into either proportional only or proportional plus reset mode with a minimum of parts. Rugged Design Die cast aluminum

### SPD Pressure Sensor Families

DATASHEET SPD Pressure Sensor Families 1/7 Introduction to the principals of Smart Pressure Devices 1 Basic principles Pressure sensors are based on the principle of bending a membrane caused by the pressure

### Gases and Pressure SECTION 11.1

SECTION 11.1 Gases and In the chapter States of Matter, you read about the kineticmolecular theory of matter. You were also introduced to how this theory explains some of the properties of ideal gases.

### Differential Pressure Transmiter

Differential Pressure Transmiter Description The is an economical alternative to established differential pressure transmitters. It combines state of the art electronics and a high performance sensor;

### In the liquid phase, molecules can flow freely from position. another. A liquid takes the shape of its container. 19.

In the liquid phase, molecules can flow freely from position to position by sliding over one another. A liquid takes the shape of its container. In the liquid phase, molecules can flow freely from position

### Unit 7. Pressure in fluids

-- Unit 7. Pressure in fluids Index 1.- Pressure...2 2.- Fluids...2 3.- Pressure in fluids...3 4.- Pascal's principle...5 5.- Archimedes principle...6 6.- Atmospheric pressure...7 6.1.- Torricelli and

### Assumptions 1 At specified conditions, air behaves as an ideal gas. 2 The volume of the tire remains constant.

PTT 04/ Applied Fluid Mechanics Sem, Session015/016 ASSIGNMENT 1 CHAPTER AND CHAPTER 1. The air in an automobile tire with a volume of 0.0740 m is at 0 C and 140 kpa. Determine the amount of air that must

### In the liquid phase, molecules can flow freely from position to position by sliding over one another. A liquid takes the shape of its container.

In the liquid phase, molecules can flow freely from position to position by sliding over one another. A liquid takes the shape of its container. In the liquid phase, molecules can flow freely from position

Pressure Variation with Depth Pressure in a static fluid does not change in the horizontal direction as the horizontal forces balance each other out. However, pressure in a static fluid does change with

### FUNDAMENTALS OF PRESSURE REGULATORS ROBERT BENNETT MANAGER OF TRAINING ELSTER AMERICAN METER

FUNDAMENTALS OF PRESSURE REGULATORS ROBERT BENNETT MANAGER OF TRAINING ELSTER AMERICAN METER SUPPLY = DEMAND FUNCTION OF A REGULATOR A regulator may be defined as a "mechanism for controlling or governing

### Measurement of Atmospheric Pressure. Atmospheric Instrumentation

Measurement of Atmospheric Pressure Outline Measurement of Atmospheric Pressure Review of Atmospheric Pressure Barometers Liquid Aneroid Wind Speed Correction Significant Exposure Error Sea-Level Correction

### Measurement And Control Appreciation

Measurement And Control Appreciation Course Content Aim To provide an appreciation of the operation and application of process plant instrumentation used for the measurement and display of the main process

### Cover Page for Lab Report Group Portion. Drag on Spheres

Cover Page for Lab Report Group Portion Drag on Spheres Prepared by Professor J. M. Cimbala, Penn State University Latest revision: 29 September 2017 Name 1: Name 2: Name 3: [Name 4: ] Date: Section number:

### Barometry. The art or science of barometric observa3on

Barometry The art or science of barometric observa3on History Giovanni Ba3sta Baliani observed that syphon pumps could not pump water higher than ~ 34 feet Galileo - proposed it was due to a vacuum Gasparo

### Gas Laws. Introduction

Gas Laws Introduction In 1662 Robert Boyle found that, at constant temperature, the pressure of a gas and its volume are inversely proportional such that P x V = constant. This relationship is known as

### BAPI Pressure Line of Products - FAQs

Table of Contents 1. Several manufacturers produce pressure transmitters, why should I purchase from BAPI?... p. 2 2. BAPI makes several styles of pressure transmitters. What are the features of each?...

### Chapter 9 Solids and Fluids

2/17/16 Chapter 9 Solids and Fluids Units of Chapter 9 Solids and Elastic Moduli Fluids: Pressure and Pascal s Buoyancy and Archimedes Fluid Dynamics and Bernoulli s Surface Tension, Viscosity, and Poiseuille

### The Principles of Vacuum Technology

The Principles of Vacuum Technology Vacuum Terminology Vacuum units Vacuum regimes How to measure vacuum. Gauge designs. How to create vacuum Pump classifications and designs UHV compatibility considerations

### Middle East Technical University Department of Mechanical Engineering ME 305 Fluid Mechanics I Fall 2018 Section 4 (Dr.

Middle East Technical University Department of Mechanical Engineering ME 305 Fluid Mechanics I Fall 2018 Section 4 (Dr. Sert) Study Set 2 Reading Assignments You can find the answers of some of the following

### Chapter 15 Fluid. Density

Density Chapter 15 Fluid Pressure Static Equilibrium in Fluids: Pressure and Depth Archimedes Principle and Buoyancy Applications of Archimedes Principle By Dr. Weining man 1 Units of Chapter 15 Fluid

### Vacuum P=0. h=76 cm A B C. Barometer

Recap: Pressure Pressure = Force per unit area (P = F /A; units: Pascals) Density of object = mass / volume (ρ = m /V; units: kg / m 3 ) Pascal s Law:Pressure is transmitted equally in all directions throughout

### Torque Tube TB300 Digital Transmitters

Page 1 of 7 7E.300-E Issue 4-2009 Description Series TB300 torque tube liquid level instruments utilize the buoyancy exerted on a displacer when immersed in a liquid. The buoyancy on the displacer is proportional

### Lecture 20. Static fluids

Lecture 20 Static fluids Today s Topics: Density Pressure, Depth and Pressure Gauges Pascal s Principle Archimedes Principle Solids and Fluids Solids Maintain their shape. Generally don t flow Limited

### Barometry. History. Atmospheric Pressure 2/1/16. The art or science of barometric observa3on

Barometry The art or science of barometric observa3on History Giovanni Ba3sta Baliani observed that syphon pumps could not pump water higher than ~ 34 feet Galileo - proposed it was due to a vacuum Gasparo

### Liquid Level Measurement

Technical Article Liquid Level Measurement A pressure transmitter can be used to determine the liquid level in a tank, well, river or other body of liquid. The pressure at the bottom of a liquid filled

### BASIC PHYSICS APPLIED TO ANAESTHESIOLOGY

BASIC PHYSICS APPLIED TO ANAESTHESIOLOGY Dr.R.Selvakumar.M.D.D.A.DNB Professor of Anaesthesiology, K.A.P.Viswanatham Govt medical college, Trichy. The current practice of Anaesthesiology demands knowledge

### Density and Specific Gravity

Fluids Phases of Matter Matter is anything that has mass and takes up space (volume). The three common phases of matter are solid, liquid, and gas. A solid has a definite shape and size. A liquid has a

### Experiment. THE RELATIONSHIP BETWEEN VOLUME AND TEMPERATURE, i.e.,charles Law. By Dale A. Hammond, PhD, Brigham Young University Hawaii

Experiment THE RELATIONSHIP BETWEEN VOLUME AND TEMPERATURE, i.e.,charles Law By Dale A. Hammond, PhD, Brigham Young University Hawaii The objectives of this experiment are to... LEARNING OBJECTIVES introduce

### Chapter 5: Subatmospheric Total Pressure Gauges

Chapter 5: Subatmospheric Total Pressure Gauges The pressure gauges that will be described in this section are used to monitor the environment in a vacuum vessel so that processes conducted under subatmospheric

### Lecture Outline Chapter 15. Physics, 4 th Edition James S. Walker. Copyright 2010 Pearson Education, Inc.

Lecture Outline Chapter 15 Physics, 4 th Edition James S. Walker Chapter 15 Fluids Density Units of Chapter 15 Pressure Static Equilibrium in Fluids: Pressure and Depth Archimedes Principle and Buoyancy

### Process Equipment Design Guidelines Chapter Four Instrumentation Selection and Sizing (Engineering Design Guidelines)

Guidelines for Processing www.klmtechgroup.com Page : 1 of 56 Rev 1 - April 2007 Rev 2 Nov 2010 Rev 3 KLM Technology #03-12 Block Aronia, Jalan Sri Perkasa 2 Taman Tampoi Utama 81200 Johor Bahru Process

### MET 335W Fluid Mechanics Laboratory. Lab 1: Bourdon Tube Calibration. Nick Peak

MET 335W Fluid Mechanics Laboratory Lab 1: Bourdon Tube Calibration Nick Peak January 15, 2018 Purpose The purpose of this experiment is to test the accuracy of a bourdon pressure gauge. This is accomplished

### TEST FOR STABILOMETER VALUE OF BITUMINOUS MIXTURES

Test Procedure for TEST FOR STABILOMETER VALUE OF BITUMINOUS MIXTURES TxDOT Designation: Tex-208-F Effective Date: February 2005 1. SCOPE 1.1 Use this test method to determine the Hveem stability value

### Cambridge International Examinations Cambridge International General Certificate of Secondary Education

Cambridge International Examinations Cambridge International General Certificate of Secondary Education *8269683414* PHYSICS 0625/31 Paper 3 Extended October/November 2014 1 hour 15 minutes Candidates

### Air Operated Hydraulic Pumping Systems to 50,000 psi

High Pressure Equipment Air Operated Hydraulic Pumping Systems to 50,000 psi PS-10: 10,000 psi PS-20: 20,000 psi PS-30: 30,000 psi PS-40: 40,000 psi PS-50: 50,000 psi PS-90: 90,000 psi High Pressure air

### Detectors and Indicators for Industrial Facility Systems

Detectors and Indicators for Industrial Facility Systems Course No: D05-002 Credit: 5 PDH Gilbert Gedeon, P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980 P:

### Introductory Lab: Vacuum Methods

Introductory Lab: Vacuum Methods Experiments in Modern Physics (P451) In this lab you will become familiar with the various components of the lab vacuum system. There are many books on this topic one of